Thats a unique idea.You should consider the following idea to enhance your idea so that it would impact uniquely:  fabricated a carbon nanotube (CNT)/ polydimethylsiloxane (PDMS) composite-based dry ECG electrode that can be readily connected to conventional ECG devices, and shod its long-term arable monitoring capability and robustness to motion and sat. While the dispersion of CNTs in PDMS is challenging,  optimized the process to disperse untreated CNTs within PDMS by mechanical force only. The electrical and mechanical characteristics of the CNT/PDMS electrode re tested according to the concentration of CNTs and its thickness. The performances of ECG electrodes re evaluated by using 36 types of electrodes which re fabricated with different concentrations of CNTs, and with a differing diameter and thickness. The ECG signals re obtained by using electrodes of diverse sizes to observe the effects of motion and sat, and the proposed electrode was shown to be robust to both factors. The CNT concentration and diameter of the electrodes re critical parameters in obtaining high-quality ECG signals. The electrode was shown to be biocompatible from the cytotoxicity test. A seven-day continuous arability test shod that the quality of the ECG signal did not degrade over time, and skin reactions such as itching or erythema re not observed. This electrode could be used for the long-term measurement of other electrical biosignals for ubiquitous health monitoring including EMG, EEG, and ERG.
['Ha Chul Jung', 'Jin Hee Moon', 'Dong Hyun Baek', 'Jae Hee Lee', 'Yoon Young Choi', 'Joung Sook Hong', 'Sang Hoon Lee']
I am writing a research paper on a idea CNT/PDMS Composite Flexible Dry Electrodesfor Long-Term ECG Monitoring
[{"from": "human", "value": "I am writing a research paper on a idea CNT/PDMS Composite Flexible Dry Electrodesfor Long-Term ECG Monitoring"}, {"from": "gpt", "value": "Thats a unique idea.You should consider the following idea to enhance your idea so that it would impact uniquely:  fabricated a carbon nanotube (CNT)/ polydimethylsiloxane (PDMS) composite-based dry ECG electrode that can be readily connected to conventional ECG devices, and shod its long-term arable monitoring capability and robustness to motion and sat. While the dispersion of CNTs in PDMS is challenging,  optimized the process to disperse untreated CNTs within PDMS by mechanical force only. The electrical and mechanical characteristics of the CNT/PDMS electrode re tested according to the concentration of CNTs and its thickness. The performances of ECG electrodes re evaluated by using 36 types of electrodes which re fabricated with different concentrations of CNTs, and with a differing diameter and thickness. The ECG signals re obtained by using electrodes of diverse sizes to observe the effects of motion and sat, and the proposed electrode was shown to be robust to both factors. The CNT concentration and diameter of the electrodes re critical parameters in obtaining high-quality ECG signals. The electrode was shown to be biocompatible from the cytotoxicity test. A seven-day continuous arability test shod that the quality of the ECG signal did not degrade over time, and skin reactions such as itching or erythema re not observed. This electrode could be used for the long-term measurement of other electrical biosignals for ubiquitous health monitoring including EMG, EEG, and ERG."}]