referring to the drawings in particular , the present invention creates an eit device and process with the following design features , corresponding to fig2 and fig3 . the voltages ( 7 ) of all m measuring channels are analyzed continuously for possible interferences . this can be done by comparing α = 1 . . . n properties e m α ( u m ) ( 11 ), the measured voltages of all m = 1 . . . m channels relative to expected desired values and / or desired ranges e soll m α ( 15 ). the desired range can be derived , for example , from device parameters and / or from theoretical considerations such as reciprocity and / or by analyzing a large data set of eit measurements from hospitals and / or laboratory tests with specific interferences . examples of measurable and derivable properties of measured voltages are , of course , the snr determined , on the one hand , but also freak values of voltages or changes of these voltages , mean value , phase and real and imaginary part of the voltage , determined crosstalk estimates of measured voltages , determined common mode estimates of voltages , reciprocity of measured voltages , voltage drifts , fluctuations in current feed , correlations between reciprocal voltage time series over a certain period , correlations between voltage and ventilation data is present , the amount and phase of the electrode skin contact impedances , characteristic properties of the value and phase distributions of the frequency spectra of the measurement series of individual channels over a certain period , e . g ., maximum , standard deviation , shape of the spectral distributions , etc . it should be noted that the n properties of the measurements do not necessarily have to be independent from each other . systematic interferences , e . g ., common mode interferences or crosstalk , may be reflected in several properties , but they do not have to be . 2 . an adaptation method is to determine the systematic error components of the m voltages from the n properties and to directly correct the voltages . 3 . another adaptation method is the channel - specific adaptation of the reconstruction rule . channel - specific and property - specific parameters q m α ( e m α , e soll m α ) ( 13 ), which can be combined into a channel - specific parameter q m ( q 1 1 , . . . , q m α , . . . , q m n ) ( 17 ), for example , by arithmetic or geometric means or variables derived therefrom or even combinations of different channels , are determined from the comparison of the α = 1 . . . n determined properties e m α of each channel m = 1 . . . m with desired values e soll m α derived empirically and / or theoretically beforehand . the channel - specific quality parameters are used by the reconstruction rule for the algorithmic suppression of the error - containing voltage of channel m ( 21 ). without limitation of the general nature , the channel - specific parameters can be imaged onto a closed value range of , for example , [ 0 , 1 ], where measurements containing great errors can be assigned to the value 0 and measurements containing hardly any errors or even intermediate values can be assigned to q = 1 . thus , channels with q m = 0 could be suppressed during the eit reconstruction ( 22 ) completely , channels with 0 & lt ; q m & lt ; 1 could be suppressed partly , and channels with q m = 1 could not be suppressed at all . however , threshold ranges for suppression strengths can be used just as well . the concrete form in which the suppression is performed depends on the concrete reconstruction rule . one method is , for example , weighting , as it is frequently used for statistical measuring errors . the consequence of this method is that any interferences in the voltage measurements , which would become noticeable as artifacts in the eit image without processing ( fig4 ), are markedly suppressed ( fig6 ). 4 . part of the process is the continuous , constant determination of the properties e m α , as well as of the property quality parameters q m α ( e m α , e soll m α ) derived therefrom and / or of combined channel - specific quality parameters q m ( q 1 1 , . . . q m α , . . . , q m n ), typically over a time window . if at least one parameter changes over a certain quality parameter range δq thr ( 20 ), the channel - specific quality parameters are used for adaptation of the reconstruction rule as described in point 2 ( 21 ). the threshold δq thr may be , depending on the concrete implementation of the process , between one value for each channel , which has proved to be advantageous , or a plurality of values for different channels , up to the definition of a threshold for each channel m and each property α . it is also possible to adapt the reconstruction rule for each frame . 5 . part of the process is the determination of β = 1 . . . g of global properties e β ( 12 ) from measurements and / or technical operating variables . examples of this are maxima or minima or mean values , variations or combinations of different methods of the channel - specific properties e m α and / or channel - specific quality parameters q m α and q m and / or operating current intensities , gain factors , dynamic range of the voltages , operating frequencies , norm of differential images with and without adaptation of the reconstruction rule , etc . the global properties e β are compared analogously to the channel - specific variables with expected desired values and / or desired ranges e soll β ( 16 ) and global quality parameters q β ( e β ) are determined ( 14 ). the desired range for the global properties can be derived here as well , for example , from device parameters and / or from theoretical considerations and / or by analysis of a large data set , of eit measurements from hospitals and / or laboratory tests with specific interferences . the global quality parameters q β ( e β ) are combined into a global quality index q ( q 1 , . . . , q g ) ( 19 ), for example , by arithmetic or geometric means . the global quality index can be imaged , without limitation of the general nature , to a closed interval , for example , [ 0 , 1 ]. the global quality index is used for the quality classification of the eit measurement . the quality classification may be carried out continuously or also discretely by classification into l classes , for example , good , medium , poor ( l = 3 ) or 1 , 2 , 3 , 4 , 5 ( l = 5 ), and the division of the q interval does not have to be , in general , equidistant and is typically substantiated empirically . 6 . the global quality index is displayed for the user on the display in a graphic form , e . g ., by bar display ( 24 ), filling level display ( 28 ) or pie chart display ( 29 ) and / or in a color - coded manner , e . g ., by traffic light type display ( 25 ), color light display ( 26 ), color - coded graphs ( 27 ) and / or in an alphanumeric manner , e . g ., in percentage ( 30 ) or in any interval including improper limits a , b and / or combinations of alphanumeric and graphic display ( 28 ). 7 . episodes of poor quality ( which cannot be clinically interpreted ) and episodes of good quality ( which can be clinically interpreted ) are marked differently ( 34 ) in a trend graph of eit variables over longer time periods ( 33 ) or curves over shorter time periods ( 32 ), for example , the summed - up impedance changes ( global curve ) or impedance changes summed up over rois ( regions of interest ) or time series derived from eit data and / or from voltage data and / or from operating data . 8 . episodes of eit images , curves , trend values with poor data quality do not participate in the scaling of the display ( 35 ), ( 36 ). 9 . eit images , curve values , trend values of episodes with poor data quality can be set at a defined value , which may be located both within the value range of their graph and outside the value range of their graph ( 37 ). 10 . eit images , curve values , trend values of episodes with poor data quality can be interpolated , and at least one eit image or value of the curve or trend before the episode with poor data quality is used for the interpolation ( 38 ). the interpolation is likewise marked separately . 11 . in case of poor data quality , a cause of error analysis ( root cause analysis , rca , in english ) is performed , and possible causes of a case of error are displayed for the user . furthermore , possibilities for eliminating the causes of error are displayed , which may lead to an improvement of the quality of measurement after an influence is exerted by the user , which makes possible a renewed interpretability of the eit data . the present invention will be explained in more detail below on the basis of an exemplary embodiment in connection with the figures . fig3 schematically shows some examples of display elements of eit data with quality information . a bar graph ( 24 ) of any orientation can be represented in one color or in a color - coded manner , where a solid bar represents good quality and an empty bar poor quality or also the other way around , depending on the definition of solid and empty . the color of the bar may also change in certain cases in which a threshold value is exceeded ( e . g ., for good = green , medium = yellow , poor = red ). a traffic light type display ( 25 ) ( poor quality = red , medium quality = yellow , good quality = green ) would be possible as well . alternative codes and / or numbers of traffic lights are , of course , conceivable as well . another display variant is , for example , a filling level display ( 28 ) or pie chart graph ( 29 ), optionally in one color or in a color - coded manner . alphanumeric values could be present , e . g ., directly in the eit images ( 30 ) or images derived from eit data or in / in case of graphic representations , e . g ., within a filling level display ( 28 ). trend graphs shall be defined here as graphs of eit - based data over longer periods of time , e . g ., several minutes to hours ( 33 ) under curves of rather rapidly varying variables in the range of seconds or a few minutes ( 32 ). episodes whose data quality is so poor that the results of the reconstruction contain or can contain great errors despite adaptation processes are marked separately ( 34 ). the reconstructed values of , e . g ., impedance curves ( 32 ) of the section with poor data quality are not used for scaling the display ( 35 ). this also applies to the trend graph ( 33 ) of the , e . g ., end - inspiratory impedance changes ( tidal strokes ). the values of the tidal strokes ( 36 ), which probably contain great errors , in the marked episode with poor data quality ( 34 ) likewise do not participate in the scaling . it is also possible to set the values at a fixed value ( 37 ) or to interpolate them ( 34 ) in the trend graphs ( 33 ) of eit - based variables in the episodes with poor data quality ( 34 ). the following example offers a more detailed explanation of the process for the determination of the channel - specific quality indices and of the global quality parameters as well as adaptation of the reconstruction rule for this example . variations of this example generate similar results . the following channel - specific properties were used here for a 16 - electrode eit system with 208 measuring channels in the adjacent daq mode : e m 1 real part re ( u m ), m = 1 . . . 208 , of the voltages u m averaged over a fixed time window . the time window equals about 30 sec and contains a plurality of breaths , e m 2 phase φ m = a tan ( u m )/ re ( u m )), m = 1 . . . 208 , of the voltages u m averaged over a fixed time window , e m 3 value of the maximum voltage max ( u m ), m = 1 . . . 208 , of the voltages u m averaged over a fixed time window , e m 4 relative common mode errors δu m α cm =| δu m cm |/| u m |, m = 1 . . . 208 , of the voltages u m averaged over a fixed time window , wherein the common mode error δu m cm can be determined from special measurements as a function of the specific hardware embodiment and model hypotheses following therefrom , e m 5 reciprocity r m =|(| u m |−| u m ′ |)/(| u m |+| u m ′ |)|, m , m ′= 1 . . . 208 , of the respective voltages u m and u m ′ , wherein m , m ′ are mutually reciprocal channels . r m = m ′ applies . e m 6 signal - to - noise ratio snr m = 20 log 10 ( u m / n m ), m = 1 . . . 208 . the channel - specific noise component n m was estimated from the effective value of a high pass - filtered voltage time series with a limit frequency above the frequency range of the physiological signal sources ( ventilation , heart beat ). the property - specific quality index q m α was determined from the following calculation instruction . q m α ⁡ ( e m α , c m α , d m α ) = { 1 , 1 + c m α - e m α d m α · c m α & gt ; 1 1 + c m α - e m α d m α · c m α , 0 ≤ 1 + c m α - e m α d m α · c m α ≤ 1 0 , 1 + c m α - e m α d m α · c m α & lt ; 0 as a variation of this instruction , a hysteresis threshold may also be agreed upon , or other , more complex variants . this depends on the particular hardware and also on the experience with the behavior of clinical data . to determine the slice parameters c m α , d m α , more than 400 data sets from clinical studies were analyzed . distributions were determined for all properties e m α ; they are schematically shown in fig7 a )- 7 c ). it was specified that a slice c m α each is set at a value e m α , where 99 . 5 % of the data are below ( fig7 a , e . g ., reciprocity ) or above ( fig7 b , e . g ., real part of the voltage ) or in a range ( fig7 c , e . g ., phase ), depending on which range is considered to increasingly contain systematic error . for example , the reciprocity is mentioned for the distribution type in fig7 a ). it is ideally 0 ; excessively high values are indicative of systematic measuring errors , and the slice is therefore at the upper end . for example , the real part of the voltage shall be mentioned for distribution type 7 b . the greater this is , the more unlikely is a high component of systematic error , and therefore the slice is for excessively low values . the phase shall be mentioned for distribution type 7 c ). they are scattered around a mean value , which depends mainly on the current feed frequency used , the average electrode - skin contact impedance and the analog electronic unit . very large negative and positive phases , which are indicative of large systematic error components , may develop in case of great magnetic uncoupling or crosstalk . the decay value d m α sets the range of quality index between 0 and 1 and the hysteresis range and prevents a possible flutter , which would occur in case of properties e m α near the slice value c m α and jump function q m α ′ between 1 and 0 at the slice value . “ bootstrapping ” could also be used in alternative variant of setting the slice by setting slices for properties except one and then estimating the probability distribution for systematic errors of the properties left out on the basis of the data set and determining the slices for the variable left out by means of these . this variable is then added , another one is left out , and the process is repeated until all slices are set in this manner . an iterative procedure is also possible for this . regardless of the determination of the slices from a large set of clinical data , the concrete values of the slice parameters c m α , d m α depend on how the concrete hardware of an eit system responds to the measuring conditions in routine clinical practice . the channel - specific quality index q m was determined here from the product of the property - and channel - specific quality indices q m α : this is a conservative hypothesis , because even one property containing great errors may lead to great interferences in the eit image and reduce possible correlations between different properties . in addition , the channel - specific quality index of adjacent channels m ± 1 of an extremely greatly disturbed channel m may conservatively also be set to 0 . this was not used in this example . the channel - specific data quality analysis according to process steps 1 ) through 3 ) led to the combined channel - specific quality parameter q m shown in fig5 . it is seen that most channels have , indeed , measured nearly error - free ( q m values close to 1 ), three channels had moderate systematic errors , but three channels had massive systematic errors , so that the q m values of 0 were assigned to them according to the method explained in detail above . it shall be noted that no major statistical errors , i . e ., noise , were present in this measurement . the snr was in the expected range . the cause was a systematic common mode error during the voltage measurement in this case ! the channel - specific quality indices q m were used to define a systematic error weighting matrix : w = diag ( q 1 , q 2 , . . . , q 208 ). a sensitivity matrix - based newton - raphson method was used to reconstruct the vector δp n of the relative impedance change from the relative voltage change δu n . sensitivity matrix s was determined from a finite element model of the thorax using the linearized geselowits relation . the adaptation of the reconstruction matrix a to the concrete systematic errors found was performed by integrating the systematic error weighting matrix w in the determination of the reconstruction matrix : a = r ( s t ws + λl t l ) − 1 s t w , δp n ( t )= aδu n ( t ). matrix l designates the regularization matrix , the scalar variable λ designates the regularization parameter , and matrix r represents a filtered registration matrix from the fem system to the pixel system of the eit image . in fig4 , we find an eit image generated from the relative voltage changes between end - inspiratory and end - expiatory states of the lungs with a data acquisition time of about 30 sec in a hospital in the adjacent daq mode with an eit system with 16 electrodes ( 208 channels ) without the use of the above - described method of adaptation of the reconstruction rule , i . e ., the above reconstruction rule was used therefor with w = i , wherein i is the 208 - dimensional identity matrix here . this means as an assumption that all measuring channels are free from systematic errors , as , e . g ., in the frequently used back projection . the horizontal structure shown in fig4 can be primarily classified as a an artifact , which was caused by the three greatly interfered channels with q m = 0 . the large artifact has nothing to do with the actual impedance change distribution , which is considered an indicator of the regional ventilation . a clinical assessment of this image would lead to an excessively clear misinterpretation . in fig6 , we find the eit image , which is based on the same interfered measured data as the image in fig4 , with applied adaptation of the reconstruction rule according to the above formula , including the quality indices in fig5 , which were determined from the quality analysis , in the systematic error weighting matrix w . complete suppression of the values of the three q m = 0 channels cause the artifact in the eit image in fig6 to disappear . a normally ventilated left lung is found here ( thoracic ct cross section convention : on the right in the figure ), but also a hypoventilated right lung in the dorsal area , in agreement with clinical x - ray findings . by applying process step 4 ), which will be explained in more detail below for this exemplary embodiment , a global quality index of q = 0 . 5 was assigned , which corresponds to medium data quality , because even when the eit data were interpreted with the use of the adaptative reconstruction method , the user should be informed that measuring errors occurred . a root cause analysis revealed large common mode components at electrode pair 11 , 12 based on poor electrode - skin contact impedance of electrode 12 . a suggestion to check the contact of the electrode according to process step 8 can put things right in such cases . the systematic errors prove to be highly constant over time . a repeated adaptation is indicated above all if a quality index worsens over a large range , e . g ., δq m & gt ; 0 . 8 , because an already bad channel may generate a large artifact . conversely , markedly improved channels need to be adapted in case of a larger number of markedly improved channels ( greater than , e . g ., 6 ) only , because only the accuracy is improved somewhat statistically compared to the previous solution without these channels . the following global properties were used in this example for the determination of the global data quality index : e 2 : maximal values of the channel - specific property e m 6 e 3 : dynamic range ( ratio of maximum voltage to minimum voltage ) e 4 : technical operating state ( flag sent by the eit system 0 or 1 ). the distributions of the global data quality parameters were determined , except for the technical operating state flag , which was provided by the device , analogously to the case of the channel - specific properties , from over 400 clinical data sets , and the slice values and decay intensities e β =( c β , d β ) were determined analogously and empirically adapted in the second iteration by expert consideration of corresponding eit images . the determination of the property - specific global data quality parameters q β was performed with an analogous formula as for the channel - specific data quality parameters q m α : q β ⁡ ( e β , c β , d β ) = { 1 , 1 + c β - e β d β · c β & gt ; 1 1 + c β - e β d β · c β , 0 ≤ 1 + c β - e β d β · c β ≤ 1 0 , 1 + c β - e β d β · c β & lt ; 0 the following values were obtained : q 1 = 0 . 89 , q 2 = 1 , q 3 = 0 . 56 , q 4 = 1 . the global quality index q was likewise determined from the product of the property - specific global quality index q β : a value , of q ≈ 0 . 5 was obtained , which is shown in the top left part of fig6 . this is a middle range . the user now knows that the measurement can be interpreted based on the adaptation of the reconstruction rule , but also that channels with systematic errors are present . while specific embodiments of the invention have been shown and described in detail to illustrate the application of the principles of the invention , it will be understood that the invention may be embodied otherwise without departing from such principles . 7 measured value vector of the voltages u of all channels of one frame 9 data set from voltages and operating information from at least one frame 12 determination of the g global properties from the data set 15 data bank or list with desired values / desired ranges of the channel - specific properties 16 data bank or list with desired values / desired ranges of global properties 18 analysis of the behavior over time of the channel - specific quality indices 20 decision for adaptation of the reconstruction rule or further use of the current one on the basis of the changes in the channel - specific quality indices compared to desired values 21 adaptation of the reconstruction rule with inclusion of the channel - specific quality indices 25 traffic light type display of the global quality index q 26 only color - and / or brightness - coded display of the global quality index q 27 bar display with quasi - continuous color - and / or bright - coded display of the global quality index q 28 filling level display with alphanumeric value of the global quality index q 29 pie chart display of the global quality index q 31 short time period of a long time period in the trend graph of the tidal strokes 32 impedance curve of the short time period ( e . g ., about 1 minute ) with ventilation 33 trend graphs of different eit - derived variables over a longer time period ( e . g ., 10 minutes ) 35 faulty values of the impedance curve during the episode with poor data quality ( adaptation of the reconstruction was , e . g ., no longer possible because of too many errors ), which do not participate in the scaling of display 32 and are not therefore fully displayed 36 faulty values for tidal strokes during the episode with poor data quality , which do not participate in the scaling 37 setting of values of a trend graph at a preset value during an episode with poor data quality 38 interpolation of values of a trend graph with the last value of the trend graph before the episode with poor data quality 39 asymmetric probability distribution for a channel property e α , for example , reciprocity and common mode voltages ( see above ) with slice at excessively high values 40 asymmetric probability distribution for a channel property e α , for example , real parts of the 4 lowest voltages per patient ( see above ) with slice at excessively low values 41 symmetrical probability distribution for a channel property e α , for example , phases of the voltages with subtracted mean value or 2f distributions of modulation with slices at excessively high values
an eit device has a plurality of electrodes that can be arranged on a body in order to reconstruct the impedance distribution of the body with a reconstruction algorithm . the control unit of the eit device is set up by suitable programming to continuously determine at least one property each from the measured signals of all measuring channels and to correct measured signals of the measuring channels on the basis of the properties or to adapt the reconstruction algorithm on the basis of the properties .