the cardiac ablation applicator in accordance with the invention is shown in an exemplary embodiment in fig1 and 2 . the applicator is shown in an unfurled state in fig1 and in a furled state in fig2 . a catheter 10 which can be of known construction and material , has disposed therein a coaxial cable 12 having an outer conductor 14 and an inner conductor 16 and an electrically insulating material 18 therebetween . the inner conductor 16 extends outwardly from the cable end and is formed into a spiral 20 to provide a spiral antenna . the spiral antenna is disposed within a balloon 22 which is sealed to the catheter wall near an end thereof as shown by seal 24 . an inflation tube 26 is disposed within the catheter 10 and has one end terminating at or near the catheter end . the spiral antenna can have one or more wire turns and can be of a wire size to suit specific embodiments . in the illustrated embodiment the spiral antenna is formed from the inner conductor of the coaxial cable . in an alternative embodiment , the spiral antenna can be formed from a separate wire which is joined to the inner conductor of the coaxial line . a low loss , low dielectric constant fluid is introduced in the balloon 22 by way of the inflation tube 26 from a suitable source ( not shown ). a physiologically benign fluid is employed which usually is air , nitrogen or a perfluorocarbon blood substitute . the fluid is also drained from the balloon via the tube 26 . to furl the antenna , the coaxial cable 12 is retracted within catheter 10 , which draws the antenna coil into the catheter and which causes a general straightening of the antenna wire within the catheter as illustrated in fig2 . the balloon 22 is deflated either before , during or after retraction of the antenna into the catheter . the catheter has the antenna retracted therein , when the catheter is inserted and snaked within a blood vessel to intended position within a heart or other organ or tissue . when in position , the balloon 22 is inflated by introduction of fluid through inflation tube 26 , and coaxial cable 12 is pushed through the catheter to a position near the end of the catheter for positioning the antenna 20 outward of the catheter and within the balloon . the antenna wire unfurls when released from the catheter by the inherent springiness of the antenna wire . the antenna can be implemented in a variety of ways . for example the antenna can be implemented by a metallized path provided on the inside wall of the balloon . the inner end of the metallized spiral terminates in a metallized path which extends to near the balloon opening . the metallized path is joined to the center conductor of the coaxial cable by an extension of the center conductor or by a separate interconnecting wire . in this embodiment the antenna is furled and unfurled during inflation and deflation of the balloon , since the metallized antenna path is sufficiently flexible to accommodate the inflated and deflated balloon configuration . there is no need for longitudinal movement of the coaxial cable within the catheter for this embodiment . to validate the new wide aperture spiral antenna applicator and define the characteristics of microwave heating in myocardial tissue , phantom tests and in vitro tissue tests as well as in vivo living animal tests were performed . the wide aperture spiral antenna was built from a coaxial rg58 cable , chosen for its high power - handling capability and ease of fabrication of the spiral antenna . the cable jacket and outer conductor were stripped to a length of 4 . 5 cm , the teflon insulator was thinned , and then this active segment was curved into a spiral with about 1¼ turns . the coaxial cable was sealed at the antenna end . a tube capable of inflating a balloon was introduced parallel to the coaxial line . the spiral and inflation tube were surrounded by the balloon and firmly sealed with silicon rubber adhesive . the total catheter length was chosen to be 100 cm . before the antenna was used for ablation , it was tested in an artificial blood / heart tissue phantom for microwave frequencies in the range from 600 to 1600 mhz using an hp8510b microwave network analyzer , as shown schematically in fig3 . the phantom consisted of a saline / sucrose solution , mixed in proportions based on standard recipes , to model the electromagnetic characteristics of high - water - content tissue — such as cardiac tissue and blood — at 915 mhz : the dielectric constant is 51 and electrical conductivity is 1 . 3 s / m [ 18 ]. measurements of s 11 were performed with the antenna embedded in the phantom . the antenna penetrated at least 10 cm below the surface to eliminate the dielectric mismatch at the air - phantom interface . the antenna input was mated to the source port of the network analyzer . [ 0034 ] fig4 shows the magnitude of the measured reflection coefficient s 11 of the antenna embedded in the phantom . at the operating frequency of 915 mhz , the value of s 11 is − 21 db , corresponding to 99 . 2 % of the power supplied being radiated from the antenna and absorbed in the feeding cable . it can thus be concluded that the antenna couples quite well to the biological tissue at 915 mhz . the transmitted power deposition was measured by using a simple short monopole probe antenna , which sampled the electric field at various locations within the phantom . an inherent and unavoidable limitation of this procedure is that the probe antenna must be very small for high - resolution measurements , but in being so small , it couples poorly to the phantom . field readings are therefore relative rather than absolute . the probe antenna consists of a coaxial cable with 10 mm of the center conductor extend beyond the outer conductor . the probe can only measure the phase and amplitude of a single polarization of electric field . the squared magnitudes of all electric field polarizations are summed to determine the relative deposited power pattern in the phantom . the spiral antenna and the monopole probe are modeled as a two - port network . according to two - port network theory , the s 11 measurement gives the ratio of power reflected back from the antenna , and hence gives a measure of how well the antenna radiates power into phantom , while s 21 represents how well the power is transferred between the two antennas through the phantom . [ 0036 ] fig5 shows the measured power deposition contours given by squaring the magnitude of s 21 . for the purpose of these plots , the spiral is oriented so that its axis is facing the forward z - direction . fig4 a gives the pattern in the x - z plane , which includes the feed line , along the negative x - direction , while fig4 b shows the pattern in the y - z plane . the above results can be compared with those obtained with a simple monopole antenna . a monopole with geometry shown in fig6 was fabricated and measured in the same phantom set - up used previously . the center conductor , pointing in the z - direction parallel and against the tissue surface , extends 2 . 5 cm , or slightly more than one - half a tissue wavelength . contour plots for the resulting deposited power patterns are shown in fig7 . fig7 a shows the power pattern in the z - p plane , where the ρ - direction points into the tissue , while fig7 b gives the circularly symmetric pattern in the plane perpendicular to the monopole axis . comparing the measured relative power deposition patterns for the spiral and monopole antennas indicates the superiority of the former in heating tissue to form well - defined lesions . whereas the monopole pattern is widely spread along the 2 . 5 cm wire length , it is narrow in the transverse direction perpendicular to the wire , and power levels fall off relatively quickly with depth into the tissue . the spiral antenna on the other hand deposits more of its power in a ring near the axis of the spiral . the spiral antenna pattern is more nearly circular and is deeper than the monopole antenna pattern . microwave heating was tested using a custom - built 915 mhz power source with output power level adjustable from 0 . 001 to 330 w and duration adjustable from 1 s to 99 min . forward and backward power were monitored throughout the course of each heating experiment . the transmitted power was set to three levels for testing ( 50 , 100 and 150 w ), the duration was set at 60 s , and the forward and reflected power was recorded in each case . the antenna was positioned with the antenna axis perpendicular to the surface of the medium to be heated . a fulcrum balanced the weight of the antenna and assured constant tissue pressure . all in vitro and in vivo temperature measurements were performed with the fluoroptic thermometry system ( luxtron model 790 , measurement range , − 200 ° c . to 450 ° c ., accuracy ± 0 . 1 c luxtron corp ., santa clara , calif .). prior to each application , 10 cc . of room air inflated the balloon surrounding the spiral antenna . during each application of microwave energy , the power and fluoroptic temperature probes were continuously monitored and recorded . in vitro experiments were performed in isolated perfused pig hearts to identify how much power would be required to effectively ablate myocardial tissue and to demonstrate the time course of temperature rise and lesion growth during microwave ablation . fig8 shows the experimental setup . we first created a perfused pig heart model to simulate the endocardial environment present in catherter ablation . the model consisted of a sectioned pig heart block suspended in a 4 × 3 × 2 cm polystyrene chamber . a perfusion pump controlled flow across the surface of each heart block , simulating blood flow . four temperature sensors were inserted into each heart block , spaced 5 mm apart . for each experiment , power was applied for 60 s . the temperature was recorded , the phantom was allowed to cool down for several minutes , the four sensors were withdrawn , the pig heart block was replaced , and the thermometry system was recalibrated . once the phantom returned to equilibrium within 0 . 5 ° c . of initial conditions , the procedure was repeated . ten tests were performed under identical treatment conditions in order to average unavoidable uncertainties in output power , heating time , and position of the antenna , catheters and temperature probes . these were repeated for each power level . once a lesion is created , the effects are clearly irreversible , and new unexposed tissue must be used for testing . however , in contrast to rf ablation , there was no noticeable input impedance change , nor heating rate change due to the lesion formation . reflected power remained nominal in all experiments for all power levels . this is partly due to lack of tissue charring , but also due to the microwave power deposition mechanism which relies less on material conduction characteristics . [ 0043 ] 3 . in vivo tests on pig thigh muscle model the experimental setup for the in vitro tests on thigh muscle model is shown in fig9 . the methods employed for the wide - aperture spiral antenna were the same used by nakagawa [ 19 ] for rf electrode tests . these studies were performed to accumulate sufficiently detailed temperature data and compare the tissue temperature at the border zone of irreversible tissue injury . five male pigs weighing between 77 - 89 kg were studied . as tested in the in vitro experiment , the catheter was positioned perpendicular to the thigh muscle , and held in contact with exposed thigh muscle at a constant pressure by use of a custom balance . tissue temperature was measured with the same fluoroptic probes as used in in vitro tests . two thermal sensor probes were bundled together with shrink tubing . one sensor tip extended 3 mm from the end of the shrink tubing , and the other sensor tip extended 6 . 0 mm . the sensor probes were inserted into the muscle ( 3 . 0 mm and 6 . 0 mm from the surface ) directly adjacent to the edge of the balloon - tissue interface . an additional fluoroptic temperature probe was attached at the center of the balloon - tissue interface . eight to ten applications of microwave energy were delivered to separate sites on the right thigh muscle . the skin incision was closed , the pig was turned onto its right side , and eight to ten applications of microwave energy were delivered at separate sites on the left thigh muscle . microwave energy was delivered for 60 seconds using one of three power levels : 50 , 100 , or 150 w . finally , the lesion was stained and its volume was determined by assuming it had an oblate spheroidal shape with volume given by : where a and c represent the surface radius and the depth of the volume , respectively . for 60 s heating with 50 , 100 and 150 w input power , the resulting lesion volumes were 260 , 1200 , and 1900 mm 3 , respectively . the relationship is not linear because the power deposition dependence is not linear with volume , and lesions created by heating tissue to above a particular temperature threshold do not have uniform temperature throughout . in the thigh muscle model , a total of 70 lesions were produced , 23 lesions created at 50 w , 24 lesions at 100 w , and 24 lesion at 150 w . for each power level , the peak and average temperatures declined as a function of depth . at 100 w the peak temperature declined from 78 . 7 ° c . at the antenna - tissue interface to 73 . 4 ° c . and 55 . 5 ° c . at 3 mm and 6 mm deep , respectively . the average temperature versus time course for the lesion formation is shown in fig1 . the peak and average temperatures increased significantly as a function of power . the peak surface temperature recorded at the antenna - tissue interface is as low as 65 . 9 ° c . at 50 w and rises to 106 . 6 ° c . at 150 w . similar relationships were observed for the average temperatures . fig1 shows the lesions produced with various applied powers . note that although the measured power patterns of fig5 indicate a ring of higher power , thermal conduction fills in the center of the ring to produce a convex lesion volume . the experiment preparation for this animal test is similar to that described in the in vivo thigh muscle model experiment . eight anesthetized pigs weighing between 70 - 90 kg were studied . a surface electrocardiogram was continually recorded during each procedure . a sternotomy incision was made to expose the heart and the pericardium was incised . the catheter was inserted into the left ventricle , and guided by fluoroscope , brought into contact with the left ventricle apex . microwave power of 150 w was then delivered to the tissue for 60 s . after completion of the experiments , the pigs were sacrificed , the hearts were removed and the lesions were examined . in the open - chest experiment , a total of 10 lesions were produced , with one or two lesions created on each pig . fig1 and 13 show the representative lesions . note that the lesions are transmural and that there is no charring of surface tissue . the depths produced are sufficient for the ablation of an arrythomogenic focus that is very deep within the ventricular myocardium or is subepicardial . the wide aperture spiral antenna effectively delivers power in cardiac tissue to create a wide , deep oblate spheroidal lesion . the spiral antenna is well matched to biological tissue at the operating frequency of 915 mhz . the antenna design is demonstrated in phantom tissue by measuring the reflection parameter s 11 and transmission parameter s 21 , and by comparing them to a monopole antenna . full power heating experiments at 50 , 100 , and 150 w on in vitro and in vivo animal tissue indicate that the field patterns in phantom combined with thermal conduction effects generate the desired lesion shape . these results show that the wide - aperture microwave spiral antenna design is capable of creating lesions of significant depth that may be applicable for the ablative therapy of ventricular tachycardia . alternative implementations may occur to those skilled in the field without departing from the spirit and true scope of the present invention . for example , the catheter may include steering mechanisms , and fluid cooling of the coaxial feed cable may be employed , as can mapping electrodes . accordingly , the invention is not limited by what has been particularly shown and described . [ 1 ] jackman , w ., wang , x ., friday , k ., et al ., “ catheter ablation of atrioventricular junction using radiofrequency current in 17 patients . comparison of standard and large - tip electrode catheters ,” circulation , 83 : 1562 , 1991 . [ 2 ] langberg , j ., chin m , schcmp , d ., et . al ., “ ablation of atrioventricaular junction using radiofrequency energy using a new electrode catheter ,” am j cardiol , 67 , pp . 142 , 1991 . [ 3 ] rosen , a . and rosen , h . ( eds . ), new frontiers in medical device technology , wiley , new york , 1995 . [ 4 ] morady , f ., harvey , m ., kalbfleisch , s ., et . al ., “ radiofrequency catheter ablation of ventricular tachycardia in patients with coronary artery disease ,” circulation , 87 , pp . 363 - 372 , 1993 . [ 5 ] kim , y ., sosa - suarez , g ., trouton , t ., et . al ., “ treatment of ventricular tachycardia by transcatheter radiofrequency ablation in patients with ischemic heart disease ,” circulation , 89 , pp . 1094 - 1102 , 1994 . [ 6 ] langberg , j ., wonnell , t ., chin , m ., et . al ., “ catheter ablation or the atrioventricular junction using a helical microwave antenna : a novel means of coupling energy to the endocardium ,” pace , 14 , pp . 2105 - 2133 , 1991 . [ 7 ] nevels , r ., dickey , g ., arndt , f ., raffoul , g ., carl , j ., and pacifico , a ., “ microwave catheter design ,” ieee trans . biomedical engineering , 45 , no . 7 , july 1998 . [ 8 ] l . n . horowitz , a . h . harken , j . a . kastar , and m . e . josephson , “ ventricular resection guided by epicardial and endocardial mapping for treatment of recurrent ventricular tachycardia ,” n . eng . j . med ., 302 , p . 590 , 1980 . [ 9 ] c . h . durney , “ electromagnetic field propagation and interaction with tissues ,” in an introduction to the practical aspects of clinical hyperthermia , s . b . field and j . w . hand , ( eds )., taylor and francis , new york , chap . 10 , 1990 . [ 10 ] james g . whayne , sunil nath , david e . haines , “ microwave catheter ablation of myocardium in vitro , assessment of the characteristics of tissue heating and injury ,” circulation 89 , no . 5 , pp . 2390 - 2395 , may 1994 . [ 11 ] brian a . vanderbrink , zeji gu , victor rodriguez , mark s . link , munther k . homoud , n . a . mark estes iii , carey m . rappaport , paul j . wang , “ microwave ablation using a wide - aperture antenna design in a porcine thigh muscle preparation : in vivo assessment of temperature profile and geometry ,” j . cardiovascular electrophysiology , february 2000 . 9 [ 12 ] l . s . taylor , “ electromagnetic syringe ,” ieee trans . biomed . eng ., bme - 25 , pp . 303 - 304 , may 1978 . [ 13 ] j . w . strohbehn , e . w . bowers , j . e . walsh , and e . b . douple , “ an invasive antenna for locally induced hyperthermia for cancer therapy ,” j . microwave power , 14 , pp . 339 - 350 , 1979 . [ 14 ] t . satoh and p . r . stauffer , “ implantable helical coil microwave antenna for interstitial hyperthermia ,” int . i . hyperthermia , 4 , no . 5 , pp . 497 - 512 , 1988 . [ 15 ] tracy wonnell , paul stauffer , and jonathan langberg , “ evaluation of microwave and radio frequency catheter ablation in a myocardium - equivalent phantom model ,” ieee trans . biomed . eng ., 39 , no . 10 , pp . 1086 - 1095 , october 1992 . [ 16 ] carey m rappaport , zeji gu , paul j wang , “ wide - aperture microwave catheter - based cardiac ablation ,” progress in electromagnetics research symposium digest , hong kong , p 316 , january 1997 . [ 17 ] zeji gu , carey m . rappaport , paul j , wang , and brian a . vanderbrink , “ a 2¼ turn spiral antenna for catheter cardiac ablation ,” ieee trans . biomed . eng ., 46 , no . 12 , pp . 1480 - 1482 , december 1999 . [ 18 ] guy , a . w ., “ analysis of electromagnetic fields induced in biomedical tissues by thermographic studies on equivalent phantom models ,” ieee trans . microwave theory and tech ., 19 , no . 2 , pp . 205 - 214 , february 1971 . [ 19 ] nakagawa h , yamanashi ws , et al , “ comparison of in vivo tissue temperature profile and lesion geometry for radiofrequency ablation with a saline - irrigated electrode versus temperature control in a canine thigh muscle preparation ,” circulation , 91 , pp . 2264 - 2273 , 1995 . 10
a catheter - based microwave antenna cardiac ablation applicator has been developed which unlike previously - developed ablation catheters , forms a wide aperture that produces a large heating pattern . the antenna comprises a spiral antenna connected to the center conductor of a coaxial line , and which is insulated from blood and tissue by a non - conductive fluid filled balloon . the antenna can be furled inside a catheter for transluminal guiding . once in place at the cardiac target , the balloon is inflated , and the coiled spiral antenna is ejected into the inflated balloon . the wide aperture antenna generates a ring - shaped power pattern . the heat generated from this deposited power is conducted through a volume larger than the spiral diameter , ablating diseased tissue . the resultant lesion profile is both wider and deeper than that of either conventionally - used rf catheter - based ablation electrodes , or that of other microwave applicators , and provides greater heating accuracy and controllability .