fig1 gives the exemplary design of a humidifier according to the here presented invention . in this design the gas first enters the humidifier , then the respiratory gas flow generator . the embodiment shows how the respiratory gas is taken from a reservoir 1 ( i . e . pressurized gas or wall outlet ). while passing the humidifier 2 the usually very dry pressurized gas is conducted to a respiratory gas flow generator 3 . within the respiratory gas flow generator 3 the gas is depressurized to the respiratory pressure level needed for ventilation or respiratory therapy which might be slightly above ambient air pressure . then the respiratory gas is conducted to the patient 5 , using active temperature controlled respiratory tubing 4 connected to the respiratory gas generator . in this embodiment the compressible volume of the breathing system leading from the respiratory gas flow generator 3 to patient 5 is very low . it is important that the respiratory gas conducted to the patient is of both constant and preset humidity and temperature . the preset humidity at 37 ° c . typically is near saturation ( i . e . 100 %). it is also important that the patient can be ventilated without interruption . this is ensured by conducting heated fluid ( in the given exemplary design water of 72 ° c .) via line 6 to the humidifier 2 and from there into the annulus 7 . here water means any fluid which might be chosen by a man skilled in the art according to requirements . in this embodiment there might be added drugs or other substances to the fluid without special notice in the following paragraphs . the heated gas leaves the annulus 7 via a sieve bottom 8 into a sprinkling type chamber 9 filled with filling material 10 . in this exemplary design this filling is made of solid structured elements inheriting a huge surface and large cut - outs . while water is trickling from the annulus 7 via the huge surface downwards into the bottom chamber 11 of the humidifier 2 it heats and wets the filling material i 0 . at the same time respiratory gas from reservoir 1 is flowing in the opposite direction through the filling 10 . by doing so the respiratory gas is heated and humidified . in the end the nearly saturated gas is conducted via the collecting chamber 12 to the respiratory gas flow generator 3 . here it should be pointed out that the flow of heated water is substantially higher than the water flow needed to humidify the respiratory gas to saturation . this is to guarantee that there is no significant cooling of the fluid in the contact area of gas and fluid , and that the gas flow is heated as required . with the here illustrated design the above mentioned process is working under pressure . water and respiratory gas are pressurized to 4 . 5 bar . the temperature is about 72 ° c . depressurization of the respiratory gas from 4 . 5 bar / 72 ° c . to nearly environmental pressure / 37 ° c . ensures that the depressurized respiratory gas has a relative humidity of nearly 100 %. the humidifier will also work with other temperature / pressure combinations as long as it is guaranteed that even after depressurization to the respiratory pressure level at the target temperature the respiratory gas will be humidified according to the target humidity . as explained above humidifier 2 collects the fluid which has trickled through the filling material in the bottom chamber 11 . controlled by valve 13 it is conducted via a backflow line 14 into a reservoir 15 , passing a depressurizer 16 . thus the reservoir 15 is depressurized . the here discussed design inherits a choke - like depressurizer . in the embodiment , the water temperature inside the reservoir 15 is heated by a controlled heater to 72 ° c . which is technically quite easy to construct . the heated water is forced by a pump via an optional filter 18 through line 6 back to the annulus 7 of the humidifier 2 . the optional filter 18 guarantees that the recirculated water , and in consequence the respiratory gas flowing into the respiratory gas flow generator 3 is nearly free of particles or microorganisms . since there are present pump 19 , valve 13 and / or depressurizer 16 , the reservoir 15 can be of the low - pressure type and may be refilled at any time . fig2 shows an alternative embodiment where the humidifier 2 is placed between the respiratory gas flow generator 3 and the patient 5 as it is in conventional state - of - the - art designs ( see above ). in order to keep small both the compressible volume and physical dimensions of the device , the device is designed to handle the range of gas flows used in respiratory therapy ( 0 to 180 l / min ), and its humidification power is complying with the respective standards . warmed fluid ( in this example heated to about 37 ° c . is conducted via line 6 to the humidifier 2 where it enters the distribution chamber 7 . from the distribution chamber 7 the heated water circulates through the sieve bottom 8 into a sprinkling type chamber 9 filled with filling material . in this example the filling is made from solid structured elements with a huge surface and large cut - outs . in order to simplify the design one could do without the filling 10 and make the fluid coming from the sieve bottom 8 simply drop through the sprinkling type chamber 9 . under certain conditions this might be regarded advantageous for purposes of cleansing and reprocessing of the humidification chamber . it might well be , however , that in order to get the same performance with respect to humidification both the volume of the sprinkling type chamber and the circulated fluid volume flow have to be substantially increased . the water coming from the distribution chamber 7 trickling along the huge surface downwards into the bottom chamber 11 of the humidifier 2 is heating and wetting the filling 10 . simultaneously by streaming counterwise through the filling 10 the respiratory gas coming from the respiratory gas flow generator 3 is heated and humidified . nearly saturated ( relative humidity 95 to 100 %) it streams via the collecting chamber to the patient 5 . in order to avoid condensate formation preferably a temperature - controlled actively heated respiratory tubing 4 of known conventional technology is used . for the same reason it is of importance that the temperature of any part of the tubing &# 39 ; s wall in contact with the gas is equal or above the gas saturation temperature . usually the gas in the respiratory tubing 4 is heated to a temperature slightly higher , i . e . 40 ° c ., in order to prevent condensate formation even at the last few centimeters of the patient sided respiratory tubing and especially the unheated endotracheal tube . one should also mention that dependent on the working principle of the particular respiratory gas flow generator 3 model used only a single , or multiple respiratory tubings 4 , may be used . the patient &# 39 ; s expiratory gas might even be recirculated to the respiratory gas flow generator . under those circumstances the discussion above with respect to necessity and possibilities of heated respiratory tubing references also to all other types of respiratory tubing and parts thereof where there is the risk of condensate formation . as already discussed , humidifier 2 collects the fluid trickling through filling 10 into its bottom chamber 11 which functionally is a fluid reservoir . here by means of the temperature - controlled heater 17 the fluid is heated to a constant temperature of in the embodiment 37 ° c . the design of such a temperature - controlled heater is technically simple . the heated water is forced by pump 19 through the optional filter 18 and line 6 back to the distribution chamber 7 of the heater 2 . the optional filter 18 guarantees that both the circulated water , and in consequence the respiratory gas conducted to the patient 5 , are nearly free of particles or microorganisms . alternatively or as a supplement the device may inherit an antibacterial inner surface . the fluid level in the bottom chamber 11 may be regulated to a constant level by means of valve 21 controlling the fluid flow from a fluid reservoir 20 ( i . e . a bottle as commonly used for iv infusion ). valve 21 may well be constructed as flotation type valve . in this case it is important that the hydrostatic pressure difference between fluid reservoir 20 and bottom chamber 11 is always higher than the highest possible peak respiratory gas pressure . in practice the fluid reservoir 20 should be positioned at least 1 m above the bottom chamber 11 . for reasons of patient safety another valve 22 controls a bypass between the humidifier &# 39 ; s 2 gas inlet and gas outlet in order to ensure that with a high difference pressure ( i . e . provoked by a humidifier failure ) respiration is left unimpaired . surplus water and water trapped by the air is separated in the water trap 23 integrated into the collection area 12 . this water is recirculated into the bottom chamber 11 . for purposes of control of the humidification process and monitoring of the device &# 39 ; s proper function the control - and - monitoring device 29 is connected to the humidifier 2 and the patient 5 by means of several sensors and control lines . the temperature sensor 24 which may be of one of those sensor types commercially easily available is sensing the patient &# 39 ; s temperature which may be fed into the humidification process as target temperature . alternatively the target temperature may be dictated by another device , i . e . a patient monitoring device , communicating with the control - and - monitoring device 29 . the fluid temperature is measured by another temperature sensor 26 and controlled with heater 17 . the third temperature sensor 25 is sensing the gas temperature at the gas inlet . in case of a too high fluid level in the bottom chamber 11 due to a failure of valve 21 or pump 19 the control - and - monitoring device 29 is able to detect an unduly high fluid level by means of detecting the increase in temperature at the site of temperature sensor 25 . in such a case the bypass valve 22 will open . in consequence the temperature sensor 27 sensing the gas temperature at the gas outlet will detect a markable decrease in temperature . this security mechanism may be checked by voluntarily switching off the pump 19 ; in consequence , the fluid level should increase if the mechanism is working properly . the humidifier 2 is connected with the control - and - monitoring device 29 via the transponder 28 firmly connected to the humidifier 2 , in order to identify the humidifier and to monitor with regard to its lifetime . the control - and - monitoring device 29 is also able to monitor the power consumption of pump 19 . since the pump &# 39 ; s power consumption is a function of the fluid level in the bottom chamber 11 , the control - and - monitoring device 29 is also able to detect a too low fluid level ( i . e . due to an empty fluid reservoir 20 , failure of valve 12 , or failure of the fluid circulation due to an obstructed sieve bottom 8 ). an alternative design might be as shown in fig3 . the humidifier 2 according to fig2 is placed in this embodiment between the respiratory gas flow generator 3 and patient 5 . the fluid is in a bottom chamber 11 , and its temperature is controlled to a preset value by means of the heater 17 . the fluid in the bottom chamber 11 is held at a constant level by means of the fluid reservoir 20 and valve 21 as within fig2 . solid structured elements of the exchanger device 30 are periodically ( preferably by means of rotation ) partially submerged into the fluid in order to achieve a sufficient frequent exchange of the water within the contact area . during this procedure the geometrical design or the structured elements ( i . e . radial ribs , or similar ) has a significant impact on the amount of fluid taken along during the phase of submerging . a high rotation frequency may further increase the energy transport . if designed according to fig1 to 3 and correctly dimensioned a heavily varying or intermittently interrupted gas flow can be conditioned to nearly constant temperature and humidity . there is no risk of overheated gas after sudden gas flow interruption and there is no risk of a too low humidity if the gas flow suddenly becomes very high . in order to perform the object of the here presented invention the fluid / gas interface has to be dimensioned and equipped with such a geometric structure that even in case of the maximally allowed peak gas flow the energy and mass transfer can take place completely . the proper dimensioning may be checked by conducting the maximum allowed peak gas flow through the device according to the present invention . if dimensioned correctly the gas temperature after having passed the humidification chamber is nearly identical to that of the fluid before it enters the humidification chamber . suitable measures ( i . e . increasing the effective flow cross - sectional area ) will prevent the dragging of water droplets along by the gas . the use of heated respiratory tubing for the further transport of the gas will effectively prevent condensate formation within the tubing . in case there is the need to humidify the gas to less than saturation this may be realized by switching a suitable heater element into the stream of the already humidified gas . the humidifier according to the here presented invention conditions the gas to a temperature and humidity where the absolute humidity is according to requirements . the additional gas heater at the gas outlet , and / or the addition of dry gas , will adjust both the temperature and relative humidity of the gas to the target values . thus it is possible to generate gas of various combinations of temperature and humidity . the device according to the here presented invention is suitable not only for purposes of respiratory therapy but for all applications where a variable gas flow has to be cleansed of particles and / or conditioned to a preset / constant temperature or humidity . application examples are the insufflation of gases into body cavities ( i . e . carbon dioxide : laparoscopy ), the provision of respiratory gas in breathing protection applications ( i . e . painting ), all types of inhalation , and air conditioning ( i . e . of houses , vehicles , airplanes ) alone or in combination with other air - conditioning devices .
a device and a process for the heating and humidification of gas , especially respiratory gas . fluid from a fluid reservoir is supplied to a sprinkling type chamber where , for the purpose of humidification , it is moved through the gas . the point is that the fluid is heated by a temperature controlled heater to a preset temperature . in addition to the description of the device , a description of the process underlying the operation of the device is described .