Systematic measurements of whole-body imaging dose distributions in image-guided radiation therapy.
PURPOSE
The full benefit of the increased precision of contemporary treatment techniques can only be exploited if the accuracy of the patient positioning is guaranteed. Therefore, more and more imaging modalities are used in the process of the patient setup in clinical routine of radiation therapy. The improved accuracy in patient positioning, however, results in additional dose contributions to the integral patient dose. To quantify this, absorbed dose measurements from typical imaging procedures involved in an image-guided radiation therapy treatment were measured in an anthropomorphic phantom for a complete course of treatment. The experimental setup, including the measurement positions in the phantom, was exactly the same as in a preceding study of radiotherapy stray dose measurements. This allows a direct combination of imaging dose distributions with the therapy dose distribution.


METHODS
Individually calibrated thermoluminescent dosimeters were used to measure absorbed dose in an anthropomorphic phantom at 184 locations. The dose distributions from imaging devices used with treatment machines from the manufacturers Accuray, Elekta, Siemens, and Varian and from computed tomography scanners from GE Healthcare were determined and the resulting effective dose was calculated. The list of investigated imaging techniques consisted of cone beam computed tomography (kilo- and megavoltage), megavoltage fan beam computed tomography, kilo- and megavoltage planar imaging, planning computed tomography with and without gating methods and planar scout views.


RESULTS
A conventional 3D planning CT resulted in an effective dose additional to the treatment stray dose of less than 1 mSv outside of the treated volume, whereas a 4D planning CT resulted in a 10 times larger dose. For a daily setup of the patient with two planar kilovoltage images or with a fan beam CT at the TomoTherapy unit, an additional effective dose outside of the treated volume of less than 0.4 mSv and 1.4 mSv was measured, respectively. Using kilovoltage or megavoltage radiation to obtain cone beam computed tomography scans led to an additional dose of 8-46 mSv. For treatment verification images performed once per week using double exposure technique, an additional effective dose of up to 18 mSv was measured.


CONCLUSIONS
Daily setup imaging using kilovoltage planar images or TomoTherapy megavoltage fan beam CT imaging can be used as a standard procedure in clinical routine. Daily kilovoltage and megavoltage cone beam computed tomography setup imaging should be applied on an individual or indication based protocol. Depending on the imaging scheme applied, image-guided radiation therapy can be administered without increasing the dose outside of the treated volume compared to therapies without image guidance.