Thats a unique idea.You should consider the following idea to enhance your idea so that it would impact uniquely: , a new approach to designing fast micro-psychrometers devoted to measuring relative humidity is presented. Such devices are based on a planar thermopile fitted with an appropriate set of microchannels arranged for monitoring the evaporation process as a Seebeck voltage. They are fabricated out of a glass substrate, sputtered metals, and near vertical side walls made of SU-8, according to standard microtechnology processes for MEMs. Considering classical psychrometers, the new design is aimed at reducing both size and response time. Thus, besides being far less invasive than a standard unit, fast measurements (tau=5 s) are enabled. A straightforward modeling, aimed at deriving the relative humidity, is given with a view to integrating the signal conditioner. As regards validation, a prototype is calibrated and compared to a reference probe, and a discussion on its enhancement is also presented
['Nicolas Giordani', 'Lionel Camberlein', 'E. Gaviot', 'F. Polet', 'Nicolas Pelletier', 'Bruno Bêche']
I am writing a research paper on a idea Fast Psychrometers as New SU-8 Based Microsystems
[{"from": "human", "value": "I am writing a research paper on a idea Fast Psychrometers as New SU-8 Based Microsystems"}, {"from": "gpt", "value": "Thats a unique idea.You should consider the following idea to enhance your idea so that it would impact uniquely: , a new approach to designing fast micro-psychrometers devoted to measuring relative humidity is presented. Such devices are based on a planar thermopile fitted with an appropriate set of microchannels arranged for monitoring the evaporation process as a Seebeck voltage. They are fabricated out of a glass substrate, sputtered metals, and near vertical side walls made of SU-8, according to standard microtechnology processes for MEMs. Considering classical psychrometers, the new design is aimed at reducing both size and response time. Thus, besides being far less invasive than a standard unit, fast measurements (tau=5 s) are enabled. A straightforward modeling, aimed at deriving the relative humidity, is given with a view to integrating the signal conditioner. As regards validation, a prototype is calibrated and compared to a reference probe, and a discussion on its enhancement is also presented"}]