reference will now be made in detail to the present preferred embodiments of the invention , examples of which are illustrated in the accompanying drawings . wherever possible , the same reference numerals will be used throughout the drawings to refer to the same or like parts . in the drawings , the depicted structural elements are not to scale and certain components are enlarged relative to the other components for purposes of emphasis and understanding . as noted above , the embodiments described herein pertain to iv connectors / ports , with or without valves . needleless iv connectors are connected to one end of an iv tube or luer and may remain in place for the duration of a patent &# 39 ; s stay in the medical facility in which the iv is administered . such needleless iv connectors may thus be handled by various staff members and may be used to administer numerous dosages of medication or to withdraw fluid . because such needleless iv connectors provide a path into the patient &# 39 ; s bloodstream , it is important that they do not harbor bacteria or other microbes . a novel needleless iv connector is disclosed herein that not only provides the requisite properties for a needleless iv connector , but also eliminates bacteria on contact . as discussed below , the components of the connector are either treated with or formed with an antimicrobial substance comprising a silane quaternary ammonium salt . one example of a needleless iv connector that may be so treated is shown in fig1 - 5 , which correspond to fig1 - 5 of the aforementioned u . s . pat . no . 5 , 685 , 866 , the entire disclosure of which is hereby incorporated herein by reference . as illustrated , needleless iv connector 10 has a valve body 12 , a seal ( or septum ) 36 , and a spike element 24 . valve body 12 includes an upper conduit 20 , a skirt 16 , an annular ring 14 , locking ears 22 ( or optionally threads ), and an annular lip 25 with an opening 25 a . valve body 12 provides an opening with threads 44 that may engage a cap or other medical device such as a container , iv tube , or luer . valve body 12 is generally tubular as discussed further below . valve body 12 may be formed of a thermal plastic . seal 36 has a seal cap 40 with a generally flat top surface 40 b , a lower lip 42 , and a tapered side - wall 38 having a plurality of longitudinal grooves 43 to facilitate compression of the seal . seal 36 has an opening at the end where the lower lip 42 is located presenting a cavity 37 for receiving spike element 24 . the outer dimensions of seal 36 are chosen to allow seal 36 to be inserted within the inside of valve body 12 . seal 36 may be formed of silicone or some other pliable material . spike element 24 includes a hollow spike 26 terminating at a tip 32 at one end and an inner conduit 18 at the other end . an annular cuff 28 is positioned between hollow spike 26 and inner conduit 18 . hollow spike 26 includes a longitudinal through hole 34 and protruding ridges 30 . annular cuff 28 may include a rim 28 a and an annular detent 28 b for receiving lower lip 42 of seal 36 . annular detent 28 b snaps into an annular groove 14 b in the underside of annular ring 14 . seal 36 fits into annular cuff 28 and is held in place by an internal lip 27 . spike element 24 may be formed of a thermal plastic . as shown in fig4 and 5 , a syringe 46 may have its nose 48 pushed against flat top surface 40 b of seal cap 40 so as to compress seal 36 until nose 48 receives the spike element 24 sufficiently deep that through hole 34 is exposed in syringe 46 . a plunger 49 may then be drawn outward to draw fluid . the valve may be rotated 180 degrees such that fluid flows in the opposite direction whereby medication may be injected from the syringe . in general , silicone such as that used for seal 36 is formed from a slurry processed at relatively low temperatures . the low temperatures at which the slurry is formed allow an antimicrobial material to be mixed in with the slurry and therefore integrated within the resulting silicone seal . also , the outer surfaces of the thermal plastic portions ( i . e ., the valve body 12 and spike element 24 ) may be wiped , sprayed or dipped in the antimicrobial material . thus , for example , an iv connector may be made having improved antimicrobial properties , not only on the outside of the connector , but also throughout the silicone seal . as mentioned above , the antimicrobial material includes a silane quaternary ammonium salt . preferred commercially available silane quaternary ammonium salts include : microbecare ™ xlp , which is available from microbecare , llc of allendale , mich . ; “ promofresh x 105 ” from piedmont chemical industries i , llc of high point , n . c . ; and aem 5772 antimicrobial , which is available from aegis environments of midland , mich .. the antimicrobial could also be aegis microbe shield ™ ( from aegis environments , midland , mich . ), which is a copolymer of chloropropyltrihydroxysilane and octadecylaminodimethyltrihydroxysilylpropyl ammonium chloride . silane quaternary ammonium salts are particularly well - suited for the antimicrobial material as they are long lasting and capable of emitting ions that aid in the destruction of a microbe . in addition , they are organofunctional silanes that include a monomer including a silane , a positively charged nitrogen molecule , and a long molecular chain . the silane bases of these monomers can covalently and permanently bond to each other and any surface . in addition , silane quaternary ammonium salts are preferable as they are substantially free from arsenic , silver , tin , heavy metals and polychlorinated phenols . also , they are safe to apply to the patient . in addition , it not only eliminates bacteria on contact , but it remains on the treated surfaces and kills any bacteria subsequently contacting these surfaces . such treatment preferably lasts at least one week , more preferably several months , and most preferably indefinitely . microbes may include bacteria , mold , mildew , algae , etc . the cell membranes of the microbes are attracted to , and then are punctured by , the long molecular chains of the monomers . as the microbes are drawn closer because of the positive - negative ion exchanges , the monomers penetrate further into the cell membranes . once the cell membranes are penetrated deeply , they are physically ruptured by a sword - like action and then electrocuted by positively charged nitrogen molecules of the monomers , thus destroying the microbes . thus , the microbes are eliminated without “ using up ” any of the antimicrobial active ingredients , which remain on and within the iv connector ready to continue protecting the patient against further microbial contamination . if the antimicrobial material used in the silicone slurry is a 3 -( trimethoxysilyl ) propyldimethyloctadecyl ammonium chloride material available from microbecare , llc , it includes an active ingredient of about 43 % by weight 3 -( trimethoxysilyl ) propyldimethyloctadecyl ammonium chloride and about 57 % by weight inert ingredients . the antimicrobial substance may then be mixed into the silicone slurry in various amounts of anywhere from about 0 . 001 % to about 50 % by weight to achieve the desired degree of antimicrobial activity while not compromising the desirable properties of the silicone sea i . to coat the thermal plastic components they may be sprayed or dipped in a solution containing the antimicrobial material . alternatively , the antimicrobial treatment solution may be applied using wipes soaked in such a solution . suitable wipes and solutions are disclosed in commonly - assigned u . s . pat . no . 8 , 491 , 922 , the entire disclosure of which is incorporated herein by reference . in this case , the antimicrobial material is again one of the silane quaternary ammonium salts described above . in a preferred form , the antimicrobial treatment solution contains 30 - 50 percent by volume isopropyl alcohol and 50 - 70 percent by volume antimicrobial treatment substance , which is preferably a silane quaternary ammonium salt having an unreacted organofunctional silane . if the antimicrobial treatment solution is applied by spraying or dipping , the solution most preferably includes 50 percent by volume isopropyl alcohol and 50 percent by volume of the unreacted antimicrobial treatment substance . if the solution is applied using the wipes , the solution is preferably 30 percent by volume isopropyl alcohol and 70 percent by volume of the unreacted antimicrobial treatment substance . the isopropyl alcohol may have a concentration of 70 - 90 percent by volume . by providing the unreacted organofunctional silane in isopropyl alcohol , the organofunctional silane does not react with the wipe substrates or the inside of the wipe container such that it is free to later react and permanently covalently bond with the inner and outer surfaces of the plastic components . isopropyl alcohol is preferred as it evaporates quickly once the solution is wiped onto the treated surface to allow the unreacted organofunctional silane to more quickly react with the treated surface . the preferred organofunctional silane quaternary ammonium salt also prevents odor , staining and product deterioration that may be associated with microbe contamination . the preferred organofunctional silane quaternary ammonium salt is also beneficial because it permanently bonds to a treated surface , covers a broad spectrum of activity with no negative effects or drawbacks , and is easily incorporated and easily verifiable . the preferred organofunctional silane quaternary ammonium salt is designed to react and create a covalent bond with the surfaces of the plastic components . the reacted substance is held onto those surfaces until the covalent bond is broken . tests have shown that most industrial cleaners or disinfectants will not remove the preferred antimicrobial treatment substance . the method of removal is by abrasion . in addition to treating the plastic components of the connector , male and female luer caps that may be placed on either ends of the connector may also be treated with the antimicrobial solution . examples of caps that may be treated include the dualcap ™ caps for use on needleless luer access valves , which is available from catheter connections , inc . of salt lake city utah . although the invention is described with respect to a particular construction of a needleless iv connector shown in fig1 - 5 , the construction thereof may vary . for example , the valve portion in the seal may include a duck bill - type valve . an example of such a needleless catheter connector with a duck bill valve is available from icu medical , inc . under the trademark neutron ™. also , the present invention may be applied to arterial connectors , catheter connectors , and dialysis connectors . the above description is considered that of the preferred embodiments only . modifications of the invention will occur to those skilled in the art and to those who make or use the invention . therefore , it is understood that the embodiments shown in the drawings and described above are merely for illustrative purposes and not intended to limit the scope of the invention , which is defined by the claims as interpreted according to the principles of patent law , including the doctrine of equivalents .
an intravenous connector is provided having a valve body having first end configured to receive a syringe for administering medicine to a patient ; a spike element having a tapered hollow spike terminating in a tip at a first end and having a second end configured for attaching to an intravenous tube , the spike element secured in the valve body such that the tip is disposed proximate the first end of the valve body and the second end of the spike element is located at a second end of the valve body ; and a pliable seal member disposed in the valve body around the tapered hollow spike and exposed at the first end of the valve body . an antimicrobial material comprising a silane quaternary ammonium salt is incorporated on or within the seal member and is coated on the valve body and the spike element , and wherein the antimicrobial material .