as shown in the exemplary drawings for the purposes of illustration , an embodiment of an integrated system made in accordance with the principals of the present invention , referred to generally by the reference numeral 10 , is provided for simplified setup and operation of an electrophysiology amplifying and switching system during an electrophysiology procedure . more specifically , as shown in fig1 the front panel of the system 10 of the present invention includes a standard twelve lead ecg input terminal 11 which allows attachment of any well known ecg lead cable through which leads extend from electrodes attached to the chest of a patient in a well known manner for transferring ecg signals into the system 10 . similarly , intracardiac input terminals 12 are positioned on the front panel , and include intervention / input terminals 13 which can be used as intracardiac input terminals or intervention ( stimulation ) terminals . the intervention / input terminals 13 are hard wired to corresponding intervention / output terminals 17 ( as shown in fig2 ). the intracardiac input terminals 12 are adapted to receive leads from the intracardiac catheters which have been placed within the patient &# 39 ; s heart to sense the electrical signals passing therethrough . the intervention / input terminals 13 are designed to pass electrical stimulation signals ( originating from an electrical stimulator , not shown ) through the system 10 into the intracardiac catheters . the intervention / input terminals 13 are also designed to receive electrical signals ( originating in the heart ) from the intracardiac catheters in the same manner as the input terminals 12 . also included on the front panel are four pressure channel input terminals 14 which are designed for receipt of pressure sensor leads which have been attached to pressure sensors positioned at desired points on or within the patient &# 39 ; s body from which blood pressure information is desired . the dominant feature of the front panel of the system 10 is an operator interactive &# 34 ; touch &# 34 ; display 15 which is programmed by an onboard microprocessor 27 ( see fig6 ) to operate as a labelling area for the input terminals 11 , 12 , 13 , 14 , and the auxiliary input terminals 16 ( see fig2 ). the display 15 is also configured by the microprocessor 27 to define touch areas thereon as &# 34 ; soft keys &# 34 ; for use in initiating setup and operation commands as will be explained below , and also to display messages related to setup and operation of the system 10 . further , the display 15 is driven by a microprocessor 27 ( see fig7 ) to display assignments of the output channels 17 ( see fig2 ) and the labels , poles , gains , filter , and clamp settings through which each channel of entering data must pass when entered into the system 10 . the display 15 is designed to simplify setup and operation of the system 10 as will be explained momentarily . software preprogrammed into the onboard microprocessor 27 , such as by a rom , is directly responsible for the operation of the display 15 and input &# 34 ; soft keys &# 34 ; thereof . the display 15 is preferably a single 640 × 480 pixel pressure responsive display commonly referred to as a &# 34 ; touch screen &# 34 ;. although the present invention is not limited to the following , it is intended that the preferred embodiment of the present invention include the ability to receive twelve leads of ecg signals as input into the input terminal 11 . further , it is preferred that sixteen or thirty - two intracardiac input terminals 12 and eight intervention or intracardiac input terminals 13 be included in the system 10 . further , it is preferred that four pressure channel input terminals 14 be included with four auxiliary input channels 16 ( see fig2 ). the display 15 is used to specify and setup labels associated with each input terminal 12 , 13 , 14 , and 16 . to describe the labeling process it is first necessary to define some terms . a label 28 ( see fig4 ( b )) is the alphanumeric assigned to an input terminal to identify the catheter or sensor and the lead attached thereto . a catheter group , or simply &# 34 ; group &# 34 ; is a group of labels 28 that have the same prefix letters but different numbers . an example of a catheter group would be : &# 34 ; rva1 , rva2 , rva3 , rva4 &# 34 ; ( see fig4 ( c )), where the prefix letters &# 34 ; rva &# 34 ;, are common to each label 28 . any intracardiac or intervention labels 28 with the same base letters , regardless of their location on the display 15 , belong to the same group . a field 29 ( see fig4 ( a )) is a label location on the display 15 . there are two basic screens which can be called to the display 15 by the operator to assist in arranging the setup configuration of the system 10 . the first is the &# 34 ; catheter placement screen &# 34 ; 43 as shown in fig4 ( a ). the second is the &# 34 ; signals screen &# 34 ; as shown in fig5 ( a ). once the electrophysiology catheters have been placed in the patient , setup of the system 10 only requires the steps of : 1 ) specifying the signal inputs by producing a label 28 for each one , and then 2 ) specifying the desired signal output parameters . the first step is accomplished through the use of the catheter placement screen 43 , the second step is accomplished with the assistance of the signals screen 44 . referring to fig4 ( a ), the catheter placement screen 43 is programmed to preferably form the following soft keys : a twelve lead key 31 , a restore key 34 , a catheter key 32 , a signals key 33 , a mark key 35 , a channels key 36 , a calibrate key 37 , a chart key 38 , an intervention key 39 , an intracardiac key 40 , an auxiliary key 41 , and a pressure key 42 . briefly , to specify an input , the operator need only touch the key describing the input type , such as the intracardiac key 40 , the auxiliary key 41 , or the pressure key 42 , then use the resulting directory 30 ( see fig4 ( b ) to assign each label 28 to the field 29 that corresponds to the correct lead input connector 12 , 13 , 14 , or 16 . to quickly assign a multiple of fields 29 to a single label 28 , the operator touches the desired label 28 in the directory 30 , then touches as many fields 29 as desired . operating in this fashion to assign labels is hereafter referred to as the batch edit mode of operation . to make a single assignment of a label 28 to a field 29 , the operator first touches the desired field 29 and then touches the desired label 28 in the directory 30 . assigning a single label 28 at a time in this fashion is referred to hereafter as the single edit mode of operation . in batch edit mode , when the user chooses which field or fields 29 to apply the label 28 to by touching in succession each desired field 29 , the plural leads of the catheter or sensor will be automatically numbered appropriately . the operator can continue in this way indefinitely . if a field 29 was empty prior to this operation , the field 29 will be filled in appropriately . however , if there was already a label 28 present in the field 29 , all associated labels 28 will be updated with the new label 28 and each lead connector will be numbered appropriately . when the operator is finished with one label 28 , another label 28 may be chosen . when all desired labels 28 are assigned , the operator chooses &# 34 ; done &# 34 ; from the touch screen , in which case the directory will disappear . if the operator selects a field 29 which cannot possibly be assigned ( such as choosing a pressure field while entering the title &# 34 ; intracardiacs &# 34 ;) an error message will be presented . if the operator chooses an empty field 29 directly from the catheter placement screen 43 , the directory 30 will appear , immediately placing the operator in the single select mode of the setup operation . when the operator then makes a label selection , the locations directory 30 disappears and the new information is displayed in the previously empty field 29 . to move a catheter label 28 in the single edit mode , the operator chooses any one of the fields 29 containing the name of the catheter to be moved . all lead names of the catheter will then be highlighted , and the directory 30 will appear . the operator can then choose the new field 29 for the catheter , after which the directory 30 will disappear and the fields 29 will automatically be updated to show the new label 28 . to remove a particular catheter in batch edit mode , the user selects &# 34 ; delete &# 34 ; from the directory 30 and then selects a field 29 associated with the particular catheter to be removed . as shown in fig4 ( c ), a popup containing the question &# 34 ; delete xxxx catheter ?&# 34 ; will appear . if the operator selects &# 34 ; yes &# 34 ;, all leads of the associated catheter will be removed . if the operator selects an invalid field 29 , such as an empty field or fields 29 from pressure or auxiliary input terminals 14 or 16 , an error message will appear . in single select mode , the operator will have first chosen the field 29 of interest , which will cause all similar fields 29 to be highlighted , then will choose &# 34 ; delete &# 34 ; from the directory 30 . all leads of the catheter will then be removed as well as the confirmation question and the directory 30 . as shown in fig4 ( d ), the softkey &# 34 ; new &# 34 ; is available under the auxiliary and pressure directory 30 when in the batch edit mode . when the &# 34 ; new &# 34 ; key is touched , it will be highlighted , and the directory 30 containing a keyboard popup will be presented . the operator may then type in the desired new label 28 and touch &# 34 ; done &# 34 ; to save the entry to the directory 30 . the operator may then select this label 28 for use as input channel label . while working with the auxiliary or pressure channels , the softkey &# 34 ; delete &# 34 ; will also remove a label 28 from the directory 30 to do so , the operator selects &# 34 ; delete &# 34 ;, which will be highlighted , and then chooses the desired label 28 from the directory 30 . a question confirming the operator &# 39 ; s intent will appear and the label 28 will be removed if the operator answers &# 34 ; yes &# 34 ; to the confirming question . zeroing of the pressure channels 14 will also be available to the operator under the pressure directory 30 . the system 10 will zero a pressure channel 14 upon request from the user . if in the single edit mode , the pressure sensor will be open to atmosphere prior to touching the &# 34 ; zero &# 34 ; softkey . the system 10 will then record the pressure signal for one second after the &# 34 ; zero &# 34 ; key is touched and compute an offset value to be used to calibrate the channel . while zeroing is taking place the softkey will be highlighted . in the batch edit mode , the operator chooses the &# 34 ; zero &# 34 ; key and then choose the desired input to be zeroed . the field 29 will then be highlighted while zeroing is taking place . after completion , in either batch or single edit mode , the letter &# 34 ; z &# 34 ; will appear to indicate that the channel has been zeroed as shown in fig4 ( e ). calibration is available while setting up the pressure channels . the &# 34 ; cal &# 34 ; key acts in a similar fashion as the &# 34 ; zero &# 34 ; key as explained above . once &# 34 ; cal &# 34 ; has been touched , a numeric keypad directory 30 as shown in fig4 ( f ) will appear . the directory 30 will preferably contain four preprogrammed &# 34 ; fast cal &# 34 ; keys 45 . the user then applies the calibrating pressure to the transducer and either enters the desired value or choose one of the &# 34 ; fast cal &# 34 ; keys 45 . once calibrated , a &# 34 ; c &# 34 ; will appear in the box beside the label 28 corresponding to the calibrated channel as shown in fig4 ( g ). to select the desired output parameters to be associated with each channel label 28 , the operator selects the &# 34 ; signals &# 34 ; softkey . at this point as shown in fig5 ( a ), the second of the two basic screens called the &# 34 ; signals screen &# 34 ; 44 will appear . the signal screen 44 contains a list of all assigned channels , their pole pairs , gains , filters , limiter level , and the output name of the channel which is to be used by the system 10 for display purposes . the signals screen 44 will behave in a manner similar to the catheter placement screen 43 described above . the signals screen 44 is preferably preprogrammed with the following soft keys : channel key 47 , leads key 48 , gain key 49 , filter key 50 , limiter key 51 , and name key 52 . also , the twelve lead key 31 , catheters key 32 , signals key 33 and restore key 34 are preprogrammed on the signals screen 44 in the identical manner as the catheter placement screen 43 . when a soft key is chosen , the operator is immediately placed in the batch edit mode of the setup operation . as shown in fig5 ( b ), a small &# 34 ; items &# 34 ; popup 46 , displaying information relative to the chosen soft key will appear . the operator must then choose the particular item 53 or label 28 from the popup 46 desired to be used and then choose which field or fields 29 to apply it to . the operator can continue in this way indefinitely . when the operator is finished , &# 34 ; done &# 34 ; can be chosen from the screen and the items popup 46 will disappear . alternatively , if the operator chooses one of the fields 29 directly , the field 29 will be highlighted , and the items popup 46 will appear and place the setup operation into the single edit mode . once the operator selects the desired item 53 , the items popup 46 disappears . at any time the operator can choose the &# 34 ; catheters &# 34 ; key 32 from the signals screen 44 and return to the catheter placement screen 43 , if desired . similarly , at any time in the catheter placement screen 43 , the operator can choose the &# 34 ; signals &# 34 ; key 33 and return to the signals screen 44 . to begin filling out the desired combination of leads for output from the system 10 , the operator first touches the leads key 48 . the items popup 46 for leads will appear as shown in fig5 ( b ) and contain a list of all available ecg , intracardiac , pressure and auxiliary channels . additionally , the ground leads &# 34 ; wct &# 34 ; and &# 34 ; rl &# 34 ; will appear along with the softkeys &# 34 ; delete &# 34 ; and &# 34 ; done &# 34 ;. in single edit mode the user will have previously chosen the desired field 29 to edit and can now choose the label 28 to use to complete an output channel . by touching the desired label 28 , the label 28 will appear in the highlighted field 29 previously selected ( ie . &# 34 ; rva 1 &# 34 ;). the operator &# 39 ; s next touch will complete the lead pair and the label 28 therefore . if the completing pair is of similar type , then the label 28 will be automatically abbreviated to appear as a single prefix and a pair of lead numbers , such as &# 34 ; rva 1 - 2 &# 34 ;. otherwise the second half of the pair will just be appended and the label 28 will appear such as &# 34 ; rva1 - hra1 &# 34 ;. when the operator selects an ecg , auxiliary or pressure channel , the label 28 will appear instantly . once a pairing has been completed , the settings as previously indicated on the signals screen 44 associated with particular cardiac locations will be applied to the gain , filter , and limiter settings . to edit the lead pair in single edit mode , the operator touches the desired field 29 ( which will be highlighted ) and the items popup 46 immediately appears . the operator then touches the desired new labels 28 from the items popup 46 . of course , both labels 28 must be chosen again to complete the new lead pair . to remove a label 28 in batch edit mode , the operator chooses the softkey &# 34 ; delete &# 34 ; from the items popup 46 . this will be highlighted and the operator then chooses the desired label 28 . the label 28 will be removed from the screen as well as all other information for that channel . &# 34 ; delete &# 34 ; will unhighlight after being used once . in single edit mode the desired label 28 will already be highlighted and will be removed once &# 34 ; delete &# 34 ; has been touched . the gains and filters are pre - set to default values which can be changed by the operator if desired . to change the default gains or filters , the operator touches the desired gain key 49 or filter key 50 , and an items popup 46 appears as shown in fig5 ( c ) with all available gains or filter settings respectively . in single edit mode , the operator will have previously chosen the field 29 to be edited and can now select the new item 53 from the presented list . the item popup 46 will then disappear . in batch edit mode the operator first selects the item 53 desired from the item popup 46 and then selects all fields 29 it is desired to apply it to . the operator may continue in this manner until &# 34 ; done &# 34 ; is finally selected . while adjusting filter settings , more than one item 53 may be chosen by the operator . these items 53 include high , low and notch filter settings . each item 53 is highlighted as it is touched to indicate which items 53 have been chosen . if more than one item 53 in a particular section is touched , the highlight moves to the new item 53 . as shown in fig5 ( d ), the limiter setting items popup 46 presents a sliding scale to indicate the relative amount of limiting . to adjust the limiter setting , the operator touches the direction arrows 54 to choose the desired amount . the numbers placed in the appropriate filed 29 on the signals screen 44 corresponding to the limiter reflect percentages of full scale . in single edit mode there is a &# 34 ; done &# 34 ; key on the popup 46 for the operator to indicate when the current level is correct . in batch edit mode , the operator selects the amount of limiting and then applies it to the desired field or fields 29 . &# 34 ; done &# 34 ; is used to indicate when the operator has finished applying new limiting levels to all the desired channels . limiting is applied in a bipolar fashion and is a ± limit . every channel preferably includes a name . the name fields are filled out with predetermined default names as the lead pairs are formed , the default name being the same as the label 28 . however , the user can edit the name using the keyboard presented in the name selection popup 46 , as shown in fig5 ( e ). once completed , the operator can select &# 34 ; done &# 34 ; and the keyboard popup 46 will disappear . in batch edit mode , the operator types in the desired name and then chooses the desired field 29 . whatever is in the keyboard buffer 55 at the time the operator touches the desired field 29 will be used as the name . the operator then touches &# 34 ; done &# 34 ; to remove the popup 46 and exit batch edit mode . single edit mode is operated in a similar manner according to the general format explained above for single edit mode operation . channel editing will only take place in single edit mode . as shown in fig5 ( f ) the operator selects the desired channel number to edit and the items popup 46 appears with all the channels and their names , and the selected channel number is highlighted . upon choosing a new channel which is not already in use , the popup 46 will disappear and the channel will be moved to it &# 39 ; s new position . the old position is initialized to an unused channel . if the operator selects a channel position from the popup 46 which is already occupied , the popup 46 will disappear , and the two channels will swap their configurations . to remove a channel the operator selects &# 34 ; delete &# 34 ; from the popup 46 and the previously selected channel will be removed . an example of a completed signals screen 44 is shown in fig5 ( g ). referring again to the catheter placement screen 43 , as shown in fig6 ( a ), the &# 34 ; catheter placement name &# 34 ; area 56 not only displays the currently invoked catheter placement name but is also active to the touch . once the operator has completed preparations of the system 10 for an electrophysiological procedure , the entire setup may be saved for recall later . to do this the operator simply touches the &# 34 ; catheter placement name &# 34 ; area 56 . if there is already a setup invoked , that setups &# 39 ; name will be present in the area 56 , otherwise the word &# 34 ; setup &# 34 ; will appear . the operator is then presented with the directory 30 containing a list of all available preset catheter setups as well as the softkeys &# 34 ; new &# 34 ;, &# 34 ; delete &# 34 ;, &# 34 ; save &# 34 ;, and &# 34 ; done &# 34 ;. upon selecting &# 34 ; save &# 34 ; the operator is presented with the keyboard directory 30 as shown in fig6 ( b ) and is prompted to fill in the name of the current setup . if a previous catheter placement had been invoked , the operator is prompted to overwrite the old one or to create a new name . &# 34 ; save &# 34 ; will be highlighted while active . the catheter placement setup will be saved both to the system 10 memory and the computer processing unit if attached . to finish saving the operator selects &# 34 ; done &# 34 ; from the keyboard directory 30 . to invoke an old catheter placement the user touches the &# 34 ; catheter placement name &# 34 ; area 56 and is presented with the directory 30 containing the list of currently stored catheter setups as again shown in fig6 ( a ). the operator then chooses the desired catheter setup by touching the desired name on the directory 30 . the catheter placement screen 43 will then be filled with that catheter setups &# 39 ; configuration such as shown in fig6 ( c ). if this is the correct catheter setup , the operator selects &# 34 ; done &# 34 ; and the directory 30 disappears and the hardware is automatically reset with the new catheter setup configuration . to remove a catheter setup configuration , the operator again initiates the catheter placement directory 30 and chooses &# 34 ; delete &# 34 ;. the operator then selects the setup to be removed , and the system 10 asks for confirmation . upon answering &# 34 ; yes &# 34 ;, the setup is removed to return to normal operation the operator selects &# 34 ; done &# 34 ;. to initialize the catheter placement screen 43 , the user selects &# 34 ; new &# 34 ; which will clear all inputs and uninvoke the current catheter setup . the &# 34 ; twelve lead &# 34 ; softkey 31 , and the &# 34 ; restore &# 34 ; softkey 34 are preferably located on opposite sides of the catheter placement name area 56 on the catheter placement screen 43 as shown in fig6 ( a ). by touching the &# 34 ; twelve lead &# 34 ; softkey 31 the operator can toggle the first twelve outputs of the output terminals 17 to receive all twelve ecg leads attached at the input ecg terminal 11 . a directory 30 indicating that the twelve lead ecg is being acquired will appear . the directory 30 contains a softkey containing the previous catheter placements name . to return to the previous placement the operator touches this softkey . if the placement has not yet been named , the softkey will contain &# 34 ; return &# 34 ;. the &# 34 ; restore &# 34 ; softkey 34 is used in the case that signals have drifted off of the computer display monitor on the chart recorder , either from movement of the patient or from a defibrillation , and the operator wishes to remove the dc offset and place the signals back into the middle of the monitor . the restore key 34 will be highlighted for appropriately one second to indicate that the signal placement is automatically being done . the &# 34 ; mark &# 34 ; key 35 is used to mark specific events in time during an electrophysiology procedure . &# 34 ; mark &# 34 ; will be highlighted for appropriately one second after it is pressed to indicate to the operator that the mark is automatically being placed in the time record . the &# 34 ; calibrate &# 34 ; key 37 is used to send a square wave of 1 mv . 5 hz ( rtt ) to all channels . once pressed the key will be highlighted . to stop the calibration pulse the operator presses calibrate key 37 again and the calibration stops . the highlight will also be removed . the &# 34 ; record &# 34 ; key 57 is used to initiate storage of data . recording will start from five seconds previous to when the record key 57 is touched , and storage thereafter is continuous . the operator will be able to stop storage by touching the record key 57 again . the &# 34 ; chart &# 34 ; key 38 delivers a ttl level to the chart recorder if attached to the system . while active , the &# 34 ; chart &# 34 ; key 38 will remain highlighted . to stop the chart recorder , the operator simply presses chart key 38 again . referring now to fig7 if the gain or phase response of the front - end amplifiers 18 are not identical , then the digital signal processor ( dsp ) 22 will not eliminate all of the common mode signal during the common mode signal rejection operation . therefor , an automatic calibration system is included in the system 10 of the present invention to automatically digitally calibrate the front - end amplifiers 18 prior to the initial use of the system 10 in order to correct for any nonuniform phase or gain performance between the front - end amplifiers 18 . the automatic calibration is performed by attaching a cable ( not shown ) from the output channels 17 to all of the intracardiac input channels 12 in parallel . the operator then enters the &# 34 ; calibration mode &# 34 ; of the system 10 and the dsp 22 automatically enters a known signal at each input channel 12 through the attached cable . the dsp 22 then samples the gain and phase of the signal it receives from each of the front - end amplifiers 18 . the difference between the gain and phase value of the known signal and the gain and phase value of the signal as received by the dsp 22 after passing through each front - end amplifier 18 is then digitally stored by the system 10 in a table . thereafter , during normal ( non - calibration mode ) operation of the system 10 , each signal received by the dsp 22 from the front - end amplifiers 18 is corrected by the stored digital value corresponding to the difference between the known calibration mode signal and the received signal of each front - end amplifier 18 . in this manner , any common mode signal received into the dsp 22 will be completely rejected regardless of which inputs 12 are used . since the automatic calibration values are stored digitally in a table in the system 10 , they do not experience any significant drift over the normal life of the system 10 . calibration of the front - end amplifiers 18 therefore is intended to be necessary only as an initial calibration , i . e . one time calibration before initial use of the system 10 . since this automatic calibration need be performed only once , it can be performed by the manufacturer of the system 10 and the subsequent operator will have no need to be concerned with it during normal use . the block diagram of fig7 shows the architecture of the most important internal electronics of the present invention . up to sixty - four electrical inputs from the input terminals of the system 10 are passed through front - end amplifiers 18 and directly into a multiplexer 19 . the signals are multiplexed into four output channels carrying sixteen input channels each and passed through a / d converters 20 and fiberoptics links 21 into the dsp switching board 22 . the dsp 22 operates as a switching matrix , such as in the manner of prior art analog switching matrixes , except that instead of switching analog signals electronically into differential amplifiers , the dsp 22 of the present invention switches digital signals and operates itself as a &# 34 ; differential amplifier &# 34 ;. this is done by electronically combining the digital representations of each signal , such as by subtraction , which results in sixteen output channels ( or thirty - two output channels if desired ) which pass directly into a dsp processing board 23 ( or two dsp processing boards in the case of thirty - two channel outputs from the dsp switching board 22 ). as is readily evident , the dsp switching board 22 of the present invention has been configured for operation to eliminate common mode signals from raw , digitized analog input signals . in this manner , the present invention is distinguished from any prior art use of digital signal processors since common mode signal noise is removed by prior art systems before any digital signal processors are utilized . the prior art use of digital signal processors has been simply to process signals which have previously been passed through an analog switching matrix . in these prior art systems , the common mode rejection function on the analog signals has already been performed through known techniques using differential amplifiers . in the present invention however , the dsp switching board 22 itself operates as a differential amplifier to perform the signal switching operation and to do common mode rejection on the raw digital signals . the signals received by the dsp processing board 23 are processed for gain , signal limiting and the application of a plurality of filters thereto . each dsp processing board 23 ( one in the case of sixteen output channels from the dsp switching board 22 , and two in the case of thirty - two output channels from the dsp switching board 22 ) outputs a single multiplexed channel to a d / a converter 24 which is then passed through a de - multiplexer 25 to restore sixteen channels . these are then passed through an analog filter 26 to the output 17 of the system 10 . the dsp switching board 22 and dsp processing board or boards 23 are driven by an onboard microprocessor 27 which is also operationally attached to the display 15 . in constructing the system 10 of the present invention using the dsp switching board 22 , a large amount of bulk is eliminated therefrom , thus allowing the system 10 to be significantly reduced in size compared to prior art hardware . also , the utilization of the dsp processing board 23 for filter , limiter , and gain application significantly aids in downsizing the overall physical dimensions of the system 10 by allowing elimination of the prior art type filter blocks which commonly include five different capacitors and an analog switch for each signal channel . the result is a system 10 which is significantly smaller than prior art hardware and which is therefore conveniently positionable directly at the patient &# 39 ; s bedside to allow bedside control of the system 10 by the operator during setup and electrophysiology procedures . the system 10 of the present invention can be attached through its output ports 17 by a cable to a computer processing unit , analog monitor , and / or chart recorder . an example of a computer processing unit usable with the system 10 of the present invention is manufactured by quinton electrophysiology corp . of markham , ontario , canada , and is presently being marketing under the trademark &# 34 ; eplab &# 34 ;. since the dsp switching board 22 is used for common mode rejection , it is very advantageous in the present invention to employ a / d converters having very high resolution , such as sixteen bit resolution . the preferred gain ranges for the system 10 include gain ranges of 100 to 5000 for ecg , intracardiac and pressure channels , and gain ranges between 1 and 5000 for the auxiliary channels . the system 10 employs three different filters , including high pass filters in the range of dc , 0 . 05 hz , 1 . 0 hz , 10 hz , and 30 hz , low pass filters in the range of 40 hz , 100 hz , 200 hz , and 400 hz , and notch filters in the range of 50 or 60 hz . the common mode rejection level is preferably set at greater than 100 db . it will be apparent from the foregoing that , while particular embodiments of the invention have been illustrated and described , various modifications can be made thereto without departing from the spirit and scope of the invention . accordingly , it is not intended that the invention be limited , except as by the append claims .
a signal management system is disclosed which integrates a patient switching box and an amplification unit into a single compact enclosure sufficiently small to reside by the patient table in an operating unit set up for electrophysiology procedures . the system includes a front panel designed to accept standard ecg leads , a plurality of intracardiac leads , including leads available for stimulation and / or lesion generation , and a plurality of pressure channels . the front panel also includes a touch screen display to allow quick assignments of labels to each of the ecg or intracardiac leads and the pressure channels . the system includes an onboard microprocessor which allows any operation performed on the system to be automatically updated on a remote computer processing unit if attached , and vice - versa . digital signal processors are used in the system to perform switching operations on the electrical signals received , and to perform gain , limiting and / or filtering processes thereon . the digital signal processors are automatically calibrated to adjust for varying gain and phase performance of the inputing front - end amplifiers which deliver the electrical signals from the intracardiac leads .