Cryoneurolysis is performed with a cryoprobe, which is composed of a hollow cannula that contains a smaller inner lumen. The pressurized coolant (nitrous oxide, carbon dioxide or liquid nitrogen) travels down the lumen and expands at the end of the lumen into the tip of the hollow cannula. No coolant exits the cryoprobe. The expansion of the pressurized liquid causes the surrounding area to cool (known as the Joule–Thomson effect) and the phase change of the liquid to gas also causes the surrounding area to cool. This causes a visible iceball to form and the tissue surrounding the end of the cryoprobe to freeze. The gas form of the coolant then travels up the length of the cryoprobe and is safely expelled. The tissue surrounding the end of the cryoprobe can reach as low as −88.5 °C with nitrous oxide as the coolant, and as low as −195.8 °C with liquid nitrogen. Temperatures below −100 °C are damaging to nerves.
Cryo-S Painless cryoanalgesia device is the next generation of apparatus used by many experts in the field since 1992. The working medium for Cryo-S Painless is carbon dioxide:  (−78 °C) or nitrous oxide:  (−89 °C), very efficient and easy to use gases. Cryo-S Painless is controlled by a microprocessor and all the parameters are displayed and monitored on a LCD screen. Mode selection probe, cleaning and freezing can be performed automatically using footswitch or touch screen which allows to keep the site of a procedure under sterile conditions. Electronic communication (chip system) between the connected probe and device allows recognition of optimal operating parameters and auto-configures to cryoprobe characteristics. Pressure and gas flow are set automatically, any manual adjustment is not necessary. Cryoprobe temperature, cylinder pressure, gas flow inside of cryoprobe and procedure time are displayed during freezing. Built-in voice communication Built-in neurostimulation (sensory, motor).