{"word_count": 1875, "character_count": 13527}
After a long wait, we are pleased to introduce our mobile meteorological measurement station. Thanks to this station, we have the opportunity to observe microclimate conditions in and around spatio-climatic devices at pedestrian level. The spatio-temporal mobility of observations is a key element in thermal comfort assessment in outdoor urban spaces (Häb et al., 2015; Le Bras, 2015; Liu et al., 2017; Lenzholzer et al., 2018); and it has many advantages: e.g. a greater spatial resolution of microclimatic properties, the ease of access of spatio-climatic devices and its urban environments, a wider scope of measurement and reduced costs (Le Bras, 2015; Liu et al. 2017; Yokoyama et al., 2018).There are many non-stationary studies, based on different microclimatic sensors and supports: portable in a backpack (Brandsma and Wolters 2012; Le Bras, 2015; Camponovo et al., 2016; Tsin et al., 2016; Chokhachian et al., 2017; Gallinelli et al. , 2017) or coming in an earpiece format (Potvin, 2000), attached to bicycles (Brandsma and Wolters, 2012; Drach and Drach, 2014; Heusinkveld et al. , 2014; Liu et al., 2017; Rajkovich and Larsen, 2016; Yokoyama et al., 2018), on carts (Kastendeuch et al., 2016;  Nakayoshi and Nakashima, 2018; Middel and Krayenhoff, 2019), on diabolos (Klok et al., 2018) or on electric golf trolleys (Häb et al., 2015). Inspired by the work of Middel and Krayenhoff (2019), we opted for robust, scientific sensors mounted on a hand-drawn cart. For non-stationary measurements, a very short response time was targeted (Table 1). Thus, the choice of equipment was based on an exhaustive bibliographic study including about thirty publications on the thermal comfort evaluation of the pedestrian and the urban microclimate in outdoor spaces. Microclimatic variables, which will be measured, include air temperature, relative humidity, wind direction and speed, and long- and short-wave solar radiation. Air temperature and relative humidity A mixed probe will provide measurements of air temperature and relative humidity. For that, the HMP155A Vaisala model has been chosen. Its installation is under a RAD14 weather shelter with natural ventilation, protecting the sensors from bad weather (precipitation, sun, wind, etc.) and insect intrusion. WindThe wind speed and direction are measured by the 2D ultrasonic anemometer, WindSonic type1. Its handling is simple and without moving parts, which reduces costs and maintenance time. Solar radiation Concerning the net radiation measurements, the NR01 sensor from Hukseflux was chosen. It is a robust set, consisting of a pair of pyranometers and a pair of pyrgeometers. Thanks to the representation of the net radiation in its components of short wavelength radiation and long wavelength infrared radiation, it is used in scientific works aiming a detailed representation of energy balances. In our case, the mobile meteorological station is equipped with 3 NR01 sensors, oriented laterally east-west, north-south and vertically between the sky and the ground. These measurements will then be used to determine the mean radiant temperature (Höppe, 1993; Spagnolo and Dear, 2003; Ali-Toudert and Mayer, 2007; Thorsson et al., 2007; Guo et al., 2020). Data logger The sensors are connected to the Campbell Scientific CR1000X data logger and control unit with a CDM-A116 expansion module. It supports sensor management, direct and remote communications, data analysis, external device control, and data and program storage. Its autonomy is ensured by a 12V - 24Ah battery (BP24) allowing measuring and storing data during several consecutive days of field trip. The set is placed in a GRP protection box type ENC 16/18. All sensors, with the exception of thermocouples, and the data logger are from Campbell Scientifics. These sensors have been tested and calibrated at the factory by professionals. Quand un projet se concrétise – volet microclimat Après une longue attente, nous avons le plaisir de vous présenter notre station de mesures microclimatiques mobile. Grâce à cette station, nous avons la possibilité d’observer les conditions microclimatiques dans et autour des dispositifs spatio-climatiques au niveau du piéton. La mobilité spatio-temporelle des observations est un élément clé dans l’évaluation du confort thermique en espace urbain extérieur&#160;(Häb et al., 2015; Le Bras, 2015; Liu et al., 2017; Lenzholzer et al., 2018) et présente des nombreux avantages, comme par exemple une plus grande résolution spatiale des propriétés microclimatiques, la facilité d’accès des dispositifs spatio-climatiques et ses environnements urbains, un périmètre de mesures élargi et des coûts réduits (Le Bras, 2015; Liu et al., 2017; Yokoyama et al., 2018). Il existe de nombreuses études non-stationnaires se basant sur différents capteurs microclimatiques et supports&#160;: portables dans un sac à dos (Brandsma and Wolters, 2012; Camponovo et al., 2016;  Le Bras, 2015; Tsin et al., 2016; Chokhachian et al., 2017; Gallinelli et al., 2017) ou en format d’écouteur (Potvin, 2000), ou bien fixés sur des vélos (Brandsma and Wolters, 2012; Drach and Drach, 2014; Heusinkveld et al., 2014; Liu et al., 2017; Rajkovich and Larsen, 2016; Yokoyama et al., 2018), sur des chariots (Kastendeuch et al., 2016; Nakayoshi and Nakashima, 2018; Middel and Krayenhoff, 2019), sur des diabolos (Klok et al., 2018) ou sur des chariots de golf électriques (Häb et al., 2015). Inspirés par les travaux de Middel and Krayenhoff (2019), nous avons opté pour des capteurs à caractère scientifique et robuste fixés sur un chariot tiré à la main. S’agissant de mesures non-stationnaires, un temps de réponse très court a été visé (Tableau 1). Ainsi, le choix d’équipement s’est basé sur une étude bibliographique exhaustive comprenant une trentaine de publications sur l’évaluation du confort thermique du piéton et du microclimat urbain en espace extérieur. Les variables microclimatiques mesurées sont notamment la température de l’air, l’humidité relative, la direction et la vitesse du vent et le rayonnement solaire à grande et courte longueur d’onde. Température de l’air et humidité relative Une sonde mixte assure les mesures de température de l’air et d’humidité relative. Il s’agit du modèle HMP155A Vaisala. Son installation est sous abri météorologique de type RAD14 avec une ventilation naturelle, afin de protéger les capteurs des intempéries (précipitations, soleil, vent, etc.) et d'intrusion d'insectes. Vent La vitesse et la direction du vent sont mesurées par un anémomètre ultrasonique 2D, de type WindSonic1. Sa manipulation est simple et sans parties mobiles, ce qui réduit le coût et le temps de la maintenance.  Rayonnement solaire Pour les mesures du rayonnement net, le capteur NR01 de chez Hukseflux a été choisi. C’est un ensemble robuste, composé d’un couple de pyranomètre et de pyrgéomètre. Grâce à la représentation du rayonnement net dans ses composantes de rayonnement à courtes longueurs d'ondes et de rayonnement infrarouge de grandes longueurs d'ondes, il est utilisé dans des travaux scientifiques ayant pour objectif une représentation de qualité du bilan énergétique. Pour le projet, la station météorologique mobile est équipée de 3 capteurs NR01, orientés latéralement est-ouest, nord-sud et verticalement entre le ciel et le sol. Ces mesures serviront ensuite à la détermination de la température rayonnante moyenne (Höppe, 1993; Spagnolo and de Dear, 2003; Ali-Toudert and Mayer, 2007; Thorsson et al., 2007; Guo et al., 2020). Centrale de mesures et de stockage Les capteurs sont connectés à une centrale de mesure et de contrôle de Campbell Scientific CR1000X avec un module d’extension CDM-A116. Elle permet la gestion des capteurs, les communications directes et à distance, les analyses des données, le contrôle des périphériques externes ainsi que le stockage des données et des programmes. Son autonomie est assurée par une batterie 12V - 24Ah (BP24) permettant de mesurer et de stocker les données pendant plusieurs journées de terrain consécutives. L’ensemble est placé dans un coffre de protection GRP de type ENC 16/18. L’ensemble de capteurs, à l’exception des thermocouples, et la centrale de mesures proviennent de chez Campbell Scientific. Ces capteurs ont été testés et étalonnés en usine par des professionnels.References Ali-Toudert, F., Mayer, H., 2007. Thermal comfort inan east–west oriented street canyon in Freiburg (Germany) under hot summerconditions. Theoretical and applied climatology 87, 223–237.Brandsma, T., Wolters, D., 2012. Measurement andStatistical Modeling of the Urban Heat Island of the City of Utrecht (theNetherlands). J. Appl. Meteor. Climatol. 51, 1046–1060.https://doi.org/10.1175/JAMC-D-11-0206.1Camponovo, R., Gallinelli, P., Guillot, V., 2016. CityFeel- An innovative protocol and instrument to better understand urbanmicroclimate. Presented at the Forschen für den Bau im Kontext von Energie undUmwelt, Genève.Chokhachian, A., Santucci, D., Vohlidka, P., Auer, T.,2017. Framework for Defining a Transient Outdoor Comfort Model in Dense UrbanSpaces, Processes &#38; Findings. Presented at the 7th International DoctoralConference, Architecture and Urbanism: Contemporary Research, Faculty ofArchitecture CTU, Prague, pp. 51–54.Drach, P., Drach, H., 2014. Mobile MeteorologicalSurvey Station: Applying Measurement Tools on a bike to create the Meteobike,in: Counting the Cost of Comfort in a changing world. Presented at the 8thWindsor Conference, Cumberland Lodge, London, UK: Network for Comfort andEnergy Une in Buildings.Gallinelli, P., Camponovo, R., Guillot, V., 2017. CityFeel- micro climate monitoring for climate mitigation and urban design. EnergyProcedia 122, 391–396. https://doi.org/10.1016/j.egypro.2017.07.427Guo, H., Aviv, D., Loyola, M., Teitelbaum, E.,Houchois, N., Meggers, F., 2020. On the understanding of the mean radianttemperature within both the indoor and outdoor environment, a critical review.Renewable and Sustainable Energy Reviews 117, 109207.https://doi.org/10.1016/j.rser.2019.06.014Häb, K., Middel, A., Ruddell, B.L., 2015. Relationshipbetween Land Use and Microclimate based on Mobile Transect Measurements.Presented at the 9th International Conference on Urban Climate, July 20-24, 2015,Toulouse.Heusinkveld, B.G., Steeneveld, G.J., Hove, L.W.A. van,Jacobs, C.M.J., Holtslag, A. a. M., 2014. Spatial variability of the Rotterdamurban heat island as influenced by urban land use. Journal of GeophysicalResearch: Atmospheres 119, 677–692. https://doi.org/10.1002/2012JD019399Höppe, P.R., 1993. Heat balance modelling. Experientia49, 741–746. https://doi.org/10.1007/BF01923542Kastendeuch, P.P., Najjar, G., Philips, N., Nerry, F.,Roupioz, L., Colin, J., Luhahe, R., 2016. Mesures pour l’étude des ambiances climatiques à Strasbourg lors de lacanicule 2015. Presented at the Colloque AIC, Besançon.Klok, L., Rood, N., Kluck, J., Kleerekoper, L., 2018.Assessment of thermally comfortable urban spaces in Amsterdam during hot summerdays. International Journal ofBiometeorology 63.Le Bras, J.,2015. Le microclimat urbain à haute résolution : mesures et modélisation (Thèsede doctorat). Université de Toulouse, Toulouse.Lenzholzer,S., Klemm, W., Vasilikou, C., 2018. Qualitative methods to explore thermo-spatialperception in outdoor urban spaces. Urban Climate, ICUC9: The 9th InternationalConference on Urban Climate 23, 231–249.https://doi.org/10.1016/j.uclim.2016.10.003Liu, L., Lin, Y., Wang, D., Liu, J., 2017. An improvedtemporal correction method for mobile measurement of outdoor thermal climates.Theor Appl Climatol 129, 201–212. https://doi.org/10.1007/s00704-016-1769-yMiddel, A., Krayenhoff, E.S., 2019.Micrometeorological determinants of pedestrian thermal exposure duringrecord-breaking heat in Tempe, Arizona: Introducing the MaRTy observationalplatform. Science of The Total Environment 687, 137–151.https://doi.org/10.1016/j.scitotenv.2019.06.085Nakayoshi, M., Nakashima, T., 2018. Performance ofGlobe Anemo-radiometer in Mean Radiant Temperature Evaluation in Stationary andMobile Observation. Presented at the 10th International Conference on UrbanClimate/14th Symposium on the Urban Environment, American MeteorologicalSociety.Potvin, A., 2000. Assessing the microclimate of urbantransitional spaces, in: Proceedings of PLEA2000. Presented at the Passive LowEnergy Architecture, Cambridge.Rajkovich, N.B., Larsen, L., 2016. A Bicycle-BasedField Measurement System for the Study of Thermal Exposure in Cuyahoga County,Ohio, USA. Int J Environ Res Public Health 13.https://doi.org/10.3390/ijerph13020159Spagnolo, J., de Dear, R., 2003. A field study ofthermal comfort in outdoor and semi-outdoor environments in subtropical SydneyAustralia. Building and Environment 38, 721–738. https://doi.org/10.1016/S0360-1323(02)00209-3Thorsson, S., Lindberg, F., Eliasson, I., Holmer, B.,2007. Different methods for estimating the mean radiant temperature in anoutdoor urban setting. International Journal of Climatology 27, 1983–1993.https://doi.org/10.1002/joc.1537Tsin, P.K., Knudby, A., Krayenhoff, E.S., Ho, H.C., Brauer, M., Henderson, S.B., 2016. Microscale mobile monitoring of urban air temperature. Urban Climate 18, 58–72. https://doi.org/10.1016/j.uclim.2016.10.001 Yokoyama, H., Ooka, R., Kikumoto, H., 2018. Study of mobile measurements for detailed temperature distribution in a high-density urban area in Tokyo. Urban Climate 24, 517–528. https://doi.org/10.1016/j.uclim.2017.06.006 AuthorsXenia Laffaille and Ignacio Requena