Describe CVE-2015-3954.
What is CVE-2015-3954?
Hospira Plum A+ Infusion System version 13.4 and prior, Plum A+3 Infusion System version 13.6 and prior, and Symbiq Infusion System, version 3.13 and prior give unauthenticated users root privileges on Port 23/TELNET by default. An unauthorized user could issue commands to the pump. Hospira recommends that customers close Port 23/TELNET on the affected devices. Hospira has also released the Plum 360 Infusion System which is not vulnerable to this issue..