def pulse_odmr_setup(inst):
    """
        Sets up PPG output channels for pulsed ODMR measurements.
        """
    print('Identification: {}'.format(inst.idn))
    inst.reset()
    inst.display = 'off'
    print('Display off?: {}'.format(inst.display))
    inst.pattern_mode = 'on'
    print('Digital pattern mode on?:{}'.format(inst.pattern_mode))
    inst.frequency = 660000000.0
    inst.volt_low[1] = 0.0
    inst.volt_high[1] = 2.5
    print('MW TTL high voltage, should be 2.5 V:{}'.format(inst.volt_high[1]))
    print('MW TTL low voltage, should be 0 V:{}'.format(inst.volt_low[1]))
    inst.data_format[1] = 'nrz'
    inst.output_on[1] = 'on'
    inst.comp_output_on[1] = 'on'
    inst.volt_low[2] = 0.0
    inst.volt_high[2] = 1.0
    print('AOM high voltage, should be 1.0 V:{}'.format(inst.volt_high[2]))
    print('AOM Low voltage, should be 0 V:{}'.format(inst.volt_low[2]))
    inst.data_format[2] = 'nrz'
    inst.output_on[2] = 'on'
    inst.comp_output_on[2] = 'on'
    print('Trigger type:{}'.format(inst.trig_output))
    print('Channel 1 timed_delay:{}'.format(inst.timed_delay[1]))
    print('Channel 2 timed_delay:{}'.format(inst.timed_delay[2]))
