Features and Benefits • Three Models with operating ranges up to 100, 200, or 250°C
• Bath capacities ranging from 3 to 12 liters
• Redundant safety temperature control and low liquid cut-off
• Built-in cooling coil for counter cooling
Click on the image to view a larger photo of the instrument.
Constant Temperature Circulating Baths Standard Model–Constant temperature circulating bath with analog temperature controller, large LED display, and an operating range to 100°C with ±0.03°C stability. Choice of 3-4.5L or 7.5-12L bath capacities. Circulator has adjustable pumping speeds (3 to 15 Lpm), and hose-barb fittings for external circulation. A separate adjustable thermostat provides safety cutoff. Bath exterior is constructed of galvanized steel with an acrylic enamel finish.
Elite Model–Similar in design and construction to the standard model, but features a microprocessor-based temperature controller with drift compensation, an operating range to 200°C, an improved temperature stability of ±0.01°C, an RS-232 interface, visual and audible overtemperature warning, and low liquid level protection.
Ultra Model–Similar in design and construction to the elite model, but features a programmable controller with self-optimizing Intelligent Cascade Control (ICC), Integrated Programmer (6 profiles, up to 60 steps), 4-line interactive LCD display, an operating range to 250°C, pumping speed up to 20 Lpm, an RS-232/485 interface, and an external port for a Pt-100 sensor.
Specifications Temperature Range (with built-in counter-cooling option): Standard model: 20°C to 100°C Elite model: 20°C to 200°C Ultra model: 20°C to 250°C
Temperature Stability: Standard model: ±0.03°C Elite and Ultra models: ±0.01°C
Electrical Requirements 115V 60Hz, Single Phase, 8.7A 230V 50Hz, Single Phase, 8.7A