the present invention provides a sterilization system that utilizes high velocity dry air heat to sterilize medical , veterinary , and dental instruments and devices that are situated in a novel container which allows the high velocity hot air to penetrate and diffuse within that container and which allows such sterilized instruments and devices to remain protected from environmental contaminants upon removal of the container from the sterilizer . the present invention embodies two distinct mechanisms that can be employed to open and close instrument containers . one embodiment describes a mechanism in which the sterilizer has a fixed engagement bar that physically engages a sliding cap on the container , pushing it to reveal an opening in the container &# 39 ; s base during container entry and pulling it back across the opening during the container &# 39 ; s exit from the sterilizer . another embodiment is described in which a pulley push - pull mechanism operated by the door latching and locking mechanism pulls the sliding cover to reveal an opening in the container &# 39 ; s base and pushes the sliding cover back across the opening to a closed position once the instrument container has been loaded into the sterilizer . the present invention is also envisioned applicable for other sterilization systems that may employ steam or sterilant gas or vapor , or heat to inactivate or kill microorganisms by providing direct access of the sterilizing agent to a medical instrument or article during a sterilization process and by protecting the sterilized instruments or articles from environmental contaminants once the container is removed from the sterilizer . the present invention may also be envisioned to bring air into the instrument container through the top ( lid ) or sides of the container . the preferred and described embodiment of the present invention is described below . fig1 a is a perspective view of the preferred rectangular sterilization container 15 . the sterilization container 15 is typically rectangular in shape with solid sidewalls 1 and end walls 2 and having a base 3 on which reside support feet 4 or perpendicular base extensions on its exterior . the solid base 3 contains an air entry portal 5 over which are baffles 6 that serve to deflect forced high - velocity hot air from the sterilizer into all portions of the container 15 . fig1 b and 2a are views of the container lid 7 . the lid 7 ( shown inverted ) has solid sidewalls 8 and end walls 9 . fig1 b and 2a depict the container lid top 10 , which is solid with a cut out portion 11 containing a nested and sealed thermal - resistant spunbond or nonwoven nylon , removable microbial filter 12 which is designed to assist in the cooling of the interior of the container 15 after the sterilization process . during sterilization the lid 7 is affixed to the container 15 and remains affixed until such time the sterile instruments are required . fig1 c is a perspective view ( shown inverted ) of the container lid 7 containing a solid baffle 28 to deflect hot air entering the container 15 from the air entry portal 5 during the sterilization process to aid in its diffusion throughout the container 15 . fig2 b , 3 , 4 , and 5 depict an embodiment in which an instrument container 15 has an external slider cap 20 to which is inserted a fixed engagement bar 31 that is attached to the sterilizer 40 . entry of the instrument container 15 into the sterilizer 40 results in the external slider cap 20 remaining stationary while the container 15 is manually pushed to the rear of the sterilizer 40 , thus uncovering the air entry portal 5 and allowing rapidly flowing hot air to enter the container 15 and sterilize instruments . upon completion of the sterilization process the container 15 is manually pulled to the front of the sterilizer 15 , closing and sealing the external slider cap 20 across the air entry portal 5 to preclude microorganisms from contaminating the sterilized instruments . the closed instrument container 15 can now be removed from the sterilizer 40 by lifting up the instrument container 15 , thus disengaging the fixed engagement bar 31 from the slider cap &# 39 ; s 20 insert well 22 . fig2 b is a view of the underside of the container base 3 having parallel externally - mounted slide rails 18 along the length of the container base 3 on which rides a slider cap 20 containing an external insert well 22 for the slider cap engagement bar 31 ( fig3 a and 3b ). the slider cap 20 covers the air entry portal 5 when in the closed position ( fig3 a and 4b ). the underside of the base 3 contains four container support feet 4 that protect the slider cap 20 and parallel mounted slide rails 18 that run along the length of the container base 3 . fig3 depicts a cross - sectional side elevation view of a preferred embodiment of the instrument container 15 and lid 7 with slider cap 20 and slide rails 18 with rail end caps 19 situated in both closed 29 and open 30 slider cap positions within the high - velocity hot air sterilization device 40 . the instrument container 15 holds a basket 14 or alternative instrument holding device which contains the instruments to be sterilized . the basket 14 sits on a perforated stainless steel floor covering 16 , which provides support for the basket 14 over the opening 5 in the container floor and the air baffles 6 . the container 15 is placed into the sterilizer 40 by means of a holding tray 17 , which provides guidance and positioning of the container 15 as it is pushed into the sterilizer 40 . positioning and guidance are required to properly seat the slider cap engagement bar 31 into the bar insert well 22 of the slider cap 20 . upon pushing the container 15 into the sterilizer chamber the sliding cap 20 is retained at the front of the sterilizer 40 while the container 15 is pushed to the rear of the sterilizer 40 thus allowing the slider cap 20 to slide away from the air entry portal 5 along the parallel slide rails 18 and allowing high velocity hot air to enter the container 15 from the bottom of the sterilizer 40 . upon completion of the sterilization process the instrument container 15 is pulled from the sterilizer 40 and the process is reversed , sliding the slider cap 20 to cover the air entry portal 5 . the sterilized container 15 is removed from the sterilizer 40 disengaging the coupling between the slider cap 20 and slider cap engagement bar 31 . subsequent to the sterilization process when the sterilizer door 41 is opened and instrument containers 15 are withdrawn from the sterilizer 40 , the sterilizer 40 continues to generate a laminar airflow curtain at the entrance to the sterilizer 40 chamber that is sufficient in airflow volume and speed to preclude entry into the sterilizer of environmental microbial contaminants . therefore the laminar airflow curtain also precludes potential microbial entry into the instrument container 15 as the slider cap 20 is pulled to the closed position during the container &# 39 ; s 15 removal . at any time during container 15 removal , the leading open edge of the air entry portal 5 always remains near the middle of the sterilizer 40 chamber , thus assuring no microbial contaminants from the outside environment can enter the instrument container 15 during its removal . to assist in the release of hot air retained inside the instrument container 15 after the sterilization process is complete , a removable spunbond or nonwoven nylon microbial filter 12 is contained in the lid 7 , sealed in a rigid housing 13 over the opening 11 . fig4 a and 4b depict cross - sectional views of the slider rail 18 and slider cap 20 with attached gasket 27 in both open 30 and closed 29 positions of the air entry portal 5 and air baffle 6 in the instrument container 15 . in the open position 30 the slider cap 20 with attached gasket 27 rides loosely on the slider rail 18 with little or no contact to the gasket 27 against bottom of the instrument container 15 . in this position the slider cap rests near or touching the front rail end cap 19 a . in the closed position 29 the slider cap 20 is pulled back into its position immediately under the air entry portal 5 . the slider cap 20 with attached gasket 27 is wedged tightly against the external bottom of the container base 3 immediately surrounding the perimeter of the air entry portal 5 by providing a slope and plateau on the back half of the interior bottoms of the parallel slide rails 18 causing the slider cap 20 to rise slightly and compress the attached gasket 27 tightly against the external bottom of the container base 3 . the rear rail end caps 19 b prevent the slider cap 20 from going beyond the correct sealable position . a rear view 32 is depicted in fig4 c of the slider rail 18 , slider cap 20 with attached gasket 27 , and rail end caps 19 in the closed 29 position under the air entry portal 5 and air baffle 6 in the instrument container 15 . the gasket 27 preferably is composed of a high - heat resistant material such as silicone or other comparable material that can be successfully compressed against the bottom of the container base 3 . the slider cap 20 , slide rails 18 , and rail end caps 19 are preferably composed of a hard , high - heat resistant plastic material or other material as found suitable for the application . fig5 a and 5b depict a cross - sectional top and side elevation views , respectively , of a preferred embodiment of a slider cap locking mechanism to retain the slider cap 20 in its closed position . channeled internally into the slider cap 20 are two cables 23 attached to a spring 26 - loaded push bar 21 on one end and on the other end to two spring - loaded pins 24 which are actively engaged into holding slots 25 located internally on each slide rail 18 at the back end of the slide rail 18 and in a position to lock the slider cap 20 in place directly centered beneath the air entry portal 5 in the instrument container 15 . upon entering the sterilizer 40 , the instrument container 15 is lowered onto the holding tray 17 and positioned by inserting the slider cap engagement bar 31 into the bar insert well 22 of the slider cap 20 . by pushing the instrument container 15 toward the back of the sterilizer 40 the slider cap 20 is retained in place , pushing the slider cap engagement bar 31 against the spring 26 - loaded push bar 21 toward the front of the bar insert well 22 thus tightening the two cables 23 attached the two spring - loaded pins 24 and pulling them out of their respective holding slots 25 in the slide rails 18 . the slider cap 20 is then free to move to the front portion of the container 15 in its open position . at the completion of the sterilization cycle , the instrument container 15 is pulled out of the sterilizer 40 , thereby forcing the slide cap engagement bar 31 against the back portion of bar insert well 22 and releasing the spring - loaded 26 push bar 21 to relax the two cables 23 and cause the two spring - loaded pins 24 to re - engage into their respective holding slots 25 , locking the slide cap 20 back in its closed position . although the preferred embodiment is the use of spring - loaded pins and push bar , it is envisioned that other locking mechanisms may also serve to lock and unlock the slider cap 20 which may or may not include springs or cables and may include the use of dense memory foams or metal clips . fig6 , 7 , and 8 depict a second embodiment in which an instrument container 15 has an internal sliding cover 50 which is moved from a closed position over the air entry portal 5 by a pulley 44 pull - push mechanism connected to the sterilizer door 41 latching and locking apparatus . upon positioning the container 15 on the holding tray 17 and insertion into the sterilizer 40 , a cable 45 is pulled forward via a pulley 44 when the door handle 42 is turned to its closed and locked position , pulling an internal sliding cover 50 and opening the air entry portal 5 . the mechanism is reversed upon completion of the sterilization cycle when the sterilizer door 41 is unlocked and the latch is positioned to the open position , pushing the cable 45 and thus the sliding cover 50 into the closed position over the air entry portal 5 . the closing of the sliding cover 50 is aided by a spring - loaded piston 53 or similar device located on the external bottom of the container base 3 . the spring - loaded piston 53 also serves to lock or hold the sliding cover 50 in place over the air entry portal 5 . fig6 a and 6b and fig6 c and d depict the internal sliding cover 50 configuration on the inside and outside , respectively , of the container base 3 in both the closed ( fig6 a and c ) and open positions ( fig6 b and d ). two parallel sliding cover rails 52 position the sliding cover 50 within the container 15 and allow it to slide freely from the closed position covering the air entry portal 5 to the open position uncovering the air entry portal 5 . attached to the bottom of the sliding cover 50 is a sliding cover push - pull bar 51 extending downward from the sliding cover 50 to engage a push - pull engagement bar 49 that pushes the sliding cover 50 to the open position uncovering the air entry portal 5 when the sterilizer door 41 is latched and locked ( fig7 ). parallel external rail guides 55 located externally on the container base 3 assist in the positioning of the push - pull engagement bar 49 during the uncovering and covering of the air entry portal 5 . a push - pull compression spring piston 53 is used to assist in the return of sliding cover 50 to the closed position covering the air entry portal 5 and to hold the sliding cover 50 in place to prevent microbial contamination of sterilized instruments during storage . the compression end of a push - pull spring piston 53 is externally attached to the instrument container base 3 and the distal piston end is attached to sliding cover push - pull bar 51 . the push - pull spring piston 53 is in the extended mode when the sliding cover 50 is covers the air entry portal 5 ( fig6 a and 6c ) and is compressed when the sliding cover 50 has been pushed to the open position ( fig6 b and 6d ), uncovering the air entry portal 5 to allow hot rapidly flowing air entry into the instrument container 15 . although the preferred embodiment is the use of a push - pull compression spring or rod , or combination thereof , it is envisioned that other similarly functional devices or mechanisms may also be employed . alternative means to move the sliding cover 50 or slider cap 20 are also envisioned using other mechanical means with ( e . g ., electromagnetic means or servo motors ) with or without electrical or hydraulic / pneumatic assist and with or without pulley assist ( e . g ., levers or other mechanical means ). fig7 depicts a cross - sectional side elevation view of a preferred embodiment of the instrument container 15 incorporating an internal sliding cover 50 . fig7 a depicts the sliding cover 50 in the closed position over the air entry portal 5 and fig7 b depicts the sliding cover 50 in the open position aside the air entry portal 5 . the sliding mechanism is activated once the container 15 has been placed on the holding tray 17 , inserted completely into the sterilizer 40 , and during the closing and locking the sterilizer door 41 . rotating the sterilizer door latch handle 42 engages the sterilizer door latch 43 to the locking position , simultaneously rotating the cable pulley 44 to which is attached a cable 45 . pulled around the cable pulley 44 , the cable 45 pulls the push - pull engagement bar 49 forward by way of a push - pull rod 46 intermediate . thus engaged , the sliding cover push - pull bar 51 pushes the sliding cover 50 , positioned by the parallel external guide rails 55 , to uncover the air entry portal 5 to the backstop 47 . the push - pull engagement bar 49 is held in position and kept on track by way of push - pull engagement bar guide 48 . as the push - pull engagement bar 49 pushes the sliding cover push - pull bar 51 , the push - pull spring piston 53 compresses against the internal ( or external ) compression spring 54 . the push - pull spring piston 53 is held in this position until the sterilizer door latch 43 is released and the sterilizer door latch handle 42 is counter - rotated causing the cable pulley 44 to unwind the cable 45 and release the tension to the push - pull engagement bar 49 . as the tension is released , the pressure against the push - pull spring piston 53 is subsequently lessened and the push - pull spring piston 53 pushes the sliding cover push - pull bar 51 and sliding cover 50 back to the closed position where the remaining tension in the push - pull spring piston 53 holds the sliding cover 50 in place over the air entry portal 5 . fig8 a and fig8 b represent the spatial configuration of the push - pull engagement bar 49 within the sterilizer 40 in relationship to an instrument container or containers 15 , holding tray 17 , and the sliding cover push - pull bar 51 . fig8 depicts a sterilizer 40 holding two instrument containers 15 side - by - side with the push - pull engagement bar 49 engaging both containers 15 and pulled / pushed by a single cable 46 attached to the cable pulley 44 . the push - pull engagement bar 49 is positioned and guided by parallel push - pull engagement bar guides 48 attached in the sterilizer 40 floor . this configuration allows multiple containers 15 to be handled by a single push - pull engagement bar 49 whether on a single plane as depicted and / or on multiple tiers or levels .
a device and system is disclosed for sterilizing objects , commonly dental , medical , or veterinary instruments , by enhancing rapid hot air diffusion into an instrument container by means of a sliding cover overlaying an opening in the bottom of the instrument container when in the closed position and sliding horizontally to reveal the opening in the open position , allowing rapidly flowing hot air into the container for medical article sterilization . the sliding cover is moved to the open position just prior to the sterilization process by a mechanical means incorporated into the sterilizer with the sliding cover remaining in the open position through the completion of the sterilization process . once the sterilization cycle is complete , the mechanical means is reversed , closing , sealing , and locking the sliding cover in place over the opening before the instrument container is removed from the sterilizer , ensuring continued medical instrumentation sterility inside the container .