The beginnings of tandem mass spectrometry (MS/MS or MSn) date back to the first mass-analyzed ion kinetic energy spectrometer (MIKES) developed at Purdue University. Tandem MS, the production and mass analysis of fragment ions from mass-selected precursor ions, is particularly useful for complex mixture analysis and has served as the backbone of fields as diverse as proteomics, forensics, environmental monitoring, and biomarker discovery.
Amongst the activation methods for MS/MS are collision-induced dissociation (CID), ultraviolet photo dissociation, infrared multiphoton dissociation, electron transfer dissociation, surface-induced dissociation, and others. Collision-induced dissociation has been especially notable in the development of the suite of MS/MS scan modes which includes three prominent members—product ion scans, precursor ion scans, and neutral loss scans—as well as other notable modes—doubly charged ion scans, reaction intermediate scans, multiple reaction monitoring, and functional relationship scans.
Although neutrals are not directly measurable by mass spectrometers, they are indirectly accessible by a variety of methods and they carry important analytical information. The two most prominent techniques for probing neutral species are neutralization-reionization mass spectrometry (NRMS) and the neutral loss scan in MS/MS. The NRMS experiment neutralizes a mass-selected ion, usually by charge exchange or CID, and the resulting neutral undergoes energetic collisions which produce neutral fragments that are re-ionized and mass analyzed. Hypervalent and other unusual species can be produced and characterized, a unique capability.
By contrast, in a neutral loss MS/MS experiment a precursor ion is mass-selected by a first mass analyzer and undergoes activation to produce a product ion and a neutral. The product ion is mass selected for detection by a second analyzer. For the neutral loss scan, the relationship between the precursor ion mass-to-charge ratio (m/z) and the product ion m/z is fixed—that is, the neutral mass is constant—and as such it describes a shared molecular functionality of a group of precursor ions. In comparison, the precursor ion scan selects a fixed product ion m/z which might also correspond to a common functionality in all precursor ions which yield this fragment.
Because mass selection of both precursor and product ion is necessitated in precursor ion and neutral loss scans, the prevailing wisdom in mass spectrometry has been that multiple mass analyzers are required.
USPTO Backgrounds
USPTO Backgrounds