fig1 is a schematic illustration of a cross section of a human heart showing a catheter stimulating the heart . here the right and left atria 3 , 5 and right and left ventricles 9 , 11 are shown . an upper section atrial his bundle 71 of a his perkinje nerve fiber bundle 70 ( his bundle ) is present in the wall of right atrium 3 which passes down a septum 13 to a right ventricle 9 and a left ventricle 11 . in normal operation , electrical stimulation of the atrial his bundle 71 causes the atria 3 , 5 to simultaneously contract passing blood through the tricuspid valve 17 and into the right ventricle 9 . the signal moves slowly downward into the right his bundle branch 73 and the left his bundle branch 75 causing right and left : ventricles 9 , 11 , respectively , to begin to contract as the atria 3 , 5 have just about completed their contraction . the delayed contraction of ventricles 9 , 11 allows maximal ejection fraction of the heart and optimum pumping efficiency . when the timing is thrown off , by disease or other disorder , the efficiency drops considerably . this may be due to a non - conductive section of the his bundle . thus , it is very important to correct contractions regarding improper timing . usually , a cardiac pacemaker system is used to correct said contractions . cardiac pacemaker systems ( pacemakers ) typically comprise : a power / control unit ( not depicted ); at least one electrode 121 ; a wire lead 103 ; an electromechanical connector 131 ; and a stylet 141 . electrode 121 is permanently attached to one end of wire lead 103 and an electromechanical connector 131 is permanently attached to the other end of wire lead 103 . electrode 121 typically comprises an electrode collar 127 and a helix 122 . electrode collar 127 is located on the distal surface of electrode 121 . see fig3 . electrode collar 127 is typically the outer ring on the distal surface of the electrode 121 . electrode collar 127 functions to make an electrical connection to heart tissue . helix 122 is a corkscrew shaped screw capable of being inserted into heart tissue by screw means . helix 122 typically has more rounded bends rather than sharp cornered bends as depicted in fig3 . helix 122 has a tip 123 that may be used to puncture heart tissue in order to retrieve a proper electrical reading or to start the screw attachment means . helix 122 also functions to make an electrical connection to heart tissue . wire lead 103 includes a catheter , cannula , or tube 105 running longitudinally along the full length of wire lead 103 , where one end of cannula 105 leads to electrode 21 and the other end to the electromechanical connector 131 , where cannula 105 extends through electromechanical connector . cannula 105 is sized to accept a specific stylet 141 where they are typically designed in pairs , where stylet 141 is a rigid or semi - rigid wire used to guide , move , and position electrode 121 inside a body cavity . stylet 141 comprises and guide wire 143 and a wrench 145 . 143 and 145 are rigidly connected so that each rotates along with the other . the whole structure of 141 is also known as a guide - wire . in order to use stylet 141 , guide wire 143 must be completely inserted into cannula 105 and seated onto the proximal end of electrode 121 . guide wire 143 may be pre - formed in various shapes such as straight or j - shaped in order to allow easier positioning and attaching of electrode 121 . typically , electromechanical connector 131 is an “ is - 1 pacemaker connector ”. is - 1 pacemaker connector comprises two electromechanical connectors : 133 and 135 . electromechanical connector 133 is electrically connected to electrode collar 127 and electromechanical connector 135 is electrically connected to helix 122 . electromechanical connectors 133 and 135 are each mechanically connected to different structure on electrode 121 and each may be rotated relative to the other , thereby rotating different structure on electrode 121 , relative to the other . with rotational screw electrodes , either stylet 141 or electromechanical connector 133 or both must be rotated in order to extend helix 122 from the center of electrode collar 127 . alternately , with fixed screw electrodes , either stylet 141 or electromechanical connector 133 or both must be rotated in order to start helix 122 to screw into heart tissue . there are many specifically designed electrodes 121 on the market ; however , practically all include an is - 1 pacemaker connector at their proximal end , where each design has its own specifically designed stylet 141 or set of such that may be used for a specific purposes such as attaching fixed screw or rotational screw electrodes of other function . this invention is first to provide continuously viable electrical connections between connectors 133 and 135 with concurrent ability to rotate such independently from each other , thereby yielding the ability to produce a high - quality electrocardiogram of the his bundle electric potential with continuous viable electric potential readings during the complete process of mapping , rotating of electrode or helix or both , screw attachment to heart tissue , and afterwards . pacemakers typically function by inserting stimulation electrode 121 into the superior vena cava , through the tricuspid valve 17 and rest it on the ventricle muscle , as described in thompson . wire lead 103 and electrode 121 must pass through catheter 101 for body insertion . this is not optimal because wire lead passes through the tricuspid valve and thus may interfere with the valve or cause an irritation . pacemakers are intended to remain in place permanently , thus lead 103 may have long - term effects or otherwise cause damage to the valve and inefficiency of its operation . also , the electrical stimulation of the right ventricle 9 also has long - term adverse effects on the muscle . to avoid these issues and to have better pacing , the pacing electrode should be placed and anchored into the his bundle 71 ( or alternately anchored to septum wall 13 ) so that the pacing electrode is electrically connected to his bundle 71 . the his bundle tissue appears to lie in a small fossa at the apex of the membranous septum , where mapping is required in order to find his bundle 70 . mapping is the process of positioning the electrode 121 , recording an electrical potential reading , repositioning the electrode 121 , recording another electrical signal , and repeating until an accurate enough map or sufficient amount of electrical readings is taken by the surgeon to render an electrical topography of the heart chamber or otherwise enough information to determine the exact position on the his bundle to attach the electrode in order to stimulate the simultaneous contraction of both ventricles in order to achieve maximum efficiency . the electrical potential reading used in the mapping process is the electric potential difference between electrode collar 127 and helix tip 123 . the electric potential has maximum amplitude when the electrode 121 is closest to the his bundle center , with the tip of the helix 123 located exactly on center of the his bundle . as noted in deshmukh , successful mapping requires a high - quality electrocardiogram of the his bundle electric potential that must be recorded and monitored by electrode 121 . electric potential mapping and electrode attachment device may be used to easily map the heart chamber in order to determine the optimal precise position for electrode 121 . electrode 121 is moved by manipulating the electric potential mapping and electrode attachment device 200 where 200 is also used as a continuous interface between the recording system ( not shown ) and the electrode 121 as the movement occurs . electric potential mapping and electrode attachment device comprises a main cylinder 210 and a secondary cylinder 250 , wherein said cylinders are concentric and mounted on a common longitudinal center axis with said secondary cylinder 250 fitting completely inside of said main cylinder 210 . common longitudinal axis is void or hollow in order to receive the stylet 141 , which must extend there through , where proper function requires that stylet 141 be threaded down the center of electric potential mapping and electrode attachment device as depicted . typically , electrode lead 103 is attached first and then stylet 141 is threaded through electric potential mapping and electrode attachment device and electrode lead 103 . there are at least two openings 211 in main cylinder 210 , through which said secondary cylinder 250 may be easily rotated by finger or hand within the main cylinder 210 and independently from main cylinder 210 . openings 211 are sized to be slightly larger than an average human thumbprint . openings 211 are oval shaped in best mode . secondary cylinder 250 has at least two ridges or other raised areas 251 on its exterior surface to provide for increased grip or friction during finger rotation of 250 within 210 . best mode 250 has eight ridges 251 . electric potential mapping and electrode attachment device further comprises a proximal end button 271 and a distal end cap 273 . proximal end button 271 is generally cylindrical with outer diameter slightly smaller than the inner diameter of main cylinder 210 . end button 271 is placed concentrically within cylinder 210 so that button 271 can slide longitudinally within cylinder 210 . proximal end of cylinder 210 has a reduced diameter end or rounded end , which acts to contain button 271 within cylinder 210 . sliding action of end button 271 within cylinder 210 is retained on the other end by spring 280 . distal cap 273 functions to provide a distal member or end of cylinder 210 . distal cap 273 is rigidly attached to cylinder 210 . members 210 , 220 , 250 , 260 , 271 , and 273 have a center passage or void 279 for receiving a catheter with electrode where the void runs from a centerpiece net 275 to a centerpiece outlet 277 . in other words , electric potential mapping and electrode attachment device effective hollow at its longitudinal core with void 279 , which is required so that a laparoscopic device , such as an electrode with lead , can be inserted through electric potential mapping and electrode attachment device for mapping , electrode attachment , or other procedure . when electric potential mapping and electrode attachment device is properly connected to electromechanical connector 131 of the pacemaker , main cylinder 210 is electrically and mechanically connected to electromechanical connector 135 on lead 103 . electric potential mapping and electrode attachment device further comprises a main cylinder - to - electrode lead electromechanical connection means where there is a rigid connection and electrical connection between main cylinder 210 and lead connector 135 . said means may be by threaded member , clamping member , collet member , chuck member , magnetic member , electromagnetic member , or similar . electrical connection of said means may be accessed at an electrical connector 290 on the electric potential mapping and electrode attachment device . electrical connector 290 provides the ability to connect an electrical connector such as an alligator clip or similar to it while also allowing the ability to move it and slightly rotate wire lead 103 , which would be adjusted in the usual way to manipulate electrode 121 . electrical connector 290 provides continuous viable electrical connection to helix 122 during rotation of helix or electrode or both . when electric potential mapping and electrode attachment device is properly connected to electromechanical connector 131 of the pacemaker , secondary cylinder 250 is electrically and mechanically connected to electromechanical connector 133 on lead 103 . electric potential mapping and electrode attachment device further comprises : a secondary cylinder to - electrode lead mechanical connection means and a secondary cylinder to - electrode lead resilient electrical connection , where there is a rigid mechanical connection and electrical connection between main cylinder 210 and lead connector 135 . this electrical connection may be accessed at an electrical connector 230 on the electric potential mapping and electrode attachment device . electrical connector 230 provides the ability to connect an electrical connector such as an alligator clip or similar to it in order to provide a continuous viable electrical connection to said electrode collar 127 , while also allowing complete rotation of electromechanical connector 133 relative to electromechanical connector 135 , as is required to attach electrode 121 to heart tissue by screw means with some electrode designs , where rotation is accomplished by rotating said secondary cylinder 250 . secondary cylinder to - electrode lead mechanical connection means may be by threaded member , clamping member , collet member , chuck member , magnetic member , electromagnetic member , or similar . in best mode , secondary cylinder - to - electrode lead mechanical connection means comprises : a collet or chuck 220 and a collet striker 260 . in best mode , secondary cylinder to - electrode lead resilient electrical connection comprises a resilient electrical connector 240 . this structure is used to attach electric potential mapping and electrode attachment device to electromechanical connector 133 . collet 220 is a chuck collar with at least two fingers 221 on its distal end . collet 220 forms a collar around the electromechanical connector 133 and exerts a strong clamping force on 133 when it is tightened on such via the collet striker 260 . when collet 220 is at rest , spring 280 forces collet 220 in the proximal direction into a taper section of collet striker 260 . collet striker 260 is a specifically sized and shaped bushing with distally facing taper on its interior diameter that acts to force collet fingers 221 closed . clamping force is provided by spring 280 . spring 280 is compressed between spring retainer ring 283 and proximal end button 271 . ring 283 is fixed while button 271 slides to compress and release spring 280 . collet fingers 221 are released when end button 271 is depressed , which forces collet 220 in the distal direction , where the taper on the collet striker widens , allowing collet fingers 221 to open , thereby releasing electromechanical connector 133 . fingers 221 naturally spring open from its shape . collet 220 is typically made of metal . collet 220 may have 2 - 20 fingers 221 . collet 220 further comprises a nut area 223 in rigid connection with the collet fingers 221 . collet is oblong shaped with fingers 221 on one end and nut area 223 on the other end or alternately in the middle of oblong shape . nut area 223 engages counterpart structure or wrench structure on the secondary cylinder 250 . nut area 223 fits within inverse matching structure on the interior surface of secondary cylinder 250 , so that cylinder 250 acts as a wrench to rotate said nut area 223 on collet , thereby rotating collet fingers 221 , thereby rotating electromechanical connector 133 on the pacemaker , when the electric potential mapping and electrode attachment device is properly connected to such . thus , connector 133 may be rotated by rotating secondary cylinder 250 when electric potential mapping and electrode attachment device is properly attached to 133 . continuous viable electrical connection may be maintained by a resilient electrical connector 240 . resilient electrical connector 240 is resilient in that it may be compressed and it will naturally spring back into shape from the compression . resilient electrical connector 240 is also capable of maintaining electrical conductivity during said compression and response . resilient electrical connector 240 electrically connects collet striker 260 to said electrical connector 230 and is physically placed between such . resilient electrical connector 240 keeps a preload positive force or pressure on collet striker 260 , directed radially inward towards the center axis of the electric potential mapping and electrode attachment device , thereby allowing 240 to remain in contact with 200 as 200 is repeatedly and completely rotated . thus , 240 must be compressed before installing between collet 220 and connector 230 . collet striker 260 maintains electrical connection with collet 220 because collet striker 260 is clamping down on collet 220 . likewise , collet 220 maintains electrical connection with electromechanical connector 133 because collet 220 is clamping down on connector 133 . in best mode , resilient electrical connector 240 is a ball detent . ball detent comprises a metal sphere , sliding within a hollow metal cylinder with one open end and one closed end , against the pressure of a spring . when elements are sized appropriately , conductivity between the ball and cylinder is maintained during sliding . the spring pushes the ball against the open end of the cylinder where it is retained within such because the open end of cylinder has a smaller diameter than the ball . when elements are sized appropriately , the ball rides along the surface of the collet , where 240 maintains electrical conductivity with such despite imperfections in outer circumference of collet , alignment of members , wear of members , or other imperfection . the flexible conductive range of this device would depend on the diameter of the ball , where larger diameters yield large flexible conductive ranges . electrical connector 290 is electrically and mechanically connected to electromechanical connector 135 . in best mode connector 290 has a threaded base 291 so that one end is a threaded member 291 , which can be mated with a taped hole in main cylinder 210 . after the electric potential mapping and electrode attachment device is properly connected to electromechanical connector 135 on the pacemaker , connector 290 may then be connected by simply screwing in down onto 135 . after electrode attachment , a programming module or the control module may be electrically connected to connectors 290 and 230 along with the previously mentioned electrical connectors used for mapping , thereby allowing continuous electrical readings throughout programming and other control process .
electric potential mapping and electrode attachment device comprises : a main cylinder ; a center piece having a center piece passage for receiving a catheter with its electrode running from a center piece inlet to a center piece outlet ; a cylinder cap for closing main cylinder ; a second cylinder which fits inside of and is allowed to rotate relative to main cylinder ; and at least two electromechanical connections used to connect the device to at least two connectors on an electrode lead , where the device facilitates rotation and movement of each connector relative to the other while retaining continuous electrical connection with heart tissue or other body cavity tissue . the device offers continuous recording of electrical potentials of a body cavity during full rotation of collars , connectors , or connections on a laparoscopic device during his bundle mapping , pacemaker electrode anchoring , post anchoring , pacemaker programming , and afterwards .