Thats a unique idea.You should consider the following idea to enhance your idea so that it would impact uniquely: A novel micro-needle array electrode (MAE) fabricated by thermal drawing and coated with Ti/Au film was proposed for bio-signals monitoring. A simple and effective setup was employed to form glassy-state poly (lactic-co-glycolic acid) (PLGA) into a micro-needle array (MA) by the thermal drawing method. The MA was composed of 6 × 6 micro-needles with an average height of about 500 μm. Electrode-skin interface impedance (EII) was recorded as the insertion force was applied on the MAE. The insertion process of the MAE was also simulated by the finite element method. Results shod that MAE could insert into skin with a relatively low compression force and maintain stable contact impedance beten the MAE and skin. Bio-signals, including electromyography (EMG), electrocardiography (ECG), and electroencephalograph (EEG) re also collected. Test results shod that the MAE could record EMG, ECG, and EEG signals with good fidelity in shape and amplitude in comparison with the commercial Ag/AgCl electrodes, which proves that MAE is an alternative electrode for bio-signals monitoring.
['Lei Ren', 'Qing Jiang', 'Keyun Chen', 'Zhipeng Chen', 'Chengfeng Pan', 'Lelun Jiang']
I am writing a research paper on a idea Fabrication of a Micro-Needle Array Electrode by Thermal Drawing for Bio-Signals Monitoring
[{"from": "human", "value": "I am writing a research paper on a idea Fabrication of a Micro-Needle Array Electrode by Thermal Drawing for Bio-Signals Monitoring"}, {"from": "gpt", "value": "Thats a unique idea.You should consider the following idea to enhance your idea so that it would impact uniquely: A novel micro-needle array electrode (MAE) fabricated by thermal drawing and coated with Ti/Au film was proposed for bio-signals monitoring. A simple and effective setup was employed to form glassy-state poly (lactic-co-glycolic acid) (PLGA) into a micro-needle array (MA) by the thermal drawing method. The MA was composed of 6 \u00d7 6 micro-needles with an average height of about 500 \u03bcm. Electrode-skin interface impedance (EII) was recorded as the insertion force was applied on the MAE. The insertion process of the MAE was also simulated by the finite element method. Results shod that MAE could insert into skin with a relatively low compression force and maintain stable contact impedance beten the MAE and skin. Bio-signals, including electromyography (EMG), electrocardiography (ECG), and electroencephalograph (EEG) re also collected. Test results shod that the MAE could record EMG, ECG, and EEG signals with good fidelity in shape and amplitude in comparison with the commercial Ag/AgCl electrodes, which proves that MAE is an alternative electrode for bio-signals monitoring."}]