reference is made to fig1 a - 1e which schematically illustrate various stages of a cell division process . fig1 a illustrates a cell 10 at its normal geometry , which can be generally spherical ( as illustrated in the drawings ), ellipsoidal , cylindrical , “ pancake - like ” or any other cell geometry , as is known in the art . fig1 b - 1d illustrate cell 10 during different stages of its division process , which results in the formation of two new cells 18 and 20 , shown in fig1 e . as shown in fig1 b - 1d , the division process of cell 10 is characterized by a slowly growing cleft 12 which gradually separates cell 10 into two units , namely sub - cells 14 and 16 , which eventually evolve into new cells 18 and 20 ( fig1 e ). a shown specifically in fig1 d , the division process is characterized by a transient period during which the structure of cell 10 is basically that of the two sub - cells 14 and 16 interconnected by a narrow “ bridge ” 22 containing cell material ( cytoplasm surrounded by cell membrane ). reference is now made to fig2 a and 2b , which schematically illustrate non - dividing cell 10 being subjected to an electric field produced by applying an alternating electric potential , at a relatively low frequency and at a relatively high frequency , respectively . cell 10 includes intracellular organelles , e . g ., a nucleus 30 . alternating electric potential is applied across electrodes 28 and 32 that can be attached externally to a patient at a predetermined region , e . g ., in the vicinity of the tumor being treated . when cell 10 is under natural conditions , i . e ., part of a living tissue , it is disposed in a conductive environment ( hereinafter referred to as a “ volume conductor ”) consisting mostly of electrolytic inter - cellular liquid . when an electric potential is applied across electrodes 28 and 32 , some of the field lines of the resultant electric field ( or the current induced in the tissue in response to the electric field ) penetrate the cell 10 , while the rest of the field lines ( or induced current ) flow in the surrounding medium . the specific distribution of the electric field lines , which is substantially consistent with the direction of current flow in this instance , depends on the geometry and the electric properties of the system components , e . g ., the relative conductivities and dielectric constants of the system components , that can be frequency dependent . for low frequencies , e . g ., frequencies lower than 10 khz , the conductance properties of the components completely dominate the current flow and the field distribution , and the field distribution is generally as depicted in fig2 a . at higher frequencies , e . g ., at frequencies of between 10 khz and 1 mhz , the dielectric properties of the components becomes more significant and eventually dominate the field distribution , resulting in field distribution lines as depicted generally in fig2 b . for constant ( i . e ., dc ) electric fields or relatively low frequency alternating electric fields , for example , frequencies under 10 khz , the dielectric properties of the various components are not significant in determining and computing the field distribution . therefore , as a first approximation , with regard to the electric field distribution , the system can be reasonably represented by the relative impedances of its various components . using this approximation , the intercellular ( i . e ., extracellular ) fluid and the intracellular fluid each has a relatively low impedance , while the cell membrane 11 has a relatively high impedance . thus , under low frequency conditions , only a fraction of the electric field lines ( or currents induced by the electric field ) penetrate membrane 11 of the cell 10 . at relatively high frequencies ( e . g ., 10 khz - 1 mhz ), in contrast , the impedance of membrane 11 relative to the intercellular and intracellular fluids decreases , and thus , the fraction of currents penetrating the cells increases significantly . it should be noted that at very high frequencies , i . e ., above 1 mhz , the membrane capacitance can short the membrane resistance and , therefore , the total membrane resistance can become negligible . in any of the embodiments described above , the electric field lines ( or induced currents ) penetrate cell 10 from a portion of the membrane 11 closest to one of the electrodes generating the current , e . g ., closest to positive electrode 28 ( also referred to herein as “ source ”). the current flow pattern across cell 10 is generally uniform because , under the above approximation , the field induced inside the cell is substantially homogeneous . the currents exit cell 10 through a portion of membrane 11 closest to the opposite electrode , e . g ., negative electrode 32 ( also referred to herein as “ sink ”). the distinction between field lines and current flow can depend on a number of factors , for example , on the frequency of the applied electric potential and on whether electrodes 28 and 32 are electrically insulated . for insulated electrodes applying a dc or low frequency alternating voltage , there is practically no current flow along the lines of the electric field . at higher frequencies , the displacement currents are induced in the tissue due to charging and discharging of the electrode insulation and the cell membranes ( which act as capacitors to a certain extent ), and such currents follow the lines of the electric field . fields generated by non - insulated electrodes , in contrast , always generate some form of current flow , specifically , dc or low frequency alternating fields generate conductive current flow along the field lines , and high frequency alternating fields generate both conduction and displacement currents along the field lines . it should be appreciated , however , that movement of polarizable intracellular organelles according to the present invention ( as described below ) is not dependent on actual flow of current and , therefore , both insulated and non - insulated electrodes can be used efficiently . advantages of insulated electrodes include lower power consumption , less heating of the treated regions , and improved patient safety . according to one exemplary embodiment of the present invention , the electric fields that are used are alternating fields having frequencies that are in the range from about 50 khz to about 500 khz , and preferably from about 100 khz to about 300 khz . for ease of discussion , these type of electric fields are also referred to below as “ tc fields ”, which is an abbreviation of “ tumor curing electric fields ”, since these electric fields fall into an intermediate category ( between high and low frequency ranges ) that have bio - effective field properties while having no meaningful stimulatory and thermal effects . these frequencies are sufficiently low so that the system behavior is determined by the system &# 39 ; s ohmic ( conductive ) properties but sufficiently high enough not to have any stimulation effect on excitable tissues . such a system consists of two types of elements , namely , the intercellular , or extracellular fluid , or medium and the individual cells . the intercellular fluid is mostly an electrolyte with a specific resistance of about 40 - 100 ohm * cm . as mentioned above , the cells are characterized by three elements , namely ( 1 ) a thin , highly electric resistive membrane that coats the cell ; ( 2 ) internal cytoplasm that is mostly an electrolyte that contains numerous macromolecules and micro - organelles , including the nucleus ; and ( 3 ) membranes , similar in their electric properties to the cell membrane , cover the micro - organelles . when this type of system is subjected to the present tc fields ( e . g ., alternating electric fields in the frequency range of 100 khz - 300 khz ) most of the lines of the electric field and currents tend away from the cells because of the high resistive cell membrane and therefore the lines remain in the extracellular conductive medium . in the above recited frequency range , the actual fraction of electric field or currents that penetrates the cells is a strong function of the frequency . fig2 schematically depicts the resulting field distribution in the system . as illustrated , the lines of force , which also depict the lines of potential current flow across the cell volume mostly in parallel with the undistorted lines of force ( the main direction of the electric field ). in other words , the field inside the cells is mostly homogeneous . in practice , the fraction of the field or current that penetrates the cells is determined by the cell membrane impedance value relative to that of the extracellular fluid . since the equivalent electric circuit of the cell membrane is that of a resistor and capacitor in parallel , the impedance is a function of the frequency . the higher the frequency , the lower the impedance , the larger the fraction of penetrating current and the smaller the field distortion ( rotshenker s . & amp ; y . palti , changes in fraction of current penetrating an axon as a function of duration of stimulating pulse , j . theor . biol . 41 ; 401 - 407 ( 1973 ). as previously mentioned , when cells are subjected to relatively weak electric fields and currents that alternate at high frequencies , such as the present tc fields having a frequency in the range of 50 khz - 500 khz , they have no effect on the non - dividing cells . while the present tc fields have no detectable effect on such systems , the situation becomes different in the presence of dividing cells . reference is now made to fig3 a - 3c which schematically illustrate the electric current flow pattern in cell 10 during its division process , under the influence of alternating fields ( tc fields ) in the frequency range from about 100 khz to about 300 khz in accordance with one exemplary embodiment . the field lines or induced currents penetrate cell 10 through a part of the membrane of sub - cell 16 closer to electrode 28 . however , they do not exit through the cytoplasm bridge 22 that connects sub - cell 16 with the newly formed yet still attached sub - cell 14 , or through a part of the membrane in the vicinity of the bridge 22 . instead , the electric field or current flow lines — that are relatively widely separated in sub - cell 16 — converge as they approach bridge 22 ( also referred to as “ neck ” 22 ) and , thus , the current / field line density within neck 22 is increased dramatically . a “ mirror image ” process takes place in sub - cell 14 , whereby the converging field lines in bridge 22 diverge as they approach the exit region of sub - cell 14 . it should be appreciated by persons skilled in the art that homogeneous electric fields do not exert a force on electrically neutral objects , i . e ., objects having substantially zero net charge , although such objects can become polarized . however , under a non - uniform , converging electric field , as shown in fig3 a - 3c , electric forces are exerted on polarized objects , moving them in the direction of the higher density electric field lines . it will be appreciated that the concentrated electric field that is present in the neck or bridge area in itself exerts strong forces on charges and natural dipoles and can disrupt structures that are associated therewith . one will understand that similar net forces act on charges in an alternating field , again in the direction of the field of higher intensity . in the configuration of fig3 a and 3b , the direction of movement of polarized and charged objects is towards the higher density electric field lines , i . e ., towards the cytoplasm bridge 22 between sub - cells 14 and 16 . it is known in the art that all intracellular organelles , for example , nuclei 24 and 26 of sub - cells 14 and 16 , respectively , are polarizable and , thus , such intracellular organelles are electrically forced in the direction of the bridge 22 . since the movement is always from lower density currents to the higher density currents , regardless of the field polarity , the forces applied by the alternating electric field to organelles , such as nuclei 24 and 26 , are always in the direction of bridge 22 . a comprehensive description of such forces and the resulting movement of macromolecules of intracellular organelles , a phenomenon referred to as “ dielectrophoresis ” is described extensively in literature , e . g ., in c . l . asbury & amp ; g . van den engh , biophys . j . 74 , 1024 - 1030 , 1998 , the disclosure of which is hereby incorporated by reference in its entirety . the movement of the organelles 24 and 26 towards the bridge 22 disrupts the structure of the dividing cell , change the concentration of the various cell constituents and , eventually , the pressure of the converging organelles on bridge membrane 22 results in the breakage of cell membrane 11 at the vicinity of the bridge 22 , as shown schematically in fig3 c . the ability to break membrane 11 at bridge 22 and to otherwise disrupt the cell structure and organization can be enhanced by applying a pulsating ac electric field , rather than a steady ac field . when a pulsating field is applied , the forces acting on organelles 24 and 26 have a “ hammering ” effect , whereby pulsed forces beat on the intracellular organelles towards the neck 22 from both sub - cells 14 and 16 , thereby increasing the probability of breaking cell membrane 11 in the vicinity of neck 22 . a very important element , which is very susceptible to the special fields that develop within the dividing cells is the microtubule spindle that plays a major role in the division process . in fig4 , a dividing cell 10 is illustrated , at an earlier stage as compared to fig3 a and 3b , under the influence of external tc fields ( e . g ., alternating fields in the frequency range of about 100 khz to about 300 khz ), generally indicated as lines 100 , with a corresponding spindle mechanism generally indicated at 120 . the lines 120 are microtubules that are known to have a very strong dipole moment . this strong polarization makes the tubules , as well as other polar macromolecules and especially those that have a specific orientation within the cells or its surrounding , susceptible to electric fields . their positive charges are located at the two centrioles while two sets of negative poles are at the center of the dividing cell and the other pair is at the points of attachment of the microtubules to the cell membrane , generally indicated at 130 . this structure forms sets of double dipoles and therefore they are susceptible to fields of different directions . it will be understood that the effect of the tc fields on the dipoles does not depend on the formation of the bridge ( neck ) and thus , the dipoles are influenced by the tc fields prior to the formation of the bridge ( neck ). since the present apparatus ( as will be described in greater detail below ) utilizes insulated electrodes , the above - mentioned negative effects obtained when conductive electrodes are used , i . e ., ion concentration changes in the cells and the formation of harmful agents by electrolysis , do not occur when the present apparatus is used . this is because , in general , no actual transfer of charges takes place between the electrodes and the medium and there is no charge flow in the medium where the currents are capacitive , i . e ., are expressed only as rotation of charges , etc . turning now to fig5 , the tc fields described above that have been found to advantageously destroy tumor cells are generated by an electronic apparatus 200 . fig5 is a simple schematic diagram of the electronic apparatus 200 illustrating the major components thereof . the electronic apparatus 200 generates the desired electric signals ( tc signals ) in the shape of waveforms or trains of pulses . the apparatus 200 includes a generator 210 and a pair of conductive leads 220 that are attached at one end thereof to the generator 210 . the opposite ends of the leads 220 are connected to insulated conductors 230 that are activated by the electric signals ( e . g ., waveforms ). the insulated conductors 230 are also referred to hereinafter as isolects 230 . optionally and according to another exemplary embodiment , the apparatus 200 includes a temperature sensor 240 and a control box 250 which are both added to control the amplitude of the electric field generated so as not to generate excessive heating in the area that is treated . the generator 210 generates an alternating voltage waveform at frequencies in the range from about 50 khz to about 500 khz ( preferably from about 100 khz to about 300 khz ) ( i . e ., the tc fields ). the required voltages are such that the electric field intensity in the tissue to be treated is in the range of about 0 . 1 v / cm to about 10 v / cm . to achieve this field , the actual potential difference between the two conductors in the isolects 230 is determined by the relative impedances of the system components , as described below . when the control box 250 is included , it controls the output of the generator 210 so that it will remain constant at the value preset by the user or the control box 250 sets the output at the maximal value that does not cause excessive heating , or the control box 250 issues a warning or the like when the temperature ( sensed by temperature sensor 240 ) exceeds a preset limit . the leads 220 are standard isolated conductors with a flexible metal shield , preferably grounded so that it prevents the spread of the electric field generated by the leads 220 . the isolects 230 have specific shapes and positioning so as to generate an electric field of the desired configuration , direction and intensity at the target volume and only there so as to focus the treatment . the specifications of the apparatus 200 as a whole and its individual components are largely influenced by the fact that at the frequency of the present tc fields ( 50 khz - 500 khz ), living systems behave according to their “ ohmic ”, rather than their dielectric properties . the only elements in the apparatus 200 that behave differently are the insulators of the isolects 230 ( see fig7 - 9 ). the isolects 200 consist of a conductor in contact with a dielectric that is in contact with the conductive tissue thus forming a capacitor . the details of the construction of the isolects 230 is based on their electric behavior that can be understood from their simplified electric circuit when in contact with tissue as generally illustrated in fig6 . in the illustrated arrangement , the potential drop or the electric field distribution between the different components is determined by their relative electric impedance , i . e ., the fraction of the field on each component is given by the value of its impedance divided by the total circuit impedance . for example , the potential drop on element δv a = a /( a + b + c + d + e ). thus , for dc or low frequency ac , practically all the potential drop is on the capacitor ( that acts as an insulator ). for relatively very high frequencies , the capacitor practically is a short and therefore , practically all the field is distributed in the tissues . at the frequencies of the present tc fields ( e . g ., 50 khz to 500 khz ), which are intermediate frequencies , the impedance of the capacitance of the capacitors is dominant and determines the field distribution . therefore , in order to increase the effective voltage drop across the tissues ( field intensity ), the impedance of the capacitors is to be decreased ( i . e ., increase their capacitance ). this can be achieved by increasing the effective area of the “ plates ” of the capacitor , decrease the thickness of the dielectric or use a dielectric with high dielectric constant . in order to optimize the field distribution , the isolects 230 are configured differently depending upon the application in which the isolects 230 are to be used . there are two principle modes for applying the present electric fields ( tc fields ). first , the tc fields can be applied by external isolects and second , the tc fields can be applied by internal isolects . electric fields ( tc fields ) that are applied by external isolects can be of a local type or widely distributed type . the first type includes , for example , the treatment of skin tumors and treatment of lesions close to the skin surface . fig7 illustrates an exemplary embodiment where the isolects 230 are incorporated in a skin patch 300 . the skin patch 300 can be a self - adhesive flexible patch with one or more pairs of isolects 230 . the patch 300 includes internal insulation 310 ( formed of a dielectric material ) and the external insulation 260 and is applied to skin surface 301 that contains a tumor 303 either on the skin surface 301 or slightly below the skin surface 301 . tissue is generally indicated at 305 . to prevent the potential drop across the internal insulation 310 to dominate the system , the internal insulation 310 must have a relatively high capacity . this can be achieved by a large surface area ; however , this may not be desired as it will result in the spread of the field over a large area ( e . g ., an area larger than required to treat the tumor ). alternatively , the internal insulation 310 can be made very thin and / or the internal insulation 310 can be of a high dielectric constant . as the skin resistance between the electrodes ( labeled as a and e in fig6 ) is normally significantly higher than that of the tissue ( labeled as c in fig6 ) underneath it ( 1 - 10 kω vs . 0 . 1 - 1 kω ), most of the potential drop beyond the isolects occurs there . to accommodate for these impedances ( z ), the characteristics of the internal insulation 310 ( labeled as b and d in fig6 ) should be such that they have impedance preferably under 100 kω at the frequencies of the present tc fields ( e . g ., 50 khz to 500 khz ). for example , if it is desired for the impedance to be about 10 k ohms or less , such that over 1 % of the applied voltage falls on the tissues , for isolects with a surface area of 10 mm 2 , at frequencies of 200 khz , the capacity should be on the order of 10 − 10 f ., which means that using standard insulations with a dielectric constant of 2 - 3 , the thickness of the insulating layer 310 should be about 50 - 100 microns . an internal field 10 times stronger would be obtained with insulators with a dielectric constant of about 20 - 50 . using an insulating material with a high dielectric constant increases the capacitance of the electrodes , which results in a reduction of the electrodes &# 39 ; impedance to the ac signal that is applied by the generator 1 ( shown in fig5 ). because the electrodes a , e are wired in series with the target tissue c , as shown in fig6 , this reduction in impedance reduces the voltage drop in the electrodes , so that a larger portion of the applied ac voltage appears across the tissue c . since a larger portion of the voltage appears across the tissue , the voltage that is being applied by the generator 1 can be advantageously lowered for a given field strength in the tissue . the desired field strength in the tissue being treated is preferably between about 0 . 1 v / cm and about 10 v / cm , and more preferably between about 2 v / cm and 3 v / cm or between about 1 v / cm and about 5 v / cm . if the dielectric constant used in the electrode is sufficiently high , the impedance of the electrodes a , e drops down to the same order of magnitude as the series combination of the skin and tissue b , c , d . one example of a suitable material with an extremely high dielectric constant is cacu 3 ti 4 o 12 , which has a dielectric constant of about 11 , 000 ( measured at 100 khz ). when the dielectric constant is this high , useful fields can be obtained using a generator voltage that is on the order of a few tens of volts . since the thin insulating layer can be very vulnerable , etc ., the insulation can be replaced by very high dielectric constant insulating materials , such as titanium dioxide ( e . g ., rutile ), the dielectric constant can reach values of about 200 . there a number of different materials that are suitable for use in the intended application and have high dielectric constants . for example , some materials include : lithium niobate ( linbo 3 ), which is a ferroelectric crystal and has a number of applications in optical , pyroelectric and piezoelectric devices ; yttrium iron garnet ( yig ) is a ferromagnetic crystal and magneto - optical devices , e . g ., optical isolator can be realized from this material ; barium titanate ( batio 3 ) is a ferromagnetic crystal with a large electro - optic effect ; potassium tantalate ( ktao 3 ) which is a dielectric crystal ( ferroelectric at low temperature ) and has very low microwave loss and tunability of dielectric constant at low temperature ; and lithium tantalate ( litao 3 ) which is a ferroelectric crystal with similar properties as lithium niobate and has utility in electro - optical , pyroelectric and piezoelectric devices . insulator ceramics with high dielectric constants may also be used , such as a ceramic made of a combination of lead magnesium niobate and lead titanate . it will be understood that the aforementioned exemplary materials can be used in combination with the present device where it is desired to use a material having a high dielectric constant . one must also consider another factor that affects the effective capacity of the isolects 230 , namely the presence of air between the isolects 230 and the skin . such presence , which is not easy to prevent , introduces a layer of an insulator with a dielectric constant of 1 . 0 , a factor that significantly lowers the effective capacity of the isolects 230 and neutralizes the advantages of the titanium dioxide ( rutile ), etc . to overcome this problem , the isolects 230 can be shaped so as to conform with the body structure and / or ( 2 ) an intervening filler 270 ( as illustrated in fig1 c ), such as a gel , that has high conductance and a high effective dielectric constant , can be added to the structure . the shaping can be pre - structured ( see fig1 a ) or the system can be made sufficiently flexible so that shaping of the isolects 230 is readily achievable . the gel can be contained in place by having an elevated rim as depicted in fig1 c and 10 c ′. the gel can be made of hydrogels , gelatins , agar , etc ., and can have salts dissolved in it to increase its conductivity . fig1 a - 10 c ′ illustrate various exemplary configurations for the isolects 230 . the exact thickness of the gel is not important so long as it is of sufficient thickness that the gel layer does not dry out during the treatment . in one exemplary embodiment , the thickness of the gel is about 0 . 5 mm to about 2 mm . preferably , the gel has high conductivity , is tacky , and is biocompatible for extended periods of time . one suitable gel is ag603 hydrogel , which is available from amgel technologies , 1667 s . mission road , fallbrook , calif . 92028 - 4115 , usa . in order to achieve the desirable features of the isolects 230 , the dielectric coating of each should be very thin , for example from between 1 - 50 microns . since the coating is so thin , the isolects 230 can easily be damaged mechanically or undergo dielectric breakdown . this problem can be overcome by adding a protective feature to the isolect &# 39 ; s structure so as to provide desired protection from such damage . for example , the isolect 230 can be coated , for example , with a relatively loose net 340 that prevents access to the surface but has only a minor effect on the effective surface area of the isolect 230 ( i . e ., the capacity of the isolects 230 ( cross section presented in fig1 b ). the loose net 340 does not effect the capacity and ensures good contact with the skin , etc . the loose net 340 can be formed of a number of different materials ; however , in one exemplary embodiment , the net 340 is formed of nylon , polyester , cotton , etc . alternatively , a very thin conductive coating 350 can be applied to the dielectric portion ( insulating layer ) of the isolect 230 . one exemplary conductive coating is formed of a metal and more particularly of gold . the thickness of the coating 350 depends upon the particular application and also on the type of material used to form the coating 350 ; however , when gold is used , the coating has a thickness from about 0 . 1 micron to about 0 . 1 mm . furthermore , the rim illustrated in fig1 can also provide some mechanical protection . however , the capacity is not the only factor to be considered . the following two factors also influence how the isolects 230 are constructed . the dielectric strength of the internal insulating layer 310 and the dielectric losses that occur when it is subjected to the tc field , i . e ., the amount of heat generated . the dielectric strength of the internal insulation 310 determines at what field intensity the insulation will be “ shorted ” and cease to act as an intact insulation . typically , insulators , such as plastics , have dielectric strength values of about 100v per micron or more . as a high dielectric constant reduces the field within the internal insulator 310 , a combination of a high dielectric constant and a high dielectric strength gives a significant advantage . this can be achieved by using a single material that has the desired properties or it can be achieved by a double layer with the correct parameters and thickness . in addition , to further decreasing the possibility that the insulating layer 310 will fail , all sharp edges of the insulating layer 310 should be eliminated as by rounding the corners , etc ., as illustrated in fig1 d using conventional techniques . fig8 and 9 illustrate a second type of treatment using the isolects 230 , namely electric field generation by internal isolects 230 . a body to which the isolects 230 are implanted is generally indicated at 311 and includes a skin surface 313 and a tumor 315 . in this embodiment , the isolects 230 can have the shape of plates , wires or other shapes that can be inserted subcutaneously or a deeper location within the body 311 so as to generate an appropriate field at the target area ( tumor 315 ). it will also be appreciated that the mode of isolects application is not restricted to the above descriptions . in the case of tumors in internal organs , for example , liver , lung , etc ., the distance between each member of the pair of isolects 230 can be large . the pairs can even by positioned opposite sides of a torso 410 , as illustrated in fig1 . the arrangement of the isolects 230 in fig1 is particularly useful for treating a tumor 415 associated with lung cancer or gastro - intestinal tumors . in this embodiment , the electric fields ( tc fields ) spread in a wide fraction of the body . in order to avoid overheating of the treated tissues , a selection of materials and field parameters is needed . the isolects insulating material should have minimal dielectric losses at the frequency ranges to be used during the treatment process . this factor can be taken into consideration when choosing the particular frequencies for the treatment . the direct heating of the tissues will most likely be dominated by the heating due to current flow ( given by the i * r product ). in addition , the isolect ( insulated electrode ) 230 and its surroundings should be made of materials that facilitate heat losses and its general structure should also facilitate head losses , i . e ., minimal structures that block heat dissipation to the surroundings ( air ) as well as high heat conductivity . using larger electrodes also minimizes the local sensation of heating , since it spreads the energy that is being transferred into the patient over a larger surface area . preferably , the heating is minimized to the point where the patient &# 39 ; s skin temperature never exceeds about 39 ° c . another way to reduce heating is to apply the field to the tissue being treated intermittently , by applying a field with a duty cycle between about 20 % and about 50 % instead of using a continuous field . for example , to achieve a duty cycle of 33 %, the field would be repetitively switched on for one second , then switched off for two seconds . preliminary experiments have shown that the efficacy of treatment using a field with a 33 % duty cycle is roughly the same as for a field with a duty cycle of 100 %. in alternative embodiments , the field could be switched on for one hour then switched off for one hour to achieve a duty cycle of 50 %. of course , switching at a rate of once per hour would not help minimize short - term heating . on the other hand , it could provide the patient with a welcome break from treatment . the effectiveness of the treatment can be enhanced by an arrangement of isolects 230 that focuses the field at the desired target while leaving other sensitive areas in low field density ( i . e ., protected areas ). the proper placement of the isolects 230 over the body can be maintained using any number of different techniques , including using a suitable piece of clothing that keeps the isolects at the appropriate positions . fig1 illustrates such an arrangement in which an area labeled as “ p ” represents a protected area . the lines of field force do not penetrate this protected area and the field there is much smaller than near the isolects 230 where target areas can be located and treated well . in contrast , the field intensity near the four poles is very high . the following example serves to illustrate an exemplary application of the present apparatus and application of tc fields ; however , this example is not limiting and does not limit the scope of the present invention in any way . to demonstrate the effectiveness of electric fields having the above described properties ( e . g ., frequencies between 50 khz and 500 khz ) in destroying tumor cells , the electric fields were applied to treat mice with malignant melanoma tumors . two pairs of isolects 230 were positioned over a corresponding pair of malignant melanomas . only one pair was connected to the generator 210 and 200 khz alternating electric fields ( tc fields ) were applied to the tumor for a period of 6 days . one melanoma tumor was not treated so as to permit a comparison between the treated tumor and the non - treated tumor . after treatment for 6 days , the pigmented melanoma tumor remained clearly visible in the non - treated side of the mouse , while , in contrast , no tumor is seen on the treated side of the mouse . the only areas that were visible discernable on the skin were the marks that represented the points of insertion of the isolects 230 . the fact that the tumor was eliminated at the treated side was further demonstrated by cutting and inversing the skin so that its inside face was exposed . such a procedure indicated that the tumor has been substantially , if not completely , eliminated on the treated side of the mouse . the success of the treatment was also further verified by histopathological examination . the present inventor has thus uncovered that electric fields having particular properties can be used to destroy dividing cells or tumors when the electric fields are applied to using an electronic device . more specifically , these electric fields fall into a special intermediate category , namely bio - effective fields that have no meaningful stimulatory and no thermal effects , and therefore overcome the disadvantages that were associated with the application of conventional electric fields to a body . it will also be appreciated that the present apparatus can further include a device for rotating the tc field relative to the living tissue . for example and according to one embodiment , the alternating electric potential applies to the tissue being treated is rotated relative to the tissue using conventional devices , such as a mechanical device that upon activation , rotates various components of the present system . moreover and according to yet another embodiment , the tc fields are applied to different pairs of the insulated electrodes 230 in a consecutive manner . in other words , the generator 210 and the control system thereof can be arranged so that signals are sent at periodic intervals to select pairs of insulated electrodes 230 , thereby causing the generation of the tc fields of different directions by these insulated electrodes 230 . because the signals are sent at select times from the generator to the insulated electrodes 230 , the tc fields of changing directions are generated consecutively by different insulated electrodes 230 . this arrangement has a number of advantages and is provided in view of the fact that the tc fields have maximal effect when they are parallel to the axis of cell division . since the orientation of cell division is in most cases random , only a fraction of the dividing cells are affected by any given field . thus , using fields of two or more orientations increases the effectiveness since it increases the chances that more dividing cells are affected by a given tc field . in vitro experiments have shown that the electric field has the maximum killing effect when the lines of force of the field are oriented generally parallel to the long axis of the hourglass - shaped cell during mitosis ( as shown in fig3 a - 3c ). in one experiment , a much higher proportion of the damaged cells had their axis of division oriented along the field : 56 % of the cells oriented at or near 0 ° with respect to the field were damaged , versus an average of 15 % of cells damaged for cells with their long axis oriented at more than 22 ° with respect to the field . the inventor has recognized that applying the field in different directions sequentially will increase the overall killing power , because the field orientation that is most effectively in killing dividing cells will be applied to a larger population of the dividing cells . a number of examples for applying the field in different directions are discussed below . fig2 a , 27 b , and 27 c show a set of 6 electrodes e 1 - e 6 , and how the direction of the field through the target tissue 1510 can be changed by applying the ac signal from the generator 1 ( shown in fig1 ) across different pairs of electrodes . for example , if the ac signal is applied across electrodes e 1 and e 4 , the field lines f would be vertical ( as shown in fig2 a ), and if the signal is applied across electrodes e 2 and e 5 , or across electrodes e 3 and e 6 , the field lines f would be diagonal ( as shown in fig2 b and 27c , respectively ). additional field directions can be obtained by applying the ac signal across other pairs of electrodes . for example , a roughly horizontal field could be obtained by applying the signal across electrodes e 2 and e 6 . in one embodiment , the ac signal is applied between the various pairs of electrodes sequentially . an example of this arrangement is to apply the ac signal across electrodes e 1 and e 4 for one second , then apply the ac signal across electrodes e 2 and e 5 for one second , and then apply the ac signal across electrodes e 3 and e 6 for one second . this three - part sequence is then repeated for the desired period of treatment . because the efficacy in cell - destruction is strongly dependant on the cell &# 39 ; s orientation , cycling the field between the different directions increases the chance that the field will be oriented in a direction that favors cell destruction at least part of the time . of course , the 6 electrode configuration shown in fig2 a - c is just one of many possible arrangement of multiple electrodes , and many other configurations of three or more electrodes could be used based on the same principles . application of the field in different directions sequentially is not limited to two dimensional embodiments , and fig2 shows how the sequential application of signals across different sets of electrodes can be extended to three dimensions . a first array of electrodes a 1 - a 9 is arranged around body part 1500 , and a last array of electrodes n 1 - n 9 is arranged around the body part 1500 a distance w away from the first array . additional arrays of electrodes may optionally be added between the first array and the last array , but these additional arrays are not illustrated for clarity ( so as not to obscure the electrodes a 5 - a 9 and b 5 - b 8 on the back of the body part 1500 ). as in the fig2 embodiment , the direction of the field through the target tissue can be changed by applying the ac signal from the generator 1 ( shown in fig1 ) across different pairs of electrodes . for example , applying the ac signal between electrodes a 2 and a 7 would result in a field in a front - to - back direction between those two electrodes , and applying the ac signal between electrodes a 5 and a 9 would result in a roughly vertical field between those two electrodes . similarly , applying the ac signal across electrodes a 2 and n 7 would generate diagonal field lines in one direction through the body part 1500 , and applying the ac signal across electrodes a 2 and b 7 would generate diagonal field lines in another direction through the body part . using a three - dimensional array of electrodes also makes it possible to energize multiple pairs of electrodes simultaneously to induce fields in the desired directions . for example , if suitable switching is provided so that electrodes a 2 through n 2 are all connected to one terminal of the generator , and so that electrodes a 7 through n 7 are all connected to the other terminal of the generator , the resulting field would be a sheet that extends in a front - to - back direction for the entire width w . after the front - to - back field is maintained for a suitable duration ( e . g ., one second ), the switching system ( not shown ) is reconfigured to connect electrodes a 3 through n 3 to one terminal of the generator , and electrodes a 8 through n 8 to the other terminal of the generator . this results in a sheet - shaped field that is rotated about the z axis by about 40 ° with respect to the initial field direction . after the field is maintained in this direction for a suitable duration ( e . g ., one second ), the next set of electrodes is activated to rotate the field an additional 40 ° to its next position . this continues until the field returns to its initial position , at which point the whole process is repeated . optionally , the rotating sheet - shaped field may be added ( sequentially in time ) to the diagonal fields described above , to better target cells that are oriented along those diagonal axes . because the electric field is a vector , the signals may optionally be applied to combinations of electrodes simultaneously in order to form a desired resultant vector . for example , a field that is rotated about the x axis by 20 ° with respect to the initial position can be obtained by switching electrodes a 2 through n 2 and a 3 through n 3 all to one terminal of the generator , and switching electrodes a 7 through n 7 and a 8 through n 8 all to the other terminal of the generator . applying the signals to other combinations of electrodes will result in fields in other directions , as will be appreciated by persons skilled in the relevant arts . if appropriate computer control of the voltages is implemented , the field &# 39 ; s direction can even be swept through space in a continuous ( i . e ., smooth ) manner , as opposed to the stepwise manner described above . fig2 a and 29b depict the results of in vitro experiments that show how the killing power of the applied field against dividing cells is a function of the field strength . in the fig2 a experiment , b16f1 melanoma cells were subjected to a 100 khz ac field at different field strengths , for a period of 24 hours at each strength . in the fig2 b experiment , f - 98 glioma cells were subjected to a 200 khz ac field at different field strengths , for a period of 24 hours at each strength . in both of these figures , the strength of the field ( ef ) is measured in volts per cm . the magnitude of the killing effect is expressed in terms of ter , which is which is the ratio of the decrease in the growth rate of treated cells ( gr t ) compared with the growth rate of control cells ( gr c ). the experimental results show that the inhibitory effect of the applied field on proliferation increases with intensity in both the melanoma and the glioma cells . complete proliferation arrest ( ter = 1 ) is seen at 1 . 35 and 2 . 25 v / cm in melanoma and glioma cells , respectively . fig3 a and 30b depict the results of in vitro experiments that show how the killing power of the applied field is a function of the frequency of the field . in the experiments , b16f1 melanoma cells ( fig3 a ) and f - 98 glioma cells ( fig3 b ) were subjected to fields with different frequencies , for a period of 24 hours at each frequency . fig3 a and 30b show the change in the growth rate , normalized to the field intensity ( ter / ef ). data are shown as mean + se . in fig3 a , a window effect is seen with maximal inhibition at 120 khz in melanoma cells . in fig3 b , two peaks are seen at 170 and 250 khz . thus , if only one frequency is available during an entire course of treatment , a field with a frequency of about 120 khz would be appropriate for destroying melanoma cells , and a field with a frequency on the order of 200 khz would be appropriate for destroying glioma cells . not all the cells of any given type will have the exact same size . instead , the cells will have a distribution of sizes , with some cells being smaller and some cells being larger . it is believed that the best frequency for damaging a particular cell is related to the physical characteristics ( e . g ., the size ) of that particular cell . thus , to best damage a population of cells with a distribution of sizes , it can be advantageous to apply a distribution of different frequencies to the population , where the selection of frequencies is optimized based on the expected size distribution of the target cells . for example , the data on fig3 b indicates that using two frequencies of 170 khz and 250 khz to destroy a population of glioma cells would be more effective than using a single frequency of 200 khz . note that the optimal field strengths and frequencies discussed herein were obtained based on in vitro experiments , and that the corresponding parameters for in vivo applications may be obtained by performing similar experiments in vivo . it is possible that relevant characteristics of the cell itself ( such as size and / or shape ) or interactions with the cell &# 39 ; s surroundings may result in a different set of optimal frequencies and / or field strengths for in vivo applications . when more than one frequency is used , the various frequencies may be applied sequentially in time . for example , in the case of glioma , field frequencies of 100 , 150 , 170 , 200 , 250 , and 300 khz may be applied during the first , second , third , fourth , fifth , and sixth minutes of treatment , respectively . that cycle of frequencies would then repeat during each successive six minutes of treatment . alternatively , the frequency of the field may be swept in a stepless manner from 100 to 300 khz . optionally , this frequency cycling may be combined with the directional cycling described above . fig3 a is an example of such a combination using three directions ( d 1 , d 2 , and d 3 ) and three frequencies ( f 1 , f 2 , and f 3 ). of course , the same scheme can be extended to any other number of directions and / or frequencies . fig3 b is an example of such a combination using three directions ( d 1 , d 2 , and d 3 ), sweeping the frequency from 100 khz to 300 khz . note that the break in the time axis between t 1 and t 2 provides the needed time for the sweeping frequency to rise to just under 300 khz . the frequency sweeping ( or stepping ) may be synchronized with directional changes , as shown in fig3 a . alternatively , the frequency sweeping ( or stepping ) may be asynchronous with respect to the directional changes , as shown in fig3 b . in an alternative embodiment , a signal that contains two or more frequencies components simultaneously ( e . g ., 170 khz and 250 khz ) is applied to the electrodes to treat a populations of cells that have a distribution of sizes . the various signals will add by superposition to create a field that includes all of the applied frequency components . turning now to fig1 in which an article of clothing 500 according to one exemplary embodiment is illustrated . more specifically , the article of clothing 500 is in the form of a hat or cap or other type of clothing designed for placement on a head of a person . for purposes of illustration , a head 502 is shown with the hat 500 being placed thereon and against a skin surface 504 of the head 502 . an intra - cranial tumor or the like 510 is shown as being formed within the head 502 underneath the skin surface 504 thereof . the hat 500 is therefore intended for placement on the head 502 of a person who has a tumor 510 or the like . unlike the various embodiments illustrated in fig1 - 13 where the insulated electrodes 230 are arranged in a more or less planar arrangement since they are placed either on a skin surface or embedded within the body underneath it , the insulated electrodes 230 in this embodiment are specifically contoured and arranged for a specific application . the treatment of intra - cranial tumors or other lesions or the like typically requires a treatment that is of a relatively long duration , e . g ., days to weeks , and therefore , it is desirable to provide as much comfort as possible to the patient . the hat 500 is specifically designed to provide comfort during the lengthy treatment process while not jeopardizing the effectiveness of the treatment . according to one exemplary embodiment , the hat 500 includes a predetermined number of insulated electrodes 230 that are preferably positioned so as to produce the optimal tc fields at the location of the tumor 510 . the lines of force of the tc field are generally indicated at 520 . as can be seen in fig1 , the tumor 510 is positioned within these lines of force 520 . as will be described in greater detail hereinafter , the insulated electrodes 230 are positioned within the hat 500 such that a portion or surface thereof is free to contact the skin surface 504 of the head 502 . in other words , when the patient wears the hat 500 , the insulated electrodes 230 are placed in contact with the skin surface 504 of the head 502 in positions that are selected so that the tc fields generated thereby are focused at the tumor 510 while leaving surrounding areas in low density . typically , hair on the head 502 is shaved in selected areas to permit better contact between the insulated electrodes 230 and the skin surface 504 ; however , this is not critical . the hat 500 preferably includes a mechanism 530 that applies a force to the insulated electrodes 230 so that they are pressed against the skin surface 502 . for example , the mechanism 530 can be of a biasing type that applies a biasing force to the insulated electrodes 230 to cause the insulated electrodes 230 to be directed outwardly away from the hat 500 . thus , when the patient places the hat 500 on his / her head 502 , the insulated electrodes 230 are pressed against the skin surface 504 by the mechanism 530 . the mechanism 530 can slightly recoil to provide a comfortable fit between the insulated electrodes 230 and the head 502 . in one exemplary embodiment , the mechanism 530 is a spring based device that is disposed within the hat 500 and has one section that is coupled to and applies a force against the insulated electrodes 230 . as with the prior embodiments , the insulated electrodes 230 are coupled to the generator 210 by means of conductors 220 . the generator 210 can be either disposed within the hat 500 itself so as to provide a compact , self - sufficient , independent system or the generator 210 can be disposed external to the hat 500 with the conductors 220 exiting the hat 500 through openings or the like and then running to the generator 210 . when the generator 210 is disposed external to the hat 500 , it will be appreciated that the generator 210 can be located in any number of different locations , some of which are in close proximity to the hat 500 itself , while others can be further away from the hat 500 . for example , the generator 210 can be disposed within a carrying bag or the like ( e . g ., a bag that extends around the patient &# 39 ; s waist ) which is worn by the patient or it can be strapped to an extremity or around the torso of the patient . the generator 210 can also be disposed in a protective case that is secured to or carried by another article of clothing that is worn by the patient . for example , the protective case can be inserted into a pocket of a sweater , etc . fig1 illustrates an embodiment where the generator 210 is incorporated directly into the hat 500 . turning now to fig1 and 16 , in one exemplary embodiment , a number of insulated electrodes 230 along with the mechanism 530 are preferably formed as an independent unit , generally indicated at 540 , that can be inserted into the hat 500 and electrically connected to the generator ( not shown ) via the conductors ( not shown ). by providing these members in the form of an independent unit , the patient can easily insert and / or remove the units 540 from the hat 500 when they may need cleaning , servicing and / or replacement . in this embodiment , the hat 500 is constructed to include select areas 550 that are formed in the hat 500 to receive and hold the units 540 . for example and as illustrated in fig1 , each area 550 is in the form of an opening ( pore ) that is formed within the hat 500 . the unit 540 has a body 542 and includes the mechanism 530 and one or more insulated electrodes 230 . the mechanism 530 is arranged within the unit 540 so that a portion thereof ( e . g ., one end thereof ) is in contact with a face of each insulated electrode 230 such that the mechanism 530 applies a biasing force against the face of the insulated electrode 230 . once the unit 540 is received within the opening 550 , it can be securely retained therein using any number of conventional techniques , including the use of an adhesive material or by using mechanical means . for example , the hat 500 can include pivotable clip members that pivot between an open position in which the opening 550 is free and a closed position in which the pivotable clip members engage portions ( e . g ., peripheral edges ) of the insulated electrodes to retain and hold the insulated electrodes 230 in place . to remove the insulated electrodes 230 , the pivotable clip members are moved to the open position . in the embodiment illustrated in fig1 , the insulated electrodes 230 are retained within the openings 550 by an adhesive element 560 which in one embodiment is a two sided self - adhesive rim member that extends around the periphery of the insulated electrode 230 . in other words , a protective cover of one side of the adhesive rim 560 is removed and it is applied around the periphery of the exposed face of the insulated electrode 230 , thereby securely attaching the adhesive rim 560 to the hat 500 and then the other side of the adhesive rim 560 is removed for application to the skin surface 504 in desired locations for positioning and securing the insulated electrode 230 to the head 502 with the tumor being positioned relative thereto for optimization of the tc fields . since one side of the adhesive rim 560 is in contact with and secured to the skin surface 540 , this is why it is desirable for the head 502 to be shaved so that the adhesive rim 560 can be placed flushly against the skin surface 540 . the adhesive rim 560 is designed to securely attach the unit 540 within the opening 550 in a manner that permits the unit 540 to be easily removed from the hat 500 when necessary and then replaced with another unit 540 or with the same unit 540 . as previously mentioned , the unit 540 includes the biasing mechanism 530 for pressing the insulated electrode 230 against the skin surface 504 when the hat 500 is worn . the unit 540 can be constructed so that side opposite the insulated electrode 230 is a support surface formed of a rigid material , such as plastic , so that the biasing mechanism 530 ( e . g ., a spring ) can be compressed therewith under the application of force and when the spring 530 is in a relaxed state , the spring 530 remains in contact with the support surface and the applies a biasing force at its other end against the insulated electrode 230 . the biasing mechanism 530 ( e . g ., spring ) preferably has a contour corresponding to the skin surface 504 so that the insulated electrode 230 has a force applied thereto to permit the insulated electrode 230 to have a contour complementary to the skin surface 504 , thereby permitting the two to seat flushly against one another . while the mechanism 530 can be a spring , there are a number of other embodiments that can be used instead of a spring . for example , the mechanism 530 can be in the form of an elastic material , such as a foam rubber , a foam plastic , or a layer containing air bubbles , etc . the unit 540 has an electric connector 570 that can be hooked up to a corresponding electric connector , such as a conductor 220 , that is disposed within the hat 500 . the conductor 220 connects at one end to the unit 540 and at the other end is connected to the generator 210 . the generator 210 can be incorporated directly into the hat 500 or the generator 210 can be positioned separately ( remotely ) on the patient or on a bedside support , etc . as previously discussed , a coupling agent , such as a conductive gel , is preferably used to ensure that an effective conductive environment is provided between the insulated electrode 230 and the skin surface 504 . suitable gel materials have been disclosed hereinbefore in the discussion of earlier embodiments . the coupling agent is disposed on the insulated electrode 230 and preferably , a uniform layer of the agent is provided along the surface of the electrode 230 . one of the reasons that the units 540 need replacement at periodic times is that the coupling agent needs to be replaced and / or replenished . in other words , after a predetermined time period or after a number of uses , the patient removes the units 540 so that the coupling agent can be applied again to the electrode 230 . fig1 and 18 illustrate another article of clothing which has the insulated electrodes 230 incorporated as part thereof . more specifically , a bra or the like 700 is illustrated and includes a body that is formed of a traditional bra material , generally indicated at 705 , to provide shape , support and comfort to the wearer . the bra 700 also includes a fabric support layer 710 on one side thereof . the support layer 710 is preferably formed of a suitable fabric material that is constructed to provide necessary and desired support to the bra 700 . similar to the other embodiments , the bra 700 includes one or more insulated electrodes 230 disposed within the bra material 705 . the one or more insulated electrodes are disposed along an inner surface of the bra 700 opposite the support 710 and are intended to be placed proximate to a tumor or the like that is located within one breast or in the immediately surrounding area . as with the previous embodiment , the insulated electrodes 230 in this embodiment are specifically constructed and configured for application to a breast or the immediate area . thus , the insulated electrodes 230 used in this application do not have a planar surface construction but rather have an arcuate shape that is complementary to the general curvature found in a typical breast . a lining 720 is disposed across the insulated electrodes 230 so as to assist in retaining the insulated electrodes in their desired locations along the inner surface for placement against the breast itself . the lining 720 can be formed of any number of thin materials that are comfortable to wear against one &# 39 ; s skin and in one exemplary embodiment , the lining 720 is formed of a fabric material . the bra 700 also preferably includes a biasing mechanism 800 as in some of the earlier embodiments . the biasing mechanism 800 is disposed within the bra material 705 and extends from the support 710 to the insulated electrode 230 and applies a biasing force to the insulated electrode 230 so that the electrode 230 is pressed against the breast . this ensures that the insulated electrode 230 remains in contact with the skin surface as opposed to lifting away from the skin surface , thereby creating a gap that results in a less effective treatment since the gap diminishes the efficiency of the tc fields . the biasing mechanism 800 can be in the form of a spring arrangement or it can be an elastic material that applies the desired biasing force to the insulated electrodes 230 so as to press the insulated electrodes 230 into the breast . in the relaxed position , the biasing mechanism 800 applies a force against the insulated electrodes 230 and when the patient places the bra 700 on their body , the insulated electrodes 230 are placed against the breast which itself applies a force that counters the biasing force , thereby resulting in the insulated electrodes 230 being pressed against the patient &# 39 ; s breast . in the exemplary embodiment that is illustrated , the biasing mechanism 800 is in the form of springs that are disposed within the bra material 705 . a conductive gel 810 can be provided on the insulated electrode 230 between the electrode and the lining 720 . the conductive gel layer 810 is formed of materials that have been previously described herein for performing the functions described above . an electric connector 820 is provided as part of the insulated electrode 230 and electrically connects to the conductor 220 at one end thereof , with the other end of the conductor 220 being electrically connected to the generator 210 . in this embodiment , the conductor 220 runs within the bra material 705 to a location where an opening is formed in the bra 700 . the conductor 220 extends through this opening and is routed to the generator 210 , which in this embodiment is disposed in a location remote from the bra 700 . it will also be appreciated that the generator 210 can be disposed within the bra 700 itself in another embodiment . for example , the bra 700 can have a compartment formed therein which is configured to receive and hold the generator 210 in place as the patient wears the bra 700 . in this arrangement , the compartment can be covered with a releasable strap that can open and close to permit the generator 210 to be inserted therein or removed therefrom . the strap can be formed of the same material that is used to construct the bra 700 or it can be formed of some other type of material . the strap can be releasably attached to the surrounding bra body by fastening means , such as a hook and loop material , thereby permitting the patient to easily open the compartment by separating the hook and loop elements to gain access to the compartment for either inserting or removing the generator 210 . the generator 210 also has a connector 211 for electrical connection to the conductor 220 and this permits the generator 210 to be electrically connected to the insulated electrodes 230 . as with the other embodiments , the insulated electrodes 230 are arranged in the bra 700 to focus the electric field ( tc fields ) on the desired target ( e . g ., a tumor ). it will be appreciated that the location of the insulated electrodes 230 within the bra 700 will vary depending upon the location of the tumor . in other words , after the tumor has been located , the physician will then devise an arrangement of insulated electrodes 230 and the bra 700 is constructed in view of this arrangement so as to optimize the effects of the tc fields on the target area ( tumor ). the number and position of the insulated electrodes 230 will therefore depend upon the precise location of the tumor or other target area that is being treated . because the location of the insulated electrodes 230 on the bra 700 can vary depending upon the precise application , the exact size and shape of the insulated electrodes 230 can likewise vary . for example , if the insulated electrodes 230 are placed on the bottom section of the bra 700 as opposed to a more central location , the insulated electrodes 230 will have different shapes since the shape of the breast ( as well as the bra ) differs in these areas . fig1 illustrates yet another embodiment in which the insulated electrodes 230 are in the form of internal electrodes that are incorporated into in the form of a probe or catheter 600 that is configured to enter the body through a natural pathway , such as the urethra , vagina , etc . in this embodiment , the insulated electrodes 230 are disposed on an outer surface of the probe 600 and along a length thereof . the conductors 220 are electrically connected to the electrodes 230 and run within the body of the probe 600 to the generator 210 which can be disposed within the probe body or the generator 210 can be disposed independent of the probe 600 in a remote location , such as on the patient or at some other location close to the patient . alternatively , the probe 600 can be configured to penetrate the skin surface or other tissues to reach an internal target that lies within the body . for example , the probe 600 can penetrate the skin surface and then be positioned adjacent to or proximate to a tumor that is located within the body . in these embodiments , the probe 600 is inserted through the natural pathway and then is positioned in a desired location so that the insulated electrodes 230 are disposed near the target area ( i . e ., the tumor ). the generator 210 is then activated to cause the insulated electrodes 230 to generate the tc fields which are applied to the tumor for a predetermined length of time . it will be appreciated that the illustrated probe 600 is merely exemplary in nature and that the probe 600 can have other shapes and configurations so long as they can perform the intended function . preferably , the conductors ( e . g ., wires ) leading from the insulated electrodes 230 to the generator 210 are twisted or shielded so as not to generate a field along the shaft . it will further be appreciated that the probes can contain only one insulated electrode while the other can be positioned on the body surface . this external electrode should be larger or consist of numerous electrodes so as to result in low lines of force - current density so as not to affect the untreated areas . in fact , the placing of electrodes should be designed to minimize the field at potentially sensitive areas . optionally , the external electrodes may be held against the skin surface by a vacuum force ( e . g ., suction ). fig2 illustrates yet another embodiment in which a high standing collar member 900 ( or necklace type structure ) can be used to treat thyroid , parathyroid , laryngeal lesions , etc . fig2 illustrates the collar member 900 in an unwrapped , substantially flat condition . in this embodiment , the insulated electrodes 230 are incorporated into a body 910 of the collar member 900 and are configured for placement against a neck area of the wearer . the insulated electrodes 230 are coupled to the generator 210 according to any of the manner described hereinbefore and it will be appreciated that the generator 210 can be disposed within the body 910 or it can be disposed in a location external to the body 910 . the collar body 910 can be formed of any number of materials that are traditionally used to form collars 900 that are disposed around a person &# 39 ; s neck . as such , the collar 900 preferably includes a means 920 for adjusting the collar 900 relative to the neck . for example , complementary fasteners ( hook and loop fasteners , buttons , etc .) can be disposed on ends of the collar 900 to permit adjustment of the collar diameter . thus , the construction of the present devices are particularly well suited for applications where the devices are incorporated into articles of clothing to permit the patient to easily wear a traditional article of clothing while at the same time the patient undergoes treatment . in other words , an extra level of comfort can be provided to the patient and the effectiveness of the treatment can be increased by incorporating some or all of the device components into the article of clothing . the precise article of clothing that the components are incorporated into will obviously vary depending upon the target area of the living tissue where tumor , lesion or the like exists . for example , if the target area is in the testicle area of a male patient , then an article of clothing in the form of a sock - like structure or wrap can be provided and is configured to be worn around the testicle area of the patient in such a manner that the insulated electrodes thereof are positioned relative to the tumor such that the tc fields are directed at the target tissue . the precise nature or form of the article of clothing can vary greatly since the device components can be incorporated into most types of articles of clothing and therefore , can be used to treat any number of different areas of the patient &# 39 ; s body where a condition may be present . now turning to fig2 - 22 in which another aspect of the present device is shown . in fig2 , a body 1000 , such as any number of parts of a human or animal body , is illustrated . as in the previous embodiments , two or more insulated electrodes 230 are disposed in proximity to the body 1000 for treatment of a tumor or the like ( not shown ) using tc fields , as has been previously described in great detail in the above discussion of other embodiments . the insulated electrode 230 has a conductive component and has external insulation 260 that surrounds the conductive component thereof . each insulated electrode 230 is preferably connected to a generator ( not shown ) by the lead 220 . between each insulated electrode 220 and the body 1000 , a conductive filler material ( e . g ., conductive gel member 270 ) is disposed . the insulated electrodes 230 are spaced apart from one another and when the generator is actuated , the insulated electrodes 230 generate the tc fields that have been previously described in great detail . the lines of the electric field ( tc field ) are generally illustrated at 1010 . as shown , the electric field lines 1010 extend between the insulated electrodes 230 and through the conductive gel member 270 . over time or as a result of some type of event , the external insulation 260 of the insulated electrode 230 can begin to breakdown at any given location thereof . for purpose of illustration only , fig2 illustrates that the external insulation 260 of one of the insulated electrodes 230 has experienced a breakdown 1020 at a face thereof which is adjacent the conductive gel member 270 . it will be appreciated that the breakdown 1020 of the external insulation 260 results in the formation of a strong current flow - current density at this point ( i . e ., at the breakdown 1020 ). the increased current density is depicted by the increased number of electric field lines 1010 and the relative positioning and distance between adjacent electric field lines 1010 . one of the side effects of the occurrence of breakdown 1020 is that current exists at this point which will generate heat and may burn the tissues / skin which have a resistance . in fig2 , an overheated area 1030 is illustrated and is a region or area of the tissues / skin where an increased current density exits due to the breakdown 1020 in the external insulation 260 . a patient can experience discomfort and pain in this area 1030 due to the strong current that exists in the area and the increased heat and possible burning sensation that exist in area 1030 . fig2 illustrates yet another embodiment in which a further application of the insulated electrodes 230 is shown . in this embodiment , the conductive gel member 270 that is disposed between the insulated electrode 230 and the body 1000 includes a conductor 1100 that is floating in that the gel material forming the member 270 completely surrounds the conductor 1100 . in one exemplary embodiment , the conductor 1100 is a thin metal sheet plate that is disposed within the conductor 1100 . as will be appreciated , if a conductor , such as the plate 1100 , is placed in a homogeneous electric field , normal to the lines of the electric field , the conductor 1100 practically has no effect on the field ( except that the two opposing faces of the conductor 1100 are equipotential and the corresponding equipotentials are slightly shifted ). conversely , if the conductor 1100 is disposed parallel to the electric field , there is a significant distortion of the electric field . the area in the immediate proximity of the conductor 1100 is not equipotential , in contrast to the situation where there is no conductor 1100 present . when the conductor 1100 is disposed within the gel member 270 , the conductor 1100 will typically not effect the electric field ( tc field ) for the reasons discussed above , namely that the conductor 1100 is normal to the lines of the electric field . if there is a breakdown of the external insulation 260 of the insulated electrode 230 , there is a strong current flow - current density at the point of breakdown as previously discussed ; however , the presence of the conductor 1100 causes the current to spread throughout the conductor 1100 and then exit from the whole surface of the conductor 1100 so that the current reaches the body 1000 with a current density that is neither high nor low . thus , the current that reaches the skin will not cause discomfort to the patient even when there has been a breakdown in the insulation 260 of the insulated electrode 230 . it is important that the conductor 1100 is not grounded as this would cause it to abolish the electric field beyond it . thus , the conductor 1100 is “ floating ” within the gel member 270 . if the conductor 1100 is introduced into the body tissues 1000 and is not disposed parallel to the electric field , the conductor 100 will cause distortion of the electric field . the distortion can cause spreading of the lines of force ( low field density - intensity ) or concentration of the lines of field ( higher density ) of the electric field , according to the particular geometries of the insert and its surroundings , and thus , the conductor 1100 can exhibit , for example , a screening effect . thus , for example , if the conductor 1100 completely encircles an organ 1101 , the electric field in the organ itself will be zero since this type of arrangement is a faraday cage . however , because it is impractical for a conductor to be disposed completely around an organ , a conductive net or similar structure can be used to cover , completely or partially , the organ , thereby resulting in the electric field in the organ itself being zero or about zero . for example , a net can be made of a number of conductive wires that are arranged relative to one another to form the net or a set of wires can be arranged to substantially encircle or otherwise cover the organ 1101 . conversely , an organ 1103 to be treated ( the target organ ) is not covered with a member having a faraday cage effect but rather is disposed in the electric field 1010 ( tc fields ). fig2 illustrates an embodiment where the conductor 1100 is disposed within the body ( i . e ., under the skin ) and it is located near a target ( e . g ., a target organ ). by placing the conductor 1100 near the target , high field density ( of the tc fields ) is realized at the target . at the same time , another nearby organ can be protected by disposing the above described protective conductive net or the like around this nearby organ so as to protect this organ from the fields . by positioning the conductor 1100 in close proximity to the target , a high field density condition can be provided near or at the target . in other words , the conductor 1100 permits the tc fields to be focused at a particular area ( i . e ., a target ). it will also be appreciated that in the embodiment of fig2 , the gel members 260 can each include a conductor as described with reference to fig2 . in such an arrangement , the conductor in the gel member 260 protects the skin surface ( tissues ) from any side effects that may be realized if a breakdown in the insulation of the insulated electrode 230 occurs . at the same time , the conductor 1100 creates a high field density near the target . there are a number of different ways to tailor the field density of the electric field by constructing the electrodes differently and / or by strategically placing the electrodes relative to one another . for example , in fig2 , a first insulated electrode 1200 and a second insulated electrode 1210 are provided and are disposed about a body 1300 . each insulated electrode includes a conductor that is preferably surrounded by an insulating material , thus the term “ insulated electrode ”. between each of the first and second electrodes 1200 , 1210 and the body 1300 , the conductive gel member 270 is provided . electric field lines are generally indicated at 1220 for this type of arrangement . in this embodiment , the first insulated electrode 1200 has dimensions that are significantly greater than the dimensions of the second insulated electrode 1210 ( the conductive gel member for the second insulated electrode 1210 will likewise be smaller ). by varying the dimensions of the insulated electrodes , the pattern of the electric field lines 1220 is varied . more specifically , the electric field tapers inwardly toward the second insulated electrode 1210 due to the smaller dimensions of the second insulated electrode 1210 . an area of high field density , generally indicated at 1230 , forms near the interface between the gel member 270 associated with the second insulated electrode 1210 and the skin surface . the various components of the system are manipulated so that the tumor within the skin or on the skin is within this high field density so that the area to be treated ( the target ) is exposed to electric field lines of a higher field density . fig2 also illustrates a tapering tc field when a conductor 1400 ( e . g ., a conductive plate ) is disposed in each of the conductive gel members 270 . in this embodiment , the size of the gel members 270 and the size of the conductors 1400 are the same or about the same despite the differences in the sizes of the insulated electrodes 1200 , 1210 . the conductors 1400 again can be characterized as “ floating plates ” since each conductor 1400 is surrounded by the material that forms the gel member 270 . as shown in fig2 , the placement of one conductor 1400 near the insulated electrode 1210 that is smaller than the other insulated electrode 1200 and is also smaller than the conductor 1400 itself and the other insulated electrode 1200 is disposed at a distance therefrom , the one conductor 1400 causes a decrease in the field density in the tissues disposed between the one conductor 1400 and the other insulated electrode 1200 . the decrease in the field density is generally indicated at 1410 . at the same time , a very inhomogeneous tapering field , generally indicated at 1420 , changing from very low density to very high density is formed between the one conductor 1400 and the insulated electrode 1210 . one benefit of this exemplary configuration is that it permits the size of the insulated electrode to be reduced without causing an increase in the nearby field density . this can be important since electrodes that having very high dielectric constant insulation can be very expensive . some insulated electrodes , for example , can cost $ 500 . 00 or more ; and further , the price is sensitive to the particular area of treatment . thus , a reduction in the size of the insulated electrodes directly leads to a reduction in cost . as used herein , the term “ tumor ” refers to a malignant tissue comprising transformed cells that grow uncontrollably . tumors include leukemias , lymphomas , myelomas , plasmacytomas , and the like ; and solid tumors . examples of solid tumors that can be treated according to the invention include sarcomas and carcinomas such as , but not limited to : fibrosarcoma , myxosarcoma , liposarcoma , chondrosarcoma , osteogenic sarcoma , chordoma , angiosarcoma , endotheliosarcoma , lymphangiosarcoma , lymphangioendotheliosarcoma , synovioma , mesothelioma , ewing &# 39 ; s tumor , leiomyosarcoma , rhabdomyosarcoma , colon carcinoma , pancreatic cancer , breast cancer , ovarian cancer , prostate cancer , squamous cell carcinoma , basal cell carcinoma , adenocarcinoma , sweat gland carcinoma , sebaceous gland carcinoma , papillary carcinoma , papillary adenocarcinomas , cystadenocarcinoma , medullary carcinoma , bronchogenic carcinoma , renal cell carcinoma , hepatoma , bile duct carcinoma , choriocarcinoma , seminoma , embryonal carcinoma , wilms &# 39 ; tumor , cervical cancer , testicular tumor , lung carcinoma , small cell lung carcinoma , bladder carcinoma , epithelial carcinoma , glioma , astrocytoma , medulloblastoma , craniopharyngioma , ependymoma , pinealoma , hemangioblastoma , acoustic neuroma , oligodendroglioma , meningioma , melanoma , neuroblastoma , and retinoblastoma . because each of these tumors undergoes rapid growth , any one can be treated in accordance with the invention . the invention is particularly advantageous for treating brain tumors , which are difficult to treat with surgery and radiation , and often inaccessible to chemotherapy or gene therapies . in addition , the present invention is suitable for use in treating skin and breast tumors because of the ease of localized treatment provided by the present invention . in addition , the present invention can control uncontrolled growth associated with non - malignant or pre - malignant conditions , and other disorders involving inappropriate cell or tissue growth by application of an electric field in accordance with the invention to the tissue undergoing inappropriate growth . for example , it is contemplated that the invention is useful for the treatment of arteriovenous ( av ) malformations , particularly in intracranial sites . the invention may also be used to treat psoriasis , a dermatologic condition that is characterized by inflammation and vascular proliferation ; and benign prostatic hypertrophy , a condition associated with inflammation and possibly vascular proliferation . treatment of other hyperproliferative disorders is also contemplated . furthermore , undesirable fibroblast and endothelial cell proliferation associated with wound healing , leading to scar and keloid formation after surgery or injury , and restenosis after angioplasty or placement of coronary stents can be inhibited by application of an electric field in accordance with the present invention . the non - invasive nature of this invention makes it particularly desirable for these types of conditions , particularly to prevent development of internal scars and adhesions , or to inhibit restenosis of coronary , carotid , and other important arteries . in addition to treating tumors that have already been detected , the above - described embodiments may also be used prophylactically to prevent tumors from ever reaching a detectable size in the first place . for example , the bra embodiment described above in connection with fig1 and 18 may be worn by a woman for an 8 hour session every day for a week , with the week - long course of treatment being repeated every few months to kill any cells that have become cancerous and started to proliferate . this mode of usage is particularly appropriate for people who are at high risk for a particular type of cancer ( e . g ., women with a strong history of breast cancer in their families , or people who have survived a bout of cancer and are at risk of a relapse ). the course of prophylactic treatment may be tailored based on the type of cancer being targeted and / or to suit the convenience of the patient . for example , undergoing a four 16 hour sessions during the week of treatment may be more convenient for some patients than seven 8 hour session , and may be equally effective . thus , the present invention provides an effective , simple method of selectively destroying dividing cells , e . g ., tumor cells and parasitic organisms , while non - dividing cells or organisms are left affected by application of the method on living tissue containing both types of cells or organisms . thus , unlike many of the conventional methods , the present invention does not damage the normal cells or organisms . in addition , the present invention does not discriminate based upon cell type ( e . g ., cells having differing sizes ) and therefore may be used to treat any number of types of sizes having a wide spectrum of characteristics , including varying dimensions . while the invention has been particularly shown and described with reference to preferred embodiments thereof , it will be understood by those skilled in the art that various changes in form and details can be made without departing from the spirit and scope of the invention .
cells that are in the late anaphase or telophase stages of cell division are vulnerable to damage by ac electric fields that have specific frequency and field strength characteristics . the selective destruction of rapidly dividing cells can therefore be accomplished by imposing an ac electric field in a target region for extended periods of time . some of the cells that divide while the field is applied will be damaged , but the cells that do not divide will not be harmed . this selectively damages rapidly dividing cells like tumor cells , but does not harm normal cells that are not dividing . since the vulnerability of the dividing cells is strongly related to the alignment between the long axis of the dividing cells and the lines of force of the electric field , improved results are obtained when the field is sequentially imposed in different directions .