turning now to the drawings and particularly fig4 there is illustrated an aspirator system generally designated 10 which is primarly used in surgery . the main innovation of the system 10 is an automatic valve bypass means s , see fig1 and 4 . mounted within the means s is a valve means generally designated vcc . a coupling 14 passing out of housing s &# 39 ; is connected to suction line 15 . the line 15 , see fig4 runs to a blood collection receptacle fc . there is also a line lga that extends from the receptacle fc to a conventional suction pump a powered by a motor not shown . the valve means vcc preferably includes an exterior cylindrical stationary portion 16 fabricated from metal or other appropriate material . the portion 16 includes a pair of threaded openings to receive an inlet coupling 17 and outlet coupling 18 . the coupling 18 is connected to a spring loaded suction responsive trigger means generally designated p by line 19 . also extending through the exterior portion 16 is an atmospheric bore 20 . the stationary portion 16 of valve vcc is fixed to a support post 21 which in turn is mounted on the bottom of housing s &# 39 ;. within a cylindrical bore 22 &# 39 ; of stationary portion 16 is an interior rotatable cylindrical valve section 22 which is preferably fabricated of leaded bronze or other appropriate material . the section 22 , see fig2 , 5 , 6 and 7 , has two annular slots 23 and 24 . each slot 23 and 24 preferably extends annularly around the section 22 less than one half the circumference . when the apparatus is in a suction mode , see fig2 the rotatable valve section 22 is in a rest position where the couplings 17 , 18 and connecting tubes are in communication with each other through groove or slot 23 . slot 24 when suction is being applied , see fig2 communicates with the atmospheric bore 20 . the rotatable valve section 22 also preferably is formed with a bore 25 to receive an end of a key pin 26 which projects from an elongated shaft 27 which horizontally passes out both sides of the valve vcc , see fig1 and 8 . the shaft 27 is mounted on bearings not seen , in the stationary valve portion 16 . thus , as the interior rotatable valve section 22 rotates the key pin 26 will also rotate and cause the shaft to rotate . in fig3 the interior rotatable valve section 22 is seen to be rotated 90 °, due to blockage of the apparatus to be explained , and the couplings 17 and 18 no longer communicate because the slots 23 and 24 have rotated with 17 communicating with slot 23 and suction is cut off . also , coupling 18 is now communicating with atmospheric bore 20 through slot 24 . the spring loaded suction responsive trigger means p includes a cylinder 29 preferable made of glass or other appropriate material surrounded by a cylindrical casing 30 of metal or other appropriate material . at one end there is a threaded cover 31 and slidably mounted within cylinder 29 is a piston or plunger 32 which is connected to piston rod 33 . the rod 33 is surrounded by a spring 34 and end 35 is attached to the rod . surrounding the spring 34 and a portion of rod 33 is a tubular spring housing 37 . at the end of housing 37 a lever arm 39 is pivoted at 38 and the arm is in turn secured to shaft 27 . in addition , there is provided a lever arm 40 which is pivotably connected to rod 33 at one end and to a relief valve va at the other . as the piston 32 and rod 33 move back and forth in the cylinder 29 the relief valve va will be opened and closed depending on pressure variances at the nozzle 42 and end 43 . the valve va is interposed in line 41 passing from coupling 14 to coupling 17 . another improvement in the present invention resides in the provision of an air pressure regulator generally designated rp . secured to the end of housing s &# 39 ; is an outer housing 44 &# 39 ; which includes a second tubular adjustment rod 44 . extending from rod 44 is an adjustment pin 45 projecting through slot 46 in rod 44 . extending from pin 45 is a tension spring 47 which terminates at a tension tab 48 fixed to shaft 27 . the adjustment of the tension on spring 47 will in turn counter balance the tension of trigger means spring 49 . the rod 45 through screw means 50 is turned by a graduated scale knob 51 . upon turning the knob 51 , the displacement occurs as described above which places upon spring 47 an increase or decrease in resistance to the negative suction pressure and controls the rotation of shaft 27 . as can be seen the spring 47 will in effect counterbalance the spring 34 whereby the suction moving through line 19 will not allow the piston 32 to move forward in the cylinder 39 . the way the piston can move to the fig3 position will be described . secured on the shaft 27 , see fig1 , 9 and 10 is a fluid bypass means generally designated o . the bypass means o includes a pair of upper pivot arms 52 and a pair of lower pivot arms 53 spaced downward of the arms 52 . the arms are pivotably mounted on a rectangular frame 62 which is secured to the bottom of housing s &# 39 ;. pivot pins 54 &# 39 ; retain the respective arms to the frame 62 . as best seen in fig1 and 13 a pair of tension springs 54 are connected to the upper and lower arms 52 and 53 to bias the same together , yet is flexible enough to allow a spreading of the same upon a camming operation to be next described . secured to the shaft 27 are a pair of cam brackets 55 and mounted within the brackets 55 are cam follows 56 . in addition , extending between each upper pair of arms 52 and each lower pair of arms 53 respectively is a pinch bar 57 axially aligned with the shaft 27 . the shaft 27 includes a pinch bushing 58 which is aligned beneath the above each bar 57 respectively . extending from couplings 59 &# 39 ; and 60 &# 39 ; mounted on an end of housing s &# 39 ; is suction line 59 and air discharge line 60 . the lines 59 and 60 extending through the frame 62 and line 59 passes between the lower pinch bar 57 and pinch bushing 58 while air discharge line 60 passes between the upper pinch bar 57 and pinch brushing 58 , see fig8 through 13 . at the housing s &# 39 ; suction line 61 connects with suction line coupling 59 &# 39 ; and air discharge line le moving from pump a connects to coupling 60 &# 39 ;. outwardly of the bypass means o the respective lines 59 and 60 are coupled to a y joint 64 and a single line 65 joins a coupling 66 to a single suction - discharge line 67 passing to nozzle 42 . in operation , the suction pump a is activated and the amount of suction can be varied depending on the type of operation being undertaken . with activation , the end 43 of nozzle 42 is inserted in the opening or wound and suction will pull blood through the nozzle 42 , line 67 , line 59 and line 61 to the receptacle fc . with suction passing into receptacle fc , suction is also created at the bifurcation 68 in line 15 through 14 , 41 to the valve vcc . with the continued suction in line 14 the valve vcc will remain in the position as seen in fig2 . the shaft 27 is positioned for communication between couplings 17 and 18 and the cam brackets being mounted on the shaft 27 will rotate and remain as seen in fig8 . that is the suction line 59 will remain open and the pinch bushing 58 will pinch closed air discharge line 60 against upper pinch bar 57 because the arms 53 are forced downward with space between the lower pinch bar 57 and pinch bushing 58 . when the line 60 is closed , air within the line may be purged to the atmosphere through a conventional bypass means ( not shown ) in suction pump a . as long as there are no obstructions such as large clots of bloods , tissue , etc . the device will remain and function as described above with no blood passing directly through the valve vcc but only through line 59 of the bypass means o . should an obstruction occur at the end 43 of nozzle 42 there will be instantaneously movement and reversal of suction to discharge air or oxygen through ejection air discharge line 60 , coupling 66 , line 67 to the nozzle end 43 to dislodge the foreign matter . when suction is interrupted and a negative pressure is presented through lines 15 and 61 , the relief valve va is closed and the pivot arm 40 rotates in turn moving piston 32 to the position shown in fig3 and the suction between couplings 17 and 18 is broken by the rotation of interior section 22 of valve vcc . in turn the shaft 27 will rotate 90 ° and at 45 both lines 59 and 60 , see fig9 are instantaneously closed . then as the shaft 27 continues rotation the cam followers 56 ride on the arms 53 causing the suction line 59 to close and open the ejection air discharge line 60 so that air or oxygen may conventionally move from pump a to the end 43 of nozzle 42 to eject the blockage . as the above takes place it can be seen that the valve vcc is completely free of binding or clogging and malfunction because the blood will move through the bypass means o and not the valve vcc or trigger means p . if upon one reversal of pressure the stoppage or blockage in the nozzle 42 is not cleared , the above described operation is repeated instantaneously so that the stoppage is subject to alternating suction and ejection pressure until it is dislodged or pulled through the nozzle to the receptacle fc . the invention and its attendant advantages will be understood from the foregoing description and it will be apparent that various changes may be made in the form , construction and arrangements of the parts without departing from the spirit and scope thereof or sacrificing its material advantages , the arrangements herein before described being merely by way of example . i do not wish to be restricted to the specific forms shown or uses mentioned , except as defined in the accompanying claims , wherein various portions have been separated for clarity of reading and not for emphasis .
a medical aspirator system which includes an automatic valve means , suction pump , waste receptacle and a bypass means to move such blood and other material from a wound or operation area without allowing the blood and other material entry into the automatic valve means . the bypass means utilizes camming operations to open and close air suction and air discharge lines by pinching action .