Thats a unique idea.You should consider the following idea to enhance your idea so that it would impact uniquely: A two-way Josephson voltage standard (JVS) direct comparison beten the National Institute of Standards and Technology (NIST) and the National Research Council (NRC) has been conducted. The process consists of two comparisons: first, using the NRC JVS with the NRC's measuring system (hardware and software) to measure the 10 V provided by the NIST JVS and then using the NIST JVS measuring system to measure the 10 V provided by the NRC JVS. The results of the two comparisons are in agreement to within 0.7 nV, and their mean indicates that the difference beten the two JVSs at 10 V is -0.28 nV, with a pooled combined uncertainty of 2.07 nV ( k  = 2) or a relative uncertainty of 2.1 parts in 10 10 .
['Yi-hua Tang', 'B.M. Wood', 'C.A. Hamilton']
I am writing a research paper on a idea A two-way Josephson voltage standard comparison between NIST and NRC
[{"from": "human", "value": "I am writing a research paper on a idea A two-way Josephson voltage standard comparison between NIST and NRC"}, {"from": "gpt", "value": "Thats a unique idea.You should consider the following idea to enhance your idea so that it would impact uniquely: A two-way Josephson voltage standard (JVS) direct comparison beten the National Institute of Standards and Technology (NIST) and the National Research Council (NRC) has been conducted. The process consists of two comparisons: first, using the NRC JVS with the NRC's measuring system (hardware and software) to measure the 10 V provided by the NIST JVS and then using the NIST JVS measuring system to measure the 10 V provided by the NRC JVS. The results of the two comparisons are in agreement to within 0.7 nV, and their mean indicates that the difference beten the two JVSs at 10 V is -0.28 nV, with a pooled combined uncertainty of 2.07 nV ( k  = 2) or a relative uncertainty of 2.1 parts in 10 10 ."}]