<?xml version="1.0" encoding="UTF-8" standalone="no" ?><CalVal xmlns:xsi="http://www.w3.org/2001/XMLSchema-instance" xsi:noNamespaceSchemaLocation="../CVALData.xsd ">
   <SensorID>
    <MxAssetID>23344</MxAssetID>
    
  </SensorID>
  <StreamCalVal>
    <CalFixture>
      <CalFixtureID/>
      <ActivityLocation/>
      <SWVersion>1.0.12</SWVersion>
      <ProcedureVersion/>
      <Station/>
    </CalFixture>
    <CalEnvironment>
      <AmbientTemperature>
        <Min>25.0</Min>
        <Max>25.0</Max>
        <Mean>25.0</Mean>
        <StdDev>0.00</StdDev>
      </AmbientTemperature>
      <AmbientPressure>
        <Min>760.0</Min>
        <Max>760.0</Max>
        <Mean>760.0</Mean>
        <StdDev>0.00</StdDev>
      </AmbientPressure>
      <AmbientHumidity>
        <Min>25.0</Min>
        <Max>25.0</Max>
        <Mean>25.0</Mean>
        <StdDev>0.00</StdDev>
      </AmbientHumidity>
    </CalEnvironment>
    <CalibrationCoefficient> 
		<Name>CVALA1</Name>
		<Value>0.1023052258071</Value>
    </CalibrationCoefficient>
	<CalibrationTable name="">
    <Metadata>
        <Entry>
            <Name> </Name>
            <Value> </Value>
        </Entry> 
    </Metadata>
	<Row index="">
    <Independent name=""> </Independent>
    <Column index="">
        <Dependent name=""> </Dependent>
    </Column>
	</Row>
	</CalibrationTable>
    <CalibrationStandards CertNumber=""> </CalibrationStandards>
    <StreamID>1</StreamID>
    <Uncertainty units="uS/cm"> 
		<Name>U_CVALR4</Name>
		<Value>105</Value>
	</Uncertainty><Uncertainty units="uS/cm"> 
		<Name>U_CVALR2</Name>
		<Value>50</Value>
	</Uncertainty><Uncertainty units="uS/cm"> 
		<Name>U_CVALR1</Name>
		<Value>104</Value>
	</Uncertainty><Uncertainty units="uS/cm"> 
		<Name>U_CVALR3</Name>
		<Value>0.00585</Value>
	</Uncertainty><Uncertainty units="uS/cm"> 
		<Name>U_CVALR2</Name>
		<Value>0.000141</Value>
	</Uncertainty><Uncertainty units="uS/cm"> 
		<Name>U_CVALR1</Name>
		<Value>0.00585</Value>
	</Uncertainty>
    <Trueness units="uS/cm">0.000000</Trueness>
    <ActivityTimeRange>
      <StartTime>2018-12-11T17:47:38.0000</StartTime>
      <EndTime>2018-12-11T17:48:35.0000</EndTime>
    </ActivityTimeRange>
    <CertificateNumber>162901</CertificateNumber>
  </StreamCalVal>
  <ValidTimeRange>
    <StartTime>2020-01-11T17:48:35.0000</StartTime>
    <EndTime>2020-02-08T17:48:35.0000</EndTime>
  </ValidTimeRange>
  <Comment>
    <CommentDescription/>
    <CommentCreator/>
    <CommentDateTime/>
  </Comment>
  <DATA>
		<MetaData>
			<Equipment> </Equipment>
			<Operator>MKelley</Operator>
			<WorkOrder>WO30149</WorkOrder>
			<MaximoID>23344</MaximoID>
			<AssetID>30000000011952</AssetID>
			<Reference/>
			<Fixture>L3F200</Fixture>
			<NEONPN>f0320170001</NEONPN>
			<SN header="">15L102472</SN>
			<FW header="">3.0.5</FW>
			<SOH header=""/>
			<UncertaintyCtl>L3F200</UncertaintyCtl>
		</MetaData>
		<PreCALDATA header=""/>
		<CALDATA header="">EXO User Cal File
File Created:12/11/2018 11:47:38 AM

[General]
Device = "Conductivity/Temp"
SN = "15L102472"
Cal Parameter = "SpCond µS/cm"
Cal Started = TRUE
Raw Data = "Cond RAW µS/cm"
Last Cal Time = "11/27/2018 18:38:33"
Cal Start Time = "12/11/2018 17:47:38"
Additional Input 1 = "N/A"
Additional Input 2 = "N/A"
Additional Input 3 = "N/A"

[Device Info]
Product = 1
Model = 3
SubModel = 1
Firmware Version = "3.0.5"

[Sonde Info]
Type = "EXO2 Sonde"
ID = "Sonde 15K101963"
SN = "15K101963"
Firmware Version = "1.0.35"

[KOR Info]
Hardware = "EXO Desktop"
KOR Version = "1.0.12"
Worksheet Version = "1"

[Additional Info]
Cond Factory Gain = "5.10"
ODO Membrane SN = ""
ODO Last Updated = ""
Wiper Factory Home = ""
Depth Mounting = ""
Depth Offset = ""
Depth Altitude = ""
Depth Latitude = ""
Cal Points Used = ""

[Current Coeffs]
A0 = 1.008114

[Cal Point 1]
Standard = 1000.000000
Type = ""
Manuf = "YSI"
Lot No = "060907"
Cal Point Accepted = TRUE
Temperature = 11.041000
Pre Value = 1000.900000
Post Value = 1000.000000
Raw Value = 728.164368
Additional Input 1 = NaN
Additional Input 2 = NaN
Additional Input 3 = NaN
Stable = TRUE

[Results]
Cal Completed = TRUE    
Cal Applied = TRUE
Cal Valid = TRUE
Device Removed = FALSE
Uncalibrated = FALSE
QC Score = 2
Cal Time = "12/11/2018 17:48:35"

[New Coeffs]
A0 = 1.007182</CALDATA>
		<ValidationDATA header=""/>
		<QCStatic header="">2.000000,2.000000,10.000000,0.950000,0.000000,0.000000,3,uS/cm</QCStatic>
		<QCDynamic header=""/>
		<Precision header=""/>
		<AsLeftCFG header="">New polys
1.007182
</AsLeftCFG>
		<AsReceived>
			<ReceivedCFG header="">Old polys
1.008114
</ReceivedCFG>
			<ReceivedCoefficient> 
				<Name> </Name>
				<Value> </Value>
			</ReceivedCoefficient>
			<ReceivedDATA header=""/>
		</AsReceived>
  </DATA>
  </CalVal>