Thats a unique idea.You should consider the following idea to enhance your idea so that it would impact uniquely: To measure precise blood pressure (BP) and pulse rate without using a cuff,  have developed an arterial pulsimeter consisting of a small, portable apparatus incorporating a Hall device. Regression analysis of the pulse wave measured during testing of the arterial pulsimeter was conducted using two equations of the BP algorithm. The estimated values of BP obtained by the cuffless arterial pulsimeter over 5 s re compared with values obtained using electronic or liquid mercury BP meters. The standard deviation beten the estimated values and the measured values for systolic and diastolic BP re 8.3 and 4.9, respectively, which are close to the range of values of the BP International Standard. Detailed analysis of the pulse wave measured by the cuffless radial artery pulsimeter by detecting changes in the magnetic field can be used to develop a new diagnostic algorithm for BP, which can be applied to new medical apparatus such as the radial artery pulsimeter.
['Sang-Suk Lee', 'Dong-Hyun Nam', 'You-Sik Hong', 'Woo-Beom Lee', 'Il-Ho Son', 'Keun Ho Kim', 'Jong-Gu Choi']
I am writing a research paper on a idea Measurement of Blood Pressure Using an Arterial Pulsimeter Equipped with a Hall Device
[{"from": "human", "value": "I am writing a research paper on a idea Measurement of Blood Pressure Using an Arterial Pulsimeter Equipped with a Hall Device"}, {"from": "gpt", "value": "Thats a unique idea.You should consider the following idea to enhance your idea so that it would impact uniquely: To measure precise blood pressure (BP) and pulse rate without using a cuff,  have developed an arterial pulsimeter consisting of a small, portable apparatus incorporating a Hall device. Regression analysis of the pulse wave measured during testing of the arterial pulsimeter was conducted using two equations of the BP algorithm. The estimated values of BP obtained by the cuffless arterial pulsimeter over 5 s re compared with values obtained using electronic or liquid mercury BP meters. The standard deviation beten the estimated values and the measured values for systolic and diastolic BP re 8.3 and 4.9, respectively, which are close to the range of values of the BP International Standard. Detailed analysis of the pulse wave measured by the cuffless radial artery pulsimeter by detecting changes in the magnetic field can be used to develop a new diagnostic algorithm for BP, which can be applied to new medical apparatus such as the radial artery pulsimeter."}]